Parameters used to simulate the receiver sensitivity performance of InP, InAlAs, and InP and InAlAs APDs.
1. Introduction
The avalanche photodiode (APD) is widely used in optical fibre communications (Campbell, 2007) due to its ability to achieve high internal gain at relatively high speeds and low excess noise (Wei et al., 2002), thus improving the system signaltonoise ratio. Its internal mechanism of gain or avalanche multiplication is a result of successive impact ionisation events. In an optical receiver system, the advantage of internal gain, in the APD, is experienced when the amplifier noise dominates that of a unitygain photodiode. This increases the signaltonoise ratio (SNR) and ultimately improves the receiver sensitivity as the gain increases until the APD noise rises to become dominant.
Indium Phosphide (InP) is widely used as the multiplication layer material in commercially available APDs for applications in the 0.9–1.7µm wavelength region with In_{0.53}Ga_{0.47}As grown latticematched to it as the absorption layer. It has been predicted that Indium Alluminium Arsenide (In_{0.52}Al_{0.48}As) will replace InP, as a more favourable multiplication layer material due to its lower excess noise characteristics (Kinsey et al., 2000). In comparison to InP, tunnelling currents remain lower in InAlAs due to its larger bandgap. While holes ionise more readily than electrons in InP, the opposite holds true for InAlAs and InGaAs, as electrons ionise more readily than holes; thus making the InGaAs/InAlAs combination superior to InGaAs/InP in a SAM APD, in terms of lower excess noise, higher gainbandwidth product, and improved sensitivity. Studies have also shown that the breakdown voltage of InAlAs APDs is less temperature dependent compared to InP (Tan et al., 2010), which would be useful in temperature sensitive applications, thus making temperature control less critical.
The sensitivity performance criterion for digital receivers is its biterror rate (BER), which is the probability of an error in the bitidentification by the receiver. The receiver sensitivity is defined as the minimum average optical power to operate at a certain BER; 10^{12} being a common standard for digital optical receivers. The sensitivity of APDbased high speed optical receivers is governed by three main competing factors, namely the excess noise, avalanchebuildup time and dark current of the APD. Generally, the excess noise and avalanchebuildup time increases with APD gain. Thus, for a fixed multiplication layer thickness, there is a sensitivityoptimised gain that offers a balance between SNR while keeping the degrading contributions from the excess noise factor and intersymbolinterference (ISI) at a minimum. More importantly, changing the thickness of the multiplication layer strongly affects the receiver sensitivity, as the aforementioned three factors change. Reducing the thickness of the multiplication layer serves to reduce the excess noise factor, due to the dead space effect, (Li et al., 1998) and minimise ISI via reducing carrier transit times across the avalanche region. On the other hand, the increase in the field in thin layers accentuates tunnelling currents at exponential rates (Forrest et al., 1980a). Thus, careful attention is required when determining the multiplication layer thickness for an optimum APD design.
It is, therefore, very useful and interesting to model the sensitivity of an APDbased receiver system accurately. Such models have been developed but none included some form of dark current mechanism, which can significantly affect the receiver’s sensitivity. Characterisation of the APD excess noise factor in test structures is also necessary in order to model the BER of an APDbased receiver system. Several efforts have been made to systematically characterise promising detector material systems including InP and InAlAs.
In this chapter, we will describe the model used to investigate the receiversensitivityoptimisation of InP and InAlAs APDs, which include dark current contributions from tunnelling current. A comprehensive assessment of the measurement systems reported in the literature is also provided followed by two suggestions for an improved design. The results of the BER calculations on receiver systems using InP APDs will be presented, followed by a discussion on the competing effects of performancedetermining factors. A straightforward comparison between InP and InAlAs APDs will then be presented with an analysis on the difference.
2. Impact ionisation
The impact ionisation process occurs when a carrier injected into a highfield region gains enough energy from the applied field and collides with the lattice structure to produce an electronhole pair. In an electroninitiated process, as depicted schematically in Figure 1(a), an energetic electron at a higher state of the conduction band scatters with an electron at the top of the valence band via Coulombic interaction, and promotes it to the bottom of the conduction band (Singh, 1995). As this process can have a cascading effect, the net result is the creation of many secondary electrons and holes from a single primary electron, generated through absorption of a photon. A similar process occurs in holeinitiated impact ionisation with similar results, as shown in Figure 1(b).
Due to conservation of energy and momentum, a threshold energy,
The generation rate or mean number of ionisation events per unit distance for a carrier is known as the impact ionisation coefficient. The electron and hole ionisation coefficients,
Carriers with energy less than
The mean multiplication factor,
where
Assuming a uniform electric field, i.e. an ideal
Hence, pure electron mean multiplication factors,
Rearranging equations #,
3. Excess noise
The stochastic nature of the impact ionisation process results in fluctuations in the multiplication factor. This noise, introduced by impact ionisation, is caused by the unpredictability in the production position of the secondary carrier.
For an APD under illumination, assuming the incident photons have a Poisson distribution generating a primary photocurrent,
where
For a measurement circuit with bandwidth
where
and from (8) and (9), noting that
These simplified derivations show that even without avalanche gain, variance in the photocurrent is expected due to the random nature of the photocurrent generation. Note, also, that (11) is identical to the shot noise formula for variance in a current.
Hence, for an APD considered as an
and the mean square noise current is given by
Equation (13) describes the ideal (noiseless) multiplication process, where the stochastic nature of the avalanche multiplication process is excluded. To account for the noise associated with the multiplication process, the excess noise factor,
where
Equation (15) shows that the average multiplication,
4. The Random Path Length model
Unlike the local model described earlier, nonlocal models account for the dead space and one such model is the Random Path Length (RPL) model (Ong et al., 1998). The RPL model is a simple model that is able to predict multiplication and excess noise characteristics in APDs by modelling the transport of carriers during the impact ionisation process. The model operates by consideration of the ionisation path length probability distribution function,
where
and
and the mean ionisation coefficient is the reciprocal of this, that is
From (16), the probability that a carrier travels a distance
Thus, a random electron ionisation path length,
Similar expressions for the hole impact ionisation path length can be obtained by substituting
The RPL simulation is composed of
where
5. BER model
The current developed in an APD, by chains of impact ionisation events, take time to build up. Materials with disparate ionisation coefficients tend to have longer chains of ionisation events; thus having longer current buildup times compared to currents developed by shorter chains. This buildup time, which is stochastic, has an associated bandwidth limit and thus governs the APD speed and ultimately, the level of ISI in the receiver system.
To understand the stochastic nature of the APD builduptimelimited bandwidth and its statistical correlation with the gain, Sun
The Gaussianapproximation method was used to calculate the BER and is described as follows. The output of the integrateanddump receiver was approximated by a Gaussian random variable with the exact mean and variance, and the BER was computed using (Agrawal, 1997)
where
which is a convenient approximation to the optimal decision threshold that minimises the BER (Agrawal, 1997). The expressions for the parameters
where
The expressions in (26)–(29) are generalisations of the traditional expressions for the output statistics of APDbased receivers found in optical communication literature (Agrawal, 1997). Whilst these expressions capture the usual effects of shot noise and the excess noise due to avalanche multiplication, they additionally capture the effects of ISI, relative speed of the detector, as well as the stochastic coupling between the APD’s gain and buildup time through the effective use of the parameters
The term in
5.1. Variant of the Gaussianapproximation
In the previous section, it was assumed that the receiver output, conditional on the state of the present bit, is a Gaussian random variable. Ong et al. (Ong et al., 2009) replaced this with the more realistic assumption that the receiver output, conditional on the state of the present bit
The mean and variance of the ISI contributions in the receiver output from the
Considering an arbitrary past bit pattern,
where
Similarly, the variance of the receiver output associated with the pattern
5.2. Generalisation of the model to include tunnelling current
The unmultiplied bandtoband tunnelling current,
where
The expression for
while the expression for
The receiver mean output,
Similarly, adding the contributions from the photons in the current bit to
Finally, for every past bit pattern
where
The bitlength parameter,
6. Measurement of excess noise: A survey
Estimation of the BER, which can be obtained from a certain diode fabricated from a particular material system, relies on the characterisation of the excess noise factor using one or more test structures. The test structure is often a
Several excess noise measurement systems have been reported in the literature and comparisons between the circuits described may be drawn. The figures of comparison are,
The system signal to noise ratio, where the signal is defined as full shot noise exhibited by 1μA.
The maximum permissible APD junction capacitance. In the case of multifrequency systems the lowest available frequency is used; this produces the most favourable result. It is assumed that the system input impedance and diode junction capacitance form a first order low pass network.
6.1. Measurement systems
The first report of noise measurements on photodiodes was by Baertsch (1966). Insufficient information is provided to estimate this system's figures of merit so it is excluded from the comparison. Xie et al. (1993) proposed a measurement system that was substantially similar to Toivonen et al. (1992). The Xie et al. (1993) system represents both. Bulman (1983), Ando and Kanbe (1981) and Lau et al. (2006) presented systems based on phase sensitive detection. Xie et al. (1993) and Toivonen et al. (1992) used a DC approach.
6.1.1. Bulman’s system
Figure 2 shows the system reported by Bulman (1983). It is a PSD system in which photocurrent and excess noise are extracted and read out by two lock in amplifiers. The APD is loaded by the AIL13680 through the bias tee. It may be assumed that the APD experiences a 50Ω load. The preamplifier output is fed to a receiver having a calibrated bandwidth. The resulting signal proportional to the noise power contained within the calibrated bandwidth is passed to a lockinamplifier.
Bulman proposes two methods to quantify the absolute noise power measured. Firstly a
Bulman's report lacks some information regarding the front end amplifier. An Analog Devices AD9618 low noise opamp in noninverting mode is used as a model. It achieves a gain of 100V/V and a bandwidth of 80MHz with 50Ω input impedance. The equivalent input noise voltage is 1.94 nV/Hz^{½}. Using this model as an approximation the signaltonoise ratio for Bulman’s system is
6.1.2. The PSD system of Ando and Kanbe
Ando and Kanbe (1981) reported the system shown in Figure 3. It is a PSD system in which the APD is loaded by 50Ω due to the input impedance of the IF amplifier. The measurement system bandwidth is defined by the IF amplifier and is 1MHz centred on 30MHz. The APD is biased using a bias tee and a variable voltage source. The noise power is read out from a lock in amplifier. A power meter and signal generator, with its output passed through a calibrated attenuator, provides a means of relating the absolute signal power to the value measured leaving the IF amplifier. Photocurrent is extracted by DC measurement of the current entering the bias tee. The measurement of devices exhibiting high dark current is therefore difficult with this system. The various connections required to calibrate the system and perform measurements are made using relays.
Ando and Kanbe did not report any attempts to measure shot noise on their system. They also do not give information regarding the model numbers or manufactures of their system components. No noise specifications for the instrumentation are given. Assuming that their system adds no noise other than the thermal noise of the 50Ω input impedance within the measurement bandwidth then the signaltonoise ratio can be computed using
The authors claim that noise power as low as 130dBm/Hz can be measured with 0.5dB accuracy. This represents a current of 0.125μA developing full shot noise.
6.1.3. A measurement after Xie et al.
The system proposed by Xie et al. (1993) is similar to that proposed by Toivonen et al. (Toivonen et al., 1992). The APD is connected to a microstrip line and DC voltage is applied via a bias tee.
The measurement is made using a CW light source and a noise figure meter such as the Hewlett Packard 8970A. The system has two significant advantages over PSD systems such as those of Bulman (1983) and Li (Lau et al., 2006). Several measurement frequencies are available up to the limit of the circuits or analyser. Presently Agilent Technologies manufactures noise figure meters capable of measuring 10MHz to 26GHz with variable effective measurement bandwidth. This upper limit can be increased by using heterodyne methods. Xie’s system (Xie et al., 1993) was limited to 1.3GHz maximum measurement frequency and 4MHz noise measurement bandwidth. The measurement is, in principle, quicker than a PSD system. The operation of PSD is discussed fully elsewhere (Horowitz and Hill, 1989) but it is sufficient to realise that the time constant of a PSD measurement may be expected to be longer than of a noise figure meter. DC measurements have several disadvantages over PSD however. For example the lowest practically measurable photogenerated noise is higher in CW systems than in some PSD systems. Using a transimpedance amplifier, Li (Li, 1999, Li et al., 1998) has shown that the transimpedance amplifier reported by Lau et al. (2006) can be used as the basis of a noise measuring system with greater (less negative) noise signal to noise ratio than is possible by using a 50Ω measurement system. A further objection to CW systems is that the noise without illumination – the dark noise  should be periodically measured in order to maintain consistency. The dark noise should be stable and sufficiently small, compared to the noise with illumination – combined light and dark noise – that the noise with illumination is dominated by the light noise. If this condition is not met the confidence of the measurement is compromised. Xie et al. (1993) reported measuring noise power as low as 182dbm/Hz without difficulty using the CW system shown in Figure 4. In a 50Ω system 182dbm/Hz is equivalent to full shot noise generated by 8μA of photocurrent. The capacitance which can be tolerated by this measurement system is computed at the lowest useable frequency, as this produces the most favourable result. By the same first order approximation used in Bulman’s and Ando and Kanbe’s systems Xie’s system will exhibit a 3dB (half power) bandwidth of 10MHz when loaded with 636pF.
6.1.4. A PSD system after Li et al.
The system of Li (Lau et al., 2006, Li, 1999) employs phase sensitive detection and a transimpedance amplifier. A schematic diagram is shown in Figure 5.
The laser is chopped by mechanical means at 180Hz and is presented to the diode via a system of optics which is not shown. The TIA is used to convert the diode current into a voltage. This voltage is amplified using a commercial low noise wide band amplifier module (Minicircuits ZFL500). A precision stepped attenuator (HP355D) is used to vary the system gain permitting measurement of high and low noise devices. The noise signal is separated from the low frequency component of the photocurrent by a Minicircuits SBP10.7+ LC ladder filter which also defines the noise measurement bandwidth. After filtration, the signal resembles an amplitude modulated noise waveform, where periods of diode illumination produce greater noise amplitude than periods of darkness. Further amplification follows, prior to a wide band squaring and averaging circuit. The output of the squaring and averaging circuit is an approximately square voltage signal, the amplitude of which is proportional to the noise power contained in the measurement bandwidth. The fundamental frequency of the noise power signal is 180Hz. The squaring circuit is based on an Analogue Devices AD835 analogue multiplier. The averaging circuit is a first order RC filter with a time constant of approximately 100μs. The output from the squaring and averaging circuit is measured using a lockinamplifier. The photocurrent signal is taken from an auxiliary output of the TIA where the amplitude of the 180Hz square wave is proportional to the photocurrent. The photocurrent signal is measured on a second lockinamplifier.
The system after Li (Lau et al., 2006, Li, 1999) is superior in noise performance to prior reported systems. The transimpedance amplifier provides a signal to noise ratio which is superior to that possible in a 50Ω system. Consider the connection of a photodiode and a 50Ω resistor. Assume that full shot noise generated by i_{ph} = 1μA flows through the resistor which exhibits thermal noise at T = 300°K. The noise signal to noise ratio is then,
The dynamic range of Li’s system is limited at the lower bound by the ability of the lock in amplifier to extract the inphase excess noise signal from the system’s background noise. Practical experimentation by the authors and their colleagues has shown that full shot noise developed by 1μA is approaching the limit and the shot noise from 0.1μA is not reliably measurable. The precise limit is difficult to quantify because it is affected by the prevailing electromagnetic conditions both radiated (passing through the experiment volume) and conducted into the power supply lines. At the upper bound the maximum attenuation of the stepped attenuator provides a limitation however more attenuation could be added without difficulty. The linearity of the transimpedance amplifier at high input current is a second limit. When driven from +/5V supplies a TIA with a gain of 2200V/A will saturate at approximately 2.25mA input current. Because the relationship between excess noise factor and photomultiplication varies between material systems it is unwise to speculate the maximum multiplication which can be used. Furthermore if a device is available which can be operated with a very large gain the optical illumination may be reduced in order to reduce the multiplied photocurrent and the excess noise power. In this way higher multiplication values may be measured. In order to measure lower multiplication values a larger primary photocurrent is required. By performing two or more measurements with differing primary photocurrents it is possible, assuming the APD is sufficiently robust, to measure multiplication and excess noise power over any desirable range above the system limit.
The capacitance tolerated by Li’s transimpedance amplifier (Lau et al., 2006, Li, 1999) is lower than all of the other systems. The interaction of the APD junction capacitance and the feedback capacitor permits the existence of resonance in the transimpedance amplifier. When the capacitance is sufficiently large oscillation breaks out and the measurement system is saturated. There limit of measureable junction capacitance is however not governed by the presence of oscillation. A result of the interaction of the diode junction capacitance and the feedback capacitance is a dependence of the effective noise power bandwidth of the system on the diode junction capacitance, which is itself dependant on the DC bias voltage applied to the APD. As a result a correction to the measurement bandwidth must be made when processing the measurement data. The limitation of the measurable device capacitance is governed by the quality of the correction which can be achieved and by the presence of oscillation. While it is known that up to 56pF does not cause oscillation, Li placed the limit at 28pF (Li, 1999). This limit was obtained by calibrating the bandwidth of the transimpedance amplifier with several values of capacitance. Having performed the calibration, shot noise due to photogenerated carriers was measured using a unitygain silicon photodiode. A second data set was gathered in which extra capacitance was placed in parallel with the photodiode to simulate a diode of greater capacitance. The simulated higher capacitance shot noise data was processed using the original calibration. The quality of the fitting of the standard photodiode shot noise and the simulated extra capacitance shot noise data was used as a basis for defining the quality of the correction and hence the maximum capacitance.
6.2. An improved CW noise measurement
We propose two possible improvements to the design proposed by Xie et al. (1993). Both are essentially improvements to the method by which the instrumentation is calibrated. The introduction of a calibrated noise source (HP346B) permits the use of direct noise figure measurement – as opposed to hot/cold measurements, which is a considerable improvement. The noise figure meter (N8973A, or an older model such as the N8970) is designed such that the noise source is connected to the device (for example an LNB) under test. Of course if the device is an electrooptical transducer this is impossible as there is no place to attach the noise source. This leads to the use of a pretest calibration followed by hot/cold measurements. It would be preferable to use the noise figure analyser (NFA) according to its design principle, i.e. with the noise source in the measurement. The NFA is provided with prior calibration  by the manufacturer  of the noise source’s contribution to the system. The system gain is also computable by measuring the effect on the noise output when the noise source is switched on and off  it is pulsed by the NFA. The time average of the change in noise level can provide the gain from the noise input port to the NFA input port. The prior knowledge of the known noise input from the calibrated source (HP346B) allows the NFA to compute the gain and noise figure nearly instantly, a considerable improvement in measurement speed, accuracy and precision. The question is then “How can the noise source be applied to the APD?” It cannot be directly applied. However, a secondary port can be created which permits the connection of an APD and the noise source to the NFA simultaneously. We provide two example designs here, the first uses a 50Ω matched topology similar to that of Xie et al. (1993). The second describes a similar overall structure but using a commercial transimpedance amplifier.
The APD multiplication, excess noise factor and noise power bandwidth can be established simultaneously in one measurement. The limitation of the system bandwidth can be alleviated by two methods. Firstly a higher maximum frequency noise figure meter can be obtained. Agilent Technologies presently manufactures noise figure meters/analysers capable of directly measuring up to 26GHz. The use of heterodyne techniques could extend this considerably. However a relatively inexpensive alternative is to use a lower bandwidth noise figure meter but begin measuring bandwidth once the APD has been biased to achieve a high gain. The high frequency roll off due to a finite gain bandwidth product can be observed at lower frequencies; the unity noise gain bandwidth product can then be inferred. The importance of correct impedance matching cannot be overemphasized.
6.2.1. 50Ω system
The system diagram in Figure 6 shows the structure of the measurement setup. A SourceMeasure Unit
A precision voltage source and current measuring device, e.g. Keithley models 237, 2400 and 2612
drives a bias tee composed of L_{1} and C_{1}. An example of a suitable tee is the PicoSecond Model 5541A. The APD is connected to a microwave DC block (C_{1}) and this is in turn connected to a termination (50Ω). The DC block and the termination must be electrically close to the APD even at the highest measurement frequency. It is preferable to fabricate the DC block and the 50Ω termination with the APD as an integrated circuit. From the point of view of the first amplifier the APD is a Norton source coupled to the end of a properly terminated transmission line. Approximately half of the noise power will escape to ground via R_{1}, the rest will enter the measurement system. It is possible to calibrate the measurement system either manually (i.e. use a 50Ω signal generator to list a table of adjustments for each frequency and post process the measured device data based on these reading) or automatically by using the HP 346B Noise source connected to the first amplifier input instead of the APD. The attenuator setting must be noted down when the calibration is carried out. The first amplifier in the chain must be of the lowest possible noise. Examples include Minicircuits ZFL1000LN+, ZX6033LN+ and Pasternack PE1513. The ZFL1000 has low noise and a reasonably flat gain vs. frequency profile from 100kHz to 1GHz however bandwidth is limited to 1GHz. The ZX6033LN+ has exceptionally low noise, and reasonable gain vs. frequency characteristics from 50MHz to 3GHz. The PE1513 has relatively poor noise especially as frequency increases, the gain vs. frequency profile is not ideal either; however it is the only device which covers the whole frequency range of the NFA, which is 3 GHz in the case of the N8973A. Unless APDs possessing bandwidths below 50MHz are to be routinely measured the authors preferred choice is the ZX6033LN.The specifications of the second and third amplifiers are considerably less critical than the first. Any microwave device with reasonable noise and gain vs. frequency characteristics will be acceptable. The stepped attenuator should be of the precision type for example the Trilithic RSA35100 (0dB to 100dB in 10dB steps) would be ideal. The power combiner may be of any type which covers the required bandwidth. A suitable resistive splitter/combiner is the Minicircuits ZX10E14S+.
The maximum device capacitance is approximately 2pF to obtain a 3dB point of approximately 3GHz. R_{1} must be electrically close to the APD, consequently it is unlikely that the noise contribution of this resistor could be minimised by cooling as was reported by Xie et al. (1993). If the APD was measured at low temperature however it would be plausible to place R_{1} and C_{1} in the cryostat chamber with the APD, thus obtaining a noise advantage at lower temperatures. A laser is often used to excite electrooptical transducers in characterisation experiments. In this case the laser should be a gas laser possessing a single longitudinal mode, preferably frequency and amplitude stabilised. The authors have met with little success in noise characterisation experiments using semiconductor lasers, the laser relative intensity noise (RIN) is often too great to permit measurement of the detector noise.
6.2.2. TIA CW noise measurement system
The structure of this measurement system is nearly identical to the 50Ω system previously described. The principle difference is the use of a transimpedance amplifier front end instead of a 50Ω system. Figure 7 shows the system diagram.
C_{1} provides an AC ground for the APD such that the very great majority of the noise current flows into the TIA. Example TIAs are given in the figure. Commercial TIAs often have input impedance which is not a good approximation to a virtual earth. As a result the maximum permissible device capacitance is often lower than in the 50Ω system case, and is dependent on the particular TIA in use. The MAX3910 provides ~9GHz small signal bandwidth and nearly linear output voltage to input current relationship for photocurrents in the range 0 to 900μA_{pkpk}. The small signal gain of this TIA is approximately 1.6kV/A in the linear region.
Unlike the 50Ω system it is not possible to connect the noise source to the TIA input for calibration purposes. Impedance matching considerations preclude it. This is a major limitation of the TIA measurement compared with the 50Ω measurement. Calibration of the TIA signal path with the noise source is only possible at the TIA output. A plausible method of calibration is to use a unity gain wide band
7. 10 Gb/s optical communications receiver BER analysis
This section will use the model described in section 3 to analyse the sensitivity of an APDbased receiver system by first investigating the performance of a 10 Gb/s receiver system using InP APDs followed by a discussion on the competing effects of excess noise, APD bandwidth, and tunnelling current on the receiver sensitivity. Similar calculations will then be performed for systems using InAlAs APDs to provide a straightforward and fair comparison with InP.
7.1. Parameters and coefficients
The nonlocal impact ionisation coefficients and threshold energies of Tan et al. (2008) for InP and Goh et al. (2007a) for InAlAs are used due to the extensive electric field range over which they are valid. The unmultiplied tunnelling current (Forrest et al., 1980b) defined by Equation (34) will use reported experimental InP (Tan et al., 2008) and InAlAs (Goh et al., 2007b) tunnelling fitting parameters. Since the tunnelling fitting parameters vary with avalanche width, the lowest value, 1.16 for InP and 1.26 for InAlAs, was used for all investigated avalanche widths to assume the worst case scenario. The Johnson noise due to the TIA in the receiver at 10 Gb/s was assumed to be 636 electrons per bit, corresponding to an input noise current density of 10.7 pA/Hz^{½}. Calculations were performed for a series of InP and InAlAs APDs, with active area radius of 15μm and avalanche widths ranging from 0.1 to 0.5µm. A complete list of the parameters used in this section is shown in Table 1.




0.68 


0.7 


2.8 


3.0 


1.344 


0.08 




7.2. InP APD optimisation
Sensitivity versus gain curves were calculated for the InP APDs and the results are shown in Figure 8. The key observation is that for each APD, there exists an optimum mean gain that achieves the lowest sensitivity. In Figure 9, the optimum sensitivity for each device and corresponding mean gain are plotted as functions of the avalanche region width. This allows identification of the optimum avalanche width for a given transmission speed, thereby yielding the optimised sensitivity for a given transmission speed; in this case, 10 Gb/s. The calculations predicted an optimum avalanche width of 0.19 μm for InP APDs, yielding a sensitivity of 28.1 dBm at a gain of 13 for a 10 Gb/s system.
7.3. Competing performancedetermining factors
In order to independently assess the significance of (i) ISI, (ii) device bandwidth, and (iii) tunnelling current, three additional sets of calculations were carried out, which shall be referred to as
Results from each of these three sets of incomplete calculations are compared to those from the
For the complete calculation results, high sensitivity values for diodes narrower than the optimum avalanche width optimum are due to high tunnelling current. For diodes wider than the optimum avalanche width, sensitivity increases with
7.4. Comparison of InP and InAlAs APDs
The optimum sensitivity (optimized over the mean gain) and its corresponding mean gain from the InP and InAlAs calculations are plotted against the avalanche region width, as shown in Figure 11, for a 10 Gb/s system. The calculations predict an optimum
For any given width, InAlAs provides better sensitivity than InP. However, the improvement is not significant. At their respective optimum avalanche widths, the difference in receiver sensitivities is only 0.5 dBm at both transmission speeds, corresponding to a reduction of 11% in optical signal power at the receiver input. This marginal improvement was also reported by Marshall et al. (2006) albeit with higher sensitivity values, as a result of ignoring the effects of APD bandwidth and ISI. The modesty in this improvement is partly due to a diminishing advantage, as
8. Conclusions
In this chapter the impact ionisation process, from the perspective of APD detector design, has been introduced. The beneficial multiplicative effect on current, and the associated detrimental current fluctuations, excess noise, has been derived. The RPL model has been introduced. This model is routinely used to compute the multiplication and excess noise of thick and thin APD structures. A comprehensive survey of the measurement systems used to characterise the excess noise properties of photodiode structures has been presented, and two improved measurement systems have been suggested. A BER model which includes ISI, excess noise, and tunnelling current has been outlined. The key performancedetermining factors which influence the APD and receiver design choices have been analysed. A comparison of InAlAs and InP APDs has been presented and InAlAs offers a marginal sensitivity improvement. An example 10 Gb/s detector and receiver combination has been presented for InAlAs and InP APDs.
Acknowledgments
The work reported here was carried out in the Department of Electronic and Electrical Engineering at the University of Sheffield, UK, within the research group of Prof. John David and Dr. Jo Shien Ng, whom the authors thank most sincerely for securing the necessary funding and helping to direct the work.
Daniel S. G. Ong is funded by the University of Sheffield studentship and James E. Green is funded by Engineering and Physical Sciences Research Council (EPSRC).
References
 1.
Agrawal G. P. 1997  2.
Ando H. Kanbe H. 1981 Ionization coefficient measurement in GaAs by using multiplication noise characteristics.  3.
Baertsch R. D. 1966 Lowfrequency noise measurements in silicon avalanche photodiodes.  4.
Bulman G. E. 1983  5.
Campbell J. C. 2007 Recent advances in telecommunications avalanche photodiodes.  6.
Forrest S. R. Di Domenico M. Jr Smith R. G. Stocker H. J. 1980a Evidence for tunneling in reversebiased IIIV photodetector diodes.  7.
Forrest S. R. Leheny R. F. Nahory R. E. Pollack M. A. 1980b In_{0.53}Ga_{0.47}As photodiodes with dark current limited by generationrecombination and tunneling.  8.
Goh Y. L. Marshall A. R. J. Massey D. J. Ng J. S. Tan C. H. Hopkinson M. David J. P. R. Jones S. K. Button C. C. Pinches S. M. 2007a Excess avalanche noise in In_{0.52}Al_{0.48}As.  9.
Goh Y. L. Massey D. J. Marshall A. R. J. Ng J. S. Tan C. H. Ng W. K. Rees G. J. Hopkinson M. David J. P. R. Jones S. K. 2007b Avalanche multiplication in InAlAs.  10.
Kinsey G. S. Hansing C. C. Holmes A. L. Streetman B. G. Campbell J. C. Dentai A. G. 2000 Waveguide In_{0.53}Ga_{0.47}AsIn_{0.52}Al_{0.48}As avalanche photodiode.  11.
Lau K. S. Tan C. H. Ng B. K. Li K. F. Tozer R. C. David J. P. R. Rees G. J. 2006 Excess noise measurement in avalanche photodiodes using a transimpedance amplifier frontend.  12.
Li K. F. 1999  13.
Li K. F. Ong D. S. David J. P. R. Rees G. J. Tozer R. C. Robson P. N. Grey R. 1998 Avalanche multiplication noise characteristics in thin GaAs p^{+}in^{+} diodes.  14.
Marshall A. R. J. Goh Y. L. Tan L. J. J. Tan C. H. Ng J. S. David J. P. R. 2006 A comparison of the lower limit of multiplication noise in InP and InAlAs based APDs for telecommunications receiver applications. 2006 IEEE LEOS Annual Meeting Conference Proceedings, Vols 1 and2 789 790 .  15.
Mc Intyre R. J. 1966 Multiplication noise in uniform avalanche diodes.  16.
Ong D. S. Li K. F. Rees G. J. David J. P. R. Robson P. N. 1998 A simple model to determine multiplication and noise in avalanche photodiodes.  17.
Ong D. S. G. Ng J. S. Hayat M. M. Sun P. David J. P. R. 2009 Optimization of InP APDs for highspeed lightwave systems.  18.
Singh J. 1995  19.
Stillman G. E. Wolfe C. M. 1977 Avalanche photodiodes.  20.
Sun P. Hayat M. M. Saleh B. E. A. Teich M. C. 2006 Statistical correlation of gain and buildup time in APDs and its effects on receiver performance.  21.
Tager A. S. 1965 Current fluctuations in a semiconductor (dielectric) under the conditions of impact ionization and avalanche breakdown.  22.
Tan L. J. J. Ng J. S. Tan C. H. David J. P. R. 2008 Avalanche noise characteristics in submicron InP diodes.  23.
Tan L. J. J. Ong D. S. G. Ng J. S. Tan C. H. Jones S. K. Qian Y. H. David J. P. R. 2010 Temperature dependence of avalanche breakdown in InP and InAlAs.  24.
Toivonen M. Salokatve A. Hovinen M. Pessa M. 1992 GaAs/AlGaAs deltadoped staircase avalanche photodiode with separated absorption layer.  25.
Wei J. Dries J. C. Wang H. S. Lange M. L. Olsen G. H. Forrest S. R. 2002 Optimization of 10Gb/s longwavelength floating guard ring InGaAsInP avalanche photodiodes.  26.
Xie F. Z. Kuhl D. Bottcher E. H. Ren S. Y. Bimberg D. 1993 Wide band frequency response measurements of photodetectors using lowlevel photocurrent noise detection.
Notes
 A precision voltage source and current measuring device, e.g. Keithley models 237, 2400 and 2612