Paprosky classification systems for femoral defects.
Total hip arthroplasty (THA) is one of the most successful surgical procedures with well documented survivorship at up to 25 years. With ageing of the population and higher arthritis prevalence in older adults, the demand for the procedure increases worldwide . In addition, over the last two decades the age range has been broadened to include younger patients. Over 270 000 hip replacements are performed annually in the US alone, and the annual volume of hip joint replacement is projected to double by the year 2030 . Although very successful procedure, significant percentage of patients undergoing total hip arthroplasty require revision within 10 to 15 years after the surgery. Aseptic loosening and the associated osteolysis have been recognized as the main reason for implant failure in 71% of cases . Other indications for revision include periprosthetic fracture, dislocation, and infection. New technologies in implant design and advances in surgical technique have improved the outcomes after primary total hip arthroplasty and decreased the rate of complications. However, as a consequence of increased rate of primary THA's the prevalence of revision hip surgery is increasing proportionally. The increased rate and costs of revision procedures impose high demands on both surgeon and healthcare system. Moreover, the cost of hip replacement is exponentially increasing .
Bone loss is the major challenge in revision setting. In 2009, Bozic et al. reviewed the most common causes for revision hip arthroplasty . Aseptic loosening, instability, and infection were reported as the main reasons for revision surgery. This study underlined the need for a complex approach to evaluation and management of patients with implant failure after hip replacement. Such approach will guarantee precise diagnosis, proper selection of revision implant and surgical approach, uncomplicated surgery, and optimal clinical result.
This chapter provides an overview of aseptic loosening of revision hip arthroplasty and outlines the management strategies in the clinical scenario of a failed hip prosthesis.
2. Patient evaluation
Various signs and symptoms can occur in the clinical setting of a failed hip prosthesis. Painful hip arthroplasty is the most common complication after total hip arthroplasty reaching 18% of patients in some series . Most of these painful hips will require revision. Groin pain can be referred to implant failure easily whereas occasional hip pain, pain in the buttock, knee pain or migrating pain can have different etiology. Other diseases and conditions such as disk disease, radiculopathy, inguinal or femoral hernia, pelvic infections, tumors, and trauma may have manifestations similar to that of a failed prosthesis.
The differential diagnosis of hip pain requires a careful history and examination. In simple cases, the reason could be identified with clinical examinations and standard radiographs only. Thorough examination elicits the underlying cause of hip complains such as infection, neurological injury, referred pain, wear, aseptic loosening or instability. In many cases, the diagnosis is a challenge to the surgeon. In addition to clinical history and physical examination, radiographic examination and advanced imaging techniques could help establish exact localization of pain, and its possible connection with the implant. Additional radiographic examinations as well as an algorithmic approach with special diagnostic imaging and tests help establishing precise diagnosis. Computed tomography and 3-D computed tomography is often helpful in establishing periprosthetic osteolysis and its severity. In addition to plain radiographs arthrography with contrast medium could be considered in certain cases.
Once extrinsic and periarticular diseases have been excluded as a reason for the hip pain, septic loosening should be excluded. Laboratory investigations are the initial tests that help differentiate septic from aseptic loosening. A standard set includes WBC, Erythrocyte sedimentation rate (ESR), C-reactive protein (CRP), complete chemistries and urinalysis. In addition to plain radiographs and laboratory tests joint aspiration is considered the most important diagnostic tool in ruling out periprosthetic infection. The aspirate should be sent for cell count and anaerobic and aerobic cultures. Recently, local markers such as interleukin-6 (IL-6) and other cytokines , synovial CRP, and leukocyte esterase (LE)  from joint aspirate have been proposed. All of these local markers have shown accuracy of more than 90% in predicting periprosthetic infection. Nuclear medicine scans with technetium-99m-HDP, gallium citrate, and labeled WBC have been used to diagnose the presence of infection. However, because of poor sensitivity, specificity, and accuracy, it is cost-prohibitive and remains a tertiary tool. Nuclear medicine is used only if infection could not be proven otherwise. Intraoperatively, diagnostic evaluations such as Gram stains and frozen sections have been proposed.
Guidelines and algorithms for evaluation of painful hip arthroplasty have been published in the literature and implemented in practice . Such approach helps eliminate infection of the failed hip that would change treatment approach and could exclude one-stage revision.
3. Classification systems for bone defects
It is important to have a practical, relatively simple classification system for assessment of bone defects associated with loose hip implants. The use of a radiographic classification system helps to establish the severity and localization of bone defects, and to guide treatment decisions. It should allow the surgeon to be prepared for the possible intraoperative findings and to plan adequate treatment approach. Numerous classification systems have been described in the literature [17,23,25,32,40,58,60,99,116,137,158].
3.1. AAOS classification
The American Academy of Orthopedic Surgeons (AAOS) classification system of bone defects, described by D’Antonio et al. identifies the pattern and localization of osteolysis but does not quantify the bone loss [23-25]. It is one of the most widely used classification system in the literature.
3.2. Paprosky classification
Perhaps the most widely used classification system, the Paprosky Classification [32,116,158] (Tables 1, 2) was developed to establish bone defect type, size, and localization in order to allow selection of appropriate cementless reconstructive option for a given bone loss pattern. We base our clinical decisions on this classification system.
The key advantage of this classification is the assessment of the host bone ability to provide initial stability of a cementless implant until bone ingrowth occurs. The bone defects are usually classified on the basis of plain radiographs. However, final assessment is made intraoperatively, after removal of the failed implant and thorough debridement of the host bone. Intraoperative assessment of implant stability is made with help of trial components. The remaining host bone determines the stability of the implant and the type of the defect.
|1||Minimal metaphyseal bone loss|
|2||Extensive metaphyseal bone loss and an intact diaphysis|
|3A||Extensive metadiaphyseal bone loss and a minimum of 4 cm of intact cortical bone in the diaphysis|
|3B||Extensive metadiaphyseal bone loss and <4 cm of intact cortical bone in the diaphysis|
|4||Extensive metadiaphyseal bone loss and a nonsupportive diaphysis|
|1||Acetabular rim, anterior-posterior column intact|
|2||Less than 3 cm superior migration|
Distorted acetabular rim. Intact anterior and posterior columns
Adequate stability with Trial. Greater than 50% contact surface
|2A||Superior and medial cavitation defect. Intact rim|
|2B||Segmental supero-lateral defect (less than 1/3 of circumference)|
|2C||Medial defect with cup medial to Kohler’s line (Protrusio)|
|3||Greater than 3 cm superior migration|
Non-supportive acetabular rim for biological fixation
|3A||Lateral to Kohler’s line. Intact medial support|
Moderate ischial lysis (<15 mm below superior obturator line)
Medial limb of teardrop is intact
Superior and lateral migration “up and out”
Contact of trial with bone over 40-60%
|3B||Broken Kohler’s line. No medial or superior support|
Extensive ischial osteolysis (>15 mm below superior obturator line)
Complete destruction of tear drop
Superior and medial migration “up and in”
Under 40% contact surface. High risk of occult pelvic discontinuity
4. Preoperative planning
Careful preoperative planning is a prerequisite for successful revision surgery. The principle aims in revision hip arthroplasty are to achieve supportive host bone, secure implant fixation and to restore hip center and joint kinematics. The type and severity of host bone loss determine the method of reconstruction. Careful preoperative planning improves effectiveness during surgery, and helps distinguish more complex alternatives for reconstruction if needed.
Thorough clinical and radiographic examination is essential for determining the extent and severity of bone loss, quality of the host bone, exclusion of infection, additional deformities, and potentially confounding factors. Computed tomography may be needed in the presence of massive bone loss. In case of medial migration of the failed components angiography with contrast medium should be considered. Manual or digital templating helps for adequate selection (size, diameter, length) of revision implant and reduce the operative room inventory. Templating also helps determine whether stable initial fixation could be obtained and the need for additional procedures. Preoperative planning is critical for the assessment of the need of graft, tools for implant removal, and selection of proper components available at the time of surgery. Appropriate surgical exposure should be planned with an extensile approach often necessary. Classification system of bone defects based on radiographs that assesses the severity of bone loss according to the type of fixation for a given bone loss pattern is beneficial. In our practice, we find the Paprosky classification system a useful tool for guidance of clinical decisions. Any attempt should be made to identify the failed implant. The implant manufacturer should be contacted for implant-specific extraction devices if available. In case of isolated partial revision, it is advisable to have an option for partial (liner or head) exchange.
5. Surgical approaches in revision arthroplasty
The aims of revision surgery are to extract the failed prosthesis with minimal soft tissue and bone damage, to restore bone loss, and to implant prosthesis with stable and durable fixation. Ultimate goals are long-lasting and painless joint function. To obtain these goals arthroplasty surgeons may require a variety of approaches for adequate exposure of the femur and acetabulum in different revision settings. Usually arthroplasty surgeons are familiar and most comfortable with a certain approach and use it in most surgeries. However, in order to obtain reproducible results after revision of most difficult cases, surgeons should be familiar with all approaches to the hip joint. Next to standard approaches used in primary total hip arthroplasty, extensile approaches were developed in order to minimize damage to the host bone, safely remove the loose implant and provide good visualization for correct insertion of the revision components. Contained defects can be reconstructed through any conventional approach. For uncontained defects, we prefer to have wide access and, therefore, we use transtrochanteric approach or trochanteric slide osteotomy with preserved insertion of vastus lateralis. If greater exposure is needed extended trochanteric osteotomy is advisable.
5.1. Extended trochanteric osteotomy
It is safe and straightforward, saving time and minimizing risk of fracture during cement and failed implant removal. However, it limits femoral component options to those that rely on distal fixation. Advantages of the technique are: predictable healing of the osteotomy, decrease in intraoperative fractures and femoral perforations, direct access to the distal canal for cement removal and neutral reaming, and decreased surgical time [9,104]. Favorable clinical results after use of ETO have been published in the literature [9,104].
6. Surgical revision options
Majority of hip revisions could be performed using cementing technique. However, patients with severe bone loss and poor bone quality require complex alternatives for revision.
6.1. Options for Femoral revision
6.1.1. Cemented fixation of the femoral component
Various studies publish the outcomes after cemented femoral revision [2,13,122]. Use of the early cementing technique produced disappointing results [2,69]. Gaining more extensive experience resulted in acceptable short- to mid-term results [13,81,122]. Re-revision rates ranging from 4.3% to 6.0% and radiographic loosening ranging from 12% to 44% after mid-term follow-up of 3.4 to 4.5 years were reported (Table 3). However, long-term studies showed suboptimal outcomes after revision with an early cementing technique. At 8.1-year follow-up of the initial group of cemented revisions, Pellicci et al. found more than doubled incidence of re-revision and radiographic loosening ranging from 5.4% to 19% and from 13.6% to 29%, respectively . Similar results were published by Kavanagh et al. in a 10-year follow-up study . Sixty-four per cent of the stems had been revised or were radiographically loose. The incidence of revision had more than doubled from 18% at 3 years to 39% at final follow-up.
|Amstutz et al. ||1982||66||2.1||9.0||29.9|
|Pellicci et al. ||1982||110||3.4||5.4||13.6|
|Pellicci et al. ||1985||99||8.1||19.0||29.0|
|Kavanagh et al. ||1985||166||4.5||6.0||44.0|
|Engelbrecht et al. ||1990||138||7.4||8.8||38.0|
|Rubash and Harris ||1988||43||6.2||2.0||11.0|
|Katz et al. ||1997||79||11.9||5.4||16.0|
|Callaghan et al. ||1985||139||3.6||4.3||12.0|
|Estok and Harris ||1994||38||11.7||10.5||10.5|
|Mulroy and Harris ||1996||43||15.1||20.0||6.0|
The main reason for suboptimal results with early cemented revisions was difficulty in obtaining stable and long-lasting fixation in compromised host bone stock where the rate of re-revision was very high . In early studies, the reactive sclerotic bone between the fibrous membrane and the native cancellous bone was not removed . Poor fixation of the revision femoral component compared to that in the primary setting may be due to inadequate excision of residual fibrous membrane, incomplete drying of bone, suboptimal cement filling technique, or insufficient cement-bone interlock on the smooth sclerotic bone surface. In such setting, even long cemented stems, are generally difficult to be inserted with adequate primary and long lasting stability. Femoral revision using so-called modern cementing techniques may yield promising results (Figure 2).
Different studies have demonstrated that modern cementing techniques have improved implant survival and clinical outcome compared with the mid-term results after revision with use of so-called first-generation technique [71,85,109,128,134] (Figure 3).
Cemented femoral revision has several advantages in elderly patients. It allows early mobilization, a shorter operating time, and possibly less risk of a peroperative fracture. Use of modern cementing techniques seems to improve fixation of the femoral components and clinical outcomes and justifies its use. Whenever possible the failed arthroplasty should be revised before occurrence of severe bone loss and femoral enlargement.
188.8.131.52. Cement-within-cement fixation of the femoral component
Removal of the well-fixed cement mantle around the stem can be extremely difficult, time consuming, and risky procedure. A solution to the problem, cement-within-cement revision was first proposed by Eftekhar who advised on preserving the existing well cemented mantle and re-cementing the new stem into it . In a biomechanical study, Greenwald et al. demonstrated that the separation strength was 94% that of a single block when the existing cement mantle was adequately prepared . The technique requires that the old cement surface be dried and roughened in order to provide contact area for fixation of the new cement. The cement should be injected in the early liquid phase in order to prevent lamination and to promote polymerization within the existing cement mantle.
This technique has been questioned by other authors , but subsequent biomechanical and clinical studies have favored its use in properly selected cases [35,67,97]. In a cadaver study, Rosenstein et al. demonstrated that cut strength at the cement-cement interface was greater than the strength at the cement bone interface . However, the cut strength of the cement bone interface was 30% weaker when cement was placed against a revised bone surface.
6.1.2. Impaction grafting technique for femoral revision
Femoral impaction grafting with a cemented stem was first performed in Exeter in 1987  (Figure 4). The rationale behind this simple concept is to rebuild femoral bone stock and to provide secure fixation to the femoral stem. The biologic approach of bone restoration during revision hip arthroplasty is a highly appealing solution for restoring host bone stock a difficult procedure with usually deficient femurs.
Following the initial report of Gie and coworkers of highly successful results after 56 revisions with follow-up of 18-49 months  the technique received wide attention and spread rapidly [64,114,141,161]. Further studies confirmed the favorable outcomes, and it became evident that the technique resulted in restoration of femoral bone loss as the impacted allograft was incorporated and remodeled [67,161].
The technique of impaction grafting appeared to be reliable, reproducible, can be learned rapidly, and produced predictably favorable outcomes. In a series of 226 revisions, Halliday et al. reported the Exeter initial experience with femoral impaction grafting . The overall rate of mechanical failure was 7% (16/221) at a minimum follow-up of five years. Ten to 11-year survivorship with removal of the stem for any reason as the end point was 90.5% and survivorship with revision for aseptic loosening as the end point was 99.1% . In 2006, Schreurs et al. published their results with the technique using a cemented polished tapered stem at an average 10.4-year follow-up . The average subsidence of the stem within the cement mantle was 3 mm, and seven stems migrated 5 mm. . No stem was revised for aseptic loosening. Three periprosthetic fractures at the sults of 1305 femoral revisions with impaction grafting from the Swedish arthroplasty registry . Survivorship at 15 years for aseptic loosening was 99.1%, for infection 98.6%, for subsidence 99.0%, and for fracture 98.7%.
However, other authors reported higher percentage of intraoperative complications, mainly femoral fractures and suboptimal cementing technique [84,86,101,123]. A high incidence of up to 12% intraoperative femoral fractures have been reported [64,140,103]. Stem subsidence of greater than 5 mm is a typical complication with this technique with a prevalence of up to 38% in some series [38,49,86,101]. Impaction grafting has certain disadvantages: it is prone to femoral fractures ; has a steep learning curve; and shows highly variable outcomes, probably related to the surgical technique. The causes of early subsidence of the stem might be insufficient impaction of the allograft, suboptimal cement penetration and interdigitation, use of synthetic graft substitutes, or other graft additives, loss of primary fixation of the allograft-cement composite due to soft-tissue infiltration and substitution of the allograft in the process of remodeling and revascularization, unrecognized femoral fracture, or fracture of the cement-allograft composite. However, in a study on saw femurs, Flannery et al. and Cummins et al. were unable to find a correlation between threshold force needed to achieve stable construct in impaction bone grafting without fracture and bone mineral density, canal-cortex ratio, or cortical thickness [22,48]. According to Gokhale et al. four variables (age, intramedullary canal diameter, stem design, and density of the graft at the tip of the stem) affected the subsidence of the stem .
The original technique of impaction grafting utilized the Exeter stem . The impacted graft is subjected to continuous loading and deformation. Thus, the use of double-tapered polished stem appears suitable option as the stem could achieve secondary stability after subsidence. Arguing that the technique is more important than the type of prosthesis other authors have used different implant designs from those of Exeter wedge shaped prosthesis [49,77, 124]. Uncemented technique was also used with an equally good outcome at mid-term follow-up .
Femoral impaction bone grafting is a suitable indication for cases with severe bone deficiency. The technique is expensive, prone to complications, hast steep learning curve, and results depend on surgical skills. It may be a viable revision option for young patients with severe bone loss.
6.1.3. Proximal femoral allografts in revision surgery
Severe bone loss in femoral revision is increasing problem as the number of patients with multiple previous revision increases. These complex cases are further increasing as the age of patients undergoing hip replacement is diminishing.
A stable initial fixation is hardly obtainable in complex cases with circumferential proximal bone loss >5 cm in length. Severe bone loss makes femoral revision using conventional techniques difficult. Alternatives include distal fixation of the stem or use of a proximal femoral allograft. Distal fixation requires the use of a proximally femoral replacement prosthesis or megaprosthesis. This has some disadvantages such as: instability due to poor soft-tissue attachment [62,120], early loosening of the distally fixed stem , stress shielding [10,46,110], intraoperative fractures  or difficulty with fixation in an ectatic femur. Various studies of revisions using megaprosthesis reported survival rate within the range 58% to 84% at five to ten year follow-up [98,120,164].
The viable revision technique using proximal femoral allograft consists of a long-stem prosthesis cemented to the allograft but not to the host bone  (Figure 5). Uncemented fixation of the allograft prosthesis construct would not result in long lasting stability of the prosthesis as neither in-growth nor on-growth could be expected at the allograft-implant interface. The importance of the allograft-host bone contact is a key factor for achieving stability of the construct and ensuring long-term stability of the implant .
Individual studies published encouraging results after use of proximal femoral allograft-prosthesis construct in large segmental defects of the proximal femur. In a series of 44 revisions with a mean follow-up of 7.2 years Vastel et al. observed two deep infections, two aseptic loosening and two fractures bellow the tip of the prosthesis . The final prosthesis survivorship rate with revision as the end point was 82.4% at 14 years of follow-up. The nonunion of the greater trochanter was considered major complication and was observed in 25 cases. In another series of 30 hips who underwent revision total hip replacement with an allograft prosthetic composite Sternheim et al. observed favorable long-term outcome . The survivorship at 10, 15 and 20 years was 93%, 75.5% and 75.5%, respectively. Encouraging results were published by Blackley et al. with 78% successful results for an average of eleven-year follow-up . The allograft-prosthesis construct survivorship at five years was 90% and at 10 years was 86%. A recent systematic review of 498 hips with a mean follow-up of 8.1 years reported survival rate of 82% . The major complications were aseptic loosening observed in 13.7% of patients followed by dislocation in 12.8%.
The use of a proximal femoral allograft-prosthesis construct has some inherited disadvantages characteristic for complex surgery. Allograft resorption eventually leading to failure of the revision is of major concern with longer follow-up . Usually, it was observed after several years of follow-up but did not progress [7,155]. Authors that utilize uncemented distal fixation support the concept of direct loading of the host-allograft junction and argue that it minimizes allograft resorption [7,59]. On the other hand, cementing the prosthesis to the distal femur and thus stress shielding the allograft may explain the high rate of allograft resorption . Nonunion of the allograft-host junction  or the greater trochanter [61,155] are of major concern with this technique. A step-cut osteotomy may provide rotational stability while an oblique osteotomy may provide greater surface area for bone healing compared to a transverse osteotomy. Dislocation is a frequent complication after revision with proximal femoral allograft with incidence ranging from to 7.3% to 16.7% [15, 61,131,155]. As with other cases, the high risk of dislocation may be lowered by optimal reconstruction of length, adequate version and high offset of the prosthesis-allograft construct and by maintaining the soft tissue tension and its attachment to the host femur . The infection rates after revision with allograft-prosthesis construct are higher than that reported after primary hip arthroplasty. Rates of infection ranging from 0 to 10.9% were reported [15,94,132,136,155]. However, considering the high complexity of the technique these levels of infection are not unacceptable.
Femoral revision using proximal femoral allograft cemented to a long-stem prosthesis is appealing option for revision. The current data from the literature support the use of the technique as a durable solution, with available evidence reporting a long-term survivorship up to 86%. It is of a particular interest in the young patients because of its potential to improve bone stock and provide a substrate for subsequent revision. The development and refinement of this technique should be encouraged.
6.1.4. Cementless fixation of the femoral component
Obtaining stable and long lasting fixation in femoral revision in patients with severe bone deficiency is a difficult task. Long-term results after cemented revision have not been optimal [81,83,121]. High failure rates ranging between 12% to 44% at mid-term follow-up have been reported [2,81,83,121]. The main reason was difficulty in obtaining stable and long-lasting fixation in severe bone loss. Cementless fixation proved a promising alternative and was soon introduced in practice. However, for fears of stress shielding the ingrowth surface of first-generation designs was confined to the proximal part of the stem. Although highly successful in primary arthroplasty, the limited amount of porous coating with proximal fixation led to less favorable results in revision surgery (Table 4). Failure rates of 4% to 10% were reported at short- to mid-term follow-up. These results were slightly better than those obtained with an early cementing technique. The technique yielded acceptable results in less severe deformities . Retrieval studies have demonstrated that less bone ingrowth occurs in revision stems compared to primary stems . Porous surface extending to the diaphysis is needed to ensure stable primary fixation. In support of this, various authors have reported promising long-term results after revision with use of extensively coated uncemented stems [1,41,88,115].
Proximal femoral deficiency results from osteolysis, infection, fracture or bone damage during implant extraction. In such cases with severe bone loss, distal fixation with cylindrical, tapered or fluted stem designs is a viable option [4,14,41]. The technique requires accurate preparation of 4 cm to 7 cm of diaphyseal bone [41,92,102]. It is adaptable and can be used in situations with different severity of bone loss. Moreover, it can be used in periprosthetic fractures and is adjustable with extended trochanteric osteotomy. Distal stem fixation is the most successful strategy in terms of primary and secondary mechanical stability, bone osteointegration, and most importantly clinical results . The main reason for its success is the fact that the implant is in contact with viable bone. Success rates of 90% to 95% have been reported with extensively coated monoblock stems over 10-year follow-up [105,115,158]. However, issues, such as thigh pain and proximal stress shielding, were reported frequently.
The principles of the Wagner stem are utilized in the tapered stems . The cone prosthesis achieves good contact between the supportive distal diaphysis and the middle or distal third of the stem (Figure 6). The conical shape and the longitudinal splines promotes primary axial and rotational stability, which are prerequisites for osteointegration and long-lasting endurance of the cementless implant. The concept of modularity was introduced in the tapered stems with the advantages of versatile proximal fill and distal fit . By modularity the sizes and shapes of the prosthesis can be increased by varying the diameter and shape of the proximal and distal part of the stem and locking them in different way. Modularity offers certain advantages such as correction and restoration of leg length, correction of offset and version, selection of optimal proximal fill, as well as compatibility with extended trochanteric osteotomy [4,92]. The potential problem of thigh pain was not associated with tapered stem design . Disadvantages of modular taper stem designs include complexity, risk of stem fracture, fretting and corrosion of the junction, increased inventory, and higher cost.
|Lawrence et al. ||1993||174||7.4||5.7||1.1|
|Engh et al. ||1988||127||4.4||1.6||2.4|
|Kang et al. ||2008||39||2 (minimum)||2.6||0|
|Krishnamurthy el al. ||1997||297||8.3||1.7||0.7|
|Harris et al. ||1988||23||2 (minimum)||0||4.3|
|Moreland and Bernstein ||1995||175||5.0||1.1||4.2*|
|Moreland and Moreno ||2001||137||9.3||4||-|
Various studies have reported favorable outcomes of various uncemented tapered distal fixation stems with high survival of more than 95% at 5 to 10-year follow-up [41,88,102]. The technique of cementless revision with distal fixation of the stem has been shown to be a reliable and straightforward. It ads no additional risks or complications. It can be used in all but the most severe segmental defects [4,14,41,102]. When the simple principles of the method are followed it provides stable and durable fixation of the revision implant. However, distal fixation does not restores host bone, thus making further revision surgery more difficult.
6.2. Options for acetabular revision
The goals of revision arthroplasty are to relieve pain and to improve function. In order to obtain these goals stable and durable fixation of the revision components must be achieved with restoration of hip center. Acetabular revision is the most difficult part of hip revision. Unfortunately, there is no single surgical technique to solve the problem of fixation. The achievement of stable initial and long-lasting fixation is challenged by the severity of different acetabular defects and soft tissue damage. The main acetabular reconstruction option is cementless pres-fit fixation of the cup with or without allograft [19,73]. When severe combined segmental and cavitary bone deficiencies, poor bone quality and viability or pelvic discontinuity are identified, other more complex options for acetabular reconstruction are required. These include trabecular metal (TM) cups, modular metallic augments, reconstruction cages, reinforcement rings, cup-cages, and structural or morsellized allografts that can be used to support the reconstruction.
6.2.1. Cemented fixation of the acetabular component
In cases with no or moderate bone loss revision could be performed with simple cemented exchange of the implant. Historically, early revisions were performed with the same technique that had been used for primary arthroplasty. However, difficulties in achieving consistent long-term results had prevented use of this technique. Failure to achieve adequate cement interdigitation explained poor results reported with early cementing techniques. Key factors for good cementing technique are optimal exposure of cancellous bone, adequate containment of the cup, and a clean and dry socket . Sutherland and colleagues demonstrated that preservation of the subchondral bone can increase stiffness and stress concentration at the bone-cement junction . Callaghan et al  reported 4.3% revisions and 34% radiographic loosening at 3.6-year follow-up of cemented revisions. Similar high rates of loosening were reported by Pellicci et al at 3.4-year follow-up . The long-term results after cemented revision were considerably worse . Even with improvement of the cementing technology, the cemented acetabular fixation has not improved (Table 5).
At longer follow-up, using modern cementing techniques failure rates ranging from 35% to 65% were reported [78,112]. Ten-year survivorship of the acetabular component with radiographical loosening as the endpoint event was 72% . Consequently, cement fixation of the acetabular component has become less popular among orthopedic surgeons. In contrast, hemispherous porous-coated cups with bone ingrowth potential were developed and demonstrated consistently better results.
|Raut et al. ||1995||387||5.5||6.2||18.8|
|Estok and Harris ||1994||32||11.7||22.0||19.0|
|Mulroy and Harris ||1996||29||15.1||38.0||44.0|
|Katz et al. ||1997||79||11.9||16.0||23.5|
|Eisler et al. *||2000||83||3.6||8.0||22.0|
|Huo and Salvati ||1993||113||4.1||1.0||5.0|
6.2.2. Cementless fixation of the acetabular component
The approach to each individual case depends upon the severity and localization of host bone loss. Results after cementless revision of the acetabular component have outperformed cemented fixation [30,31,44]. With supportive and viable host bone and a reliable ingrowth surface a hemispheric metal shell supported with screws is a straightforward solution for acetabular reconstruction (Table 6). The success of the technique has been so dramatic that it is currently considered the gold standard by most arthroplasty surgeons in the USA.
|Della Valle CJ et al. ||138||15.0 (minimum)||4.5*||1.5|
|Park et al. ||138||20.0 (minimum)||8.0*||5.0|
|Wysocki et al. ||187||5.0 (minimum)||2.0||5.0|
|Lachiewicz and Hussamy ||60||5.0||0||0|
|Tanzer et al. ||140||3.4||0.7||4.3|
|Silverton et al. ||109||8.3||0||5.0|
|Templeton et al. ||61||12.9||0||3.0|
In minor or contained defects, hemispherical cup with/without grafting produced excellent results [30,31,44]. Cementless fixation is suitable for patients with Paprosky types 1, 2A and 2B defects: without hip center migration or pelvic discontinuity. As a general rule, at least 50% of the host bone is needed to be in contact with the implant in order to support a hemispherical cup. Transfixational screws are usually used to support ingrowth of the press-fit cup. Morcellized allografts could be used to fill the cavitary defects. In a deficient acetabulum with major bone loss such as Paprosky type 3 defects a hemispherical cup could be placed against the intact supportive roof ("high hip center"). Sometimes extra large (jumbo cup) or oblong cups can bypass severe bone defects and provide stable initial fixation for bone ingrowth. Cementless fixation results in a anatomical hip center or in a high hip center .
184.108.40.206. High hip center
The failed acetabular components migrate in the direction of joint reaction forces creating a deficient acetabular bed with greater superoinferior dimension compared to the anteroposterior dimension. In such revision setting implantation of the cementless hemispheric press-fit cup in the anatomical hip center is not possible. A straightforward decision for treatment of such defects is to place a small hemispherical press-fit cup against the supportive bone at the roof of the acetabular defect - the so-called high hip center (Figure 7). Most authors consider arbitrary the hip center high if it is proximally greater than 35 mm to the inter-teardrop line .
Results after cementless press-fit fixation of the acetabular component inserted with screws outperformed cemented revision. With use of this approach, despite extensive acetabular bone loss excellent implant fixation was consistently reported. The durability of cementless acetabular fixation was proven in long-term studies, too. In the study of Templeton et al., none of the cementless cups have been revised for aseptic loosening at 12.9-year follow-up and only 3 cups have migrated . In a study with minimum follow-up of 20 years, Park et al. demonstrated 95% survivorship with revision of the cup for aseptic loosening or radiographic signs for loosening as the end point . However, with longer follow-up, the problem of polyethylene wear and osteolysis emerged. In their series, re-revisions for wear and osteolysis were first performed at approximately twelve years postoperatively . At last follow-up 20 years after revision, the incidence of reoperations for polyethylene wear and/or osteolysis continued to increase.
The technique saves costs, time, and eliminates the use of structural allografts or cement. However, high rate of complications was reported [27,74]. This might reflect the complexity of the procedure. Certain disadvantage of the technique is restoration of limb-length discrepancy on the femoral side whereas the defect is on the acetabular side. This would result in abnormal hip biomechanics. Increased hip joint reactive forces with high hip center and impingement might partially explain the relatively higher rate of dislocation with this technique [117,162].
Considering the excellent results reported with porous coated press-fit acetabular components in terms of implant fixation, we believe that the use of press-fit cups should be considered in every revision setting if there is sufficient host bone stock to support the cup.
220.127.116.11. Jumbo cups
Extra-large cups offer certain advantages in maximizing the contact area between the cup and host bone when revising deficient acetabulum. There is no universally-accepted definition of the jumbo cup. Extra-large cups are arbitrary defined compared to the size of the pelvis, the hip joint, and the previous implant. Whaley and coworkers defined jumbo cups as having a minimum outside diameter of 66 mm (men) or 62 mm (women) . This was based upon the fact that the revision cups used at their institution were 10 mm larger than the mean implant diameters used for primary hip arthroplasty.
The method has certain advantages : the acetabulum is prepared straightforward by reaming to a large hemispere; the large implant fills in the deficiencies and bone grafting is usually unnecessary; the center of rotation is transferred inferiorly and to some extent laterally restoring hip biomechanics (Figure 8); the large implant provides greater contact area and greater lever arm. Disadvantage of the technique are the limitations in restoring bone stock. Moreover, most of the defects are oblong with a greater superoinferior dimension than anteroposterior dimension. Converting an oblong defect to a hemispherical with extensive reaming may disrupt posterior wall or column which is critical for cup stability . This risk of host bone compromise may result in high implantation of the socket.
Whaley et al. reported on 89 acetabular revisions using extra-large hemispherical components from the Mayo clinic . The probability of survivorship of the cup at eight years was 93% with removal for any reason as the end point, and 95% with radiographical loosening or revision for aseptic loosening as the end point . Wedemeyer et al., Obenaus et al., and Dearborn and Harris published similar results with cup survivorship with an end point aseptic loosening higher than 94% at mid-term follow-up [26,113,157].
The technique saves costs, time, and eliminates the use of structural allografts or cement. However, the rate of complications reported was rather high. In the series of Park et al., the most common reason for revision in 11.6% of 138 hips was infection and dislocation . Similar high rates of revision were reported by Dearborn and Harris  and Della Valle et al. . This might reflect the complexity of the procedures.
Extra large cups are a reasonable alternative in patients with moderate defects (Paprosky type 2).
18.104.22.168. Oblong cups
As described earlier, large oval contained defects cannot be filled-in superoinferiorly without excessive reaming of the anterior or posterior column of the acetabulum to produce hemisphere. Another option is high placement of the cup. Attractive alternative in such cases is oblong cup. Oblong cup has smaller anteroposterior and mediolateral dimensions compared to the superiorinferior dimension. By accommodating the implant to the defect oblong cups can restore hip center and increase implant contact with host bone. Advantages of the technique are: lack of increased reaming of the anterior or posterior columns or metallization of the cup; increased contact between the device and the host bone; restoration of hip center [19,73]; and avoidance of structural allografts . Disadvantages include higher cost, difficulties in cases with insufficient contact , possible component malpositioning, failure to restore bone , and excessive bone removal in order to achieve press-fit .
In a multicenter study on 38 hips revised with oblong press-fit cups, Berry et al. published good results at mean 3 years after surgery . There was only one failure for acetabular loosening that required re-revision. Mean Harris Hip Score (HHS) increased from 50 points preoperatively to 90 points after revision. The hip center was 37 mm above the inter-teardrop line before the operation and was corrected to 25 mm above the inter-teardrop line. Civinini et al. published good mid-term results after revising with oblong cup Paprosky type 2 and 3 acetabular defects . In a series of 55 hips followed-up for an average of 7.2 years only one cup was revised for loosening. Similar good results were reported by DeBoer and Christie on 18 hips revised with oblong press-fit cups . At the latest follow-up, 4.5 years after revision no component was loose and the mean HHS increased from 41 points preoperatively to 91 points after revision. The authors reported near anatomic restoration of the hip center to 17 mm above the inter-teardrop line after revision. Chen and Engh reported less favorable results in 37 revisions (29 with massive type 3 defects) followed-up for an average of 41 months . Eight percent of the hips were probably loose and 16% were unstable. Eight of the 14 hips that had more than two centimeters of superior migration of the component and disruption of Koehler’s line on preoperative radiographs failed . The authors found a correlation between loosening and average distance from the inferior edge of the cup and the interteardrop line. Five of the six unstable components initially had not reached to the level of the radiographic teardrop or distal to it . In their early series, Sutherland  reported a 50% failure rate (3 of 6) after revision with an oblong cup of type 3 defects. Discouraging mid-term results were published by Babis et al. using a cementless oblong cup for revision of Paprosky 3A defects . After a mean follow-up of 60.5 months 18 hips (29.0%) were revised and a further four hips (6.4%) were loose and awaited revision. Further analyses proved that careful patient selection is critical for the success of this revision technique.
Acetabular revision with an oblong cup is an attractive alternative, especially when the surgeon plans correction of an elevated hip center [16,108]. The technique is suitable for Paprosky type 2A, 2B and 3A defects where the acetabular defect is oval. Obtaining initial stability and supplementary fixation by screws are mandatory for the stability of the implant and are key factors for long-term success of the reconstruction. However, the medial wall should be intact and the failed component should not have migrated more than two centimeters. Pelvic discontinuity is also a contraindication for this technique. Alternative techniques such as structural allograft or cage should be considered in such revision settings.
6.2.3. Impaction grafting technique for acetabullar revision
Cemented acetabular revision yielded unacceptably high rate of loosening [13,80,81]. Possible reason is the deficient, weaken and sclerotic acetabular bone frequently found at revision. An attractive alternative for cup fixation in massive contained defects is impaction grafting where the cup is cemented on a premoulded bed of impacted morsellized cancellous bone. In such revision setting the contact with host bone is very limited if not absent. The morsellized bone has osteoinductive and osteocunductive properties and is used as a filler scaffold of contained defects. The technique of impaction bone grafting and cemented fixation of the cup was first described in 1984 by Slooff and coworkers , and later standardized by the same authors with minor modifications and technique-specific instrumentation [11,142]. Morsellized graft could be used with a cementless hemispherous cup if more than 50% of the cup is in contact with viable host bone . Transfixing screws should be used for additional stabilization of the cup. In cases with less than 50% contact between the cup and viable host bone the cemented cup into impacted cancellous bone should be used .
The technique can provide stable and durable reconstruction of the hip joint. However, initial mechanical stability of the morsellized allograft-cemented cup composite is a prerequisite for a successful biological reconstruction. Subsequent remodeling and incorporation of grafts, provides long-term stability. In contrast to cementless revision, this technique could restore bone loss. The modern evolved technique consists of reconstruction of segmental and rim defects with use of a metal mesh or a solid graft. The sclerotic areas are perforated with multiple 2-mm drill-holes, and fresh-frozen morsellized bone chips are impacted layer by layer into the acetabulum. The clinical success of the of impaction grafting depends on the surgical technique and on the biological and mechanical properties of the morsellized bone graft. Various factors connected with the graft such as graft type (cancellous, cortical-cancellous, chemical composites, synthetic additives), graft processing (fresh frozen, freeze-dried, irradiated), graft particle size and grade influence the clinical result. The originators of the technique use fresh frozen allografts [11,142,143].
The technique of acetabular impaction grafting is well-established and various authors demonstrated reproducible results [20,51,143,154]. Schreurs et al. reported good results after acetabular revision with impaction allografting at 15 to 20-year follow-up . With revision for aseptic loosening as the endpoint, cup survivorship was 84% at 15 years . In a 20 to 28-year follow-up study, Busch et al. from the same study group evaluated 42 patients with impaction grafting younger than 50 years . With revision for aseptic loosening as the end point, survivorship was 85% after twenty years and 77% after twenty-five years . With end point event signs of loosening on radiographs, survival was 71% at twenty years and 62% at twenty-five years. Results declined over time, but the authors concluded that the technique is useful in younger patients with major bone defects .
Acetabular revision using impaction grafting is a reliable alternative for biologic restoration of hip joint mechanics. The procedure is technically demanding and exacting. Results comparable to those after revision with cementless hemispheric cups were obtained after use of correct surgical technique.
6.2.4. Reconstruction cages
Reconstruction cages and antiprotrusio rings are an established method of treatment for severe acetabular bone loss if contact with 50% host bone could not be established . They have the advantage of fixation into viable host bone of the ileum and ischium with flanges while protecting allograft (Figure 9). Failure rates higher than 60% at an average of 2.9 years have been reported in cases with massive allografts not supported by cages . Because of the poor results following use of unsupported structural allograft use of reconstruction cages has been advised . Reinforcement rings or reconstruction cages can provide adequate support for the reconstruction with massive allografts in Paprosky type 2C and type 3 defects. Favorable results were reported after use of reconstruction cages by different authors [50,68,107,125]. Recently, this treatment approach has been questioned as TM implants provide more favorable conditions for graft incorporation and bone ingrowth .
The reconstruction cages and antiprotrusio rings have definite advantages: the cage and ring allow for restoration of hip center; they provide uniform load to the allograft stimulating bone remodeling and incorporation into host bone ; cementing allows use of local antibiotic protection; allow correct placement of the cemented cup independent of the cage or ring. The cage protects either the morsellized or structural allograft while it remodels, and if the cage fails cementless revision can be done [55,57].
Disadvantages include higher cost, need for wide surgical exposure of the superolateral part of the ileum. The later may risk injury of the superior gluteal nerve and limping. The major concern with standard nonporous cages and rings is that they do not allow bone ingrowth. Finally, they loosen and break. However, this inability of bone ingrowth is compensated by the mechanical stability and incorporation of the graft reducing the risk of fatigue fracture of the cage. Close fit between the cage and the allograft as well as adequate fixation of the cage are a prerequisite for successful reconstruction. Cement augmentation of screws is recommended in cases with severe osteolysis or osteoporosis.
The limits of using antiprotrusio rings were demonstrated by Haentjens and coworkers and Zehntner and Ganz [63,163]. High rate of migration up to 44% (12/27) at mid-term follow-up of 7.2 years was reported . Previous designs of reconstruction cages did not allow bone ingrowth and a failure rate of 16.4% due to loosening was reported on average 4.6 years after surgery . Sporer et al. reported a 2- to 8-year follow-up of 45 hips where a cage was used for a type 3 defects . Nine hips were revised for aseptic loosening, and an additional nine hips were radiographically loose.
In contrast, Winter et reported no loosening or revision in 38 hips followed-up at mean 7.3 years after revision with cage . In a long-term study, 18 acetabular revisions for pelvic discontinuity have been reported on average 13.5 years after surgery . Two cages were revised for aseptic loosening, and another two allografts showed signs of severe osteolysis. Survivorship of the acetabular component at 16.6 years with end points revision for any reason, loosening or nonunion of the allograft was 72%. The increased rate of loosening and revision is probably multifactorial and reflexes the increased case load. Frequent indication for use of reconstruction cages is pelvic discontinuity. However, designs without bone ingrowth do not have potential for biologic fixation and rely solely on mechanical fixation.
Antiprotrusio cages and rings are an effective technique for treatment of severe bone defects. However, in recent years, newer implant designs have gained popularity. In cases with more than 50% host bone support cementless cup transfixed with screws is the treatment method of choice. Trabecular metal implants, porous augments, and triflanged acetabular components are an attractive alternative for complex acetabular reconstructions, [33,57,148]. TM cups have been proposed if contact with viable host bone is 30% to 50%. If contact with viable host bone is less than 30% a cup-cage construct was suggested . Longer follow-up studies are needed to support the clinical use of these new implants.
6.2.5. Structural acetabular allografts
One of the most difficult scenarios in revision surgery is a reconstruction of a massive acetabular bone loss. Structural acetabullar allografts are a suitable revision option for uncontained bone defects (Paprosky type 2B, 3A and 3B). The size of the allograft may range from a femoral head in superolateral uncontained defects to total acetabular allograft when severe uncontained defects or pelvic discontinuity are present.
Advantages of the technique include restoration of hip center and restoration of bone stock for future revisions [50,68,90,125]. However, actual restoration of viable and mechanically competent bone is questionable. Moreover, results are unpredictable. The technique is demanding and associated with various complications.
Results after revision with structural allografts have been largely controversial. Harris initially reported successful results after reconstruction of severe acetabular defects with structural allografts . However, the encouraging period of initial good functioning for 5 to 10 years was followed by later failures. In 1993, Kwong et al. reported 47% failures in 30 hips with a mean follow-up period of 10 years . In 1997, the senior author reported total rate of revision or loosening 60% at an average of 16.5 years . High hip center for placement of the cup was suggested in cases with severe acetabular bone loss .
In a series of 33 hips followed-up on average 7 years after revision with a structural allograft, Garbuz et al. reported 55% success rate . Fifteen hips were revised: seven hips because of failure of the prosthesis and eight hips because of failure of both the allograft and the prosthesis. Gross et al. reported on 107 hips reconstructed with bulk allograft . Thirty hips (28%) were revised and in 15.9% of cases (17 hips) the indication was aseptic loosening. The authors reported 76% successful results in the 33 hips with minimum duration of follow-up of 5 years (average, 7.1 years) after the revision. However, eight hips needed additional reoperation because of failure of the graft and another six hips were revised for loosening. Hooten et al. reported on a series of 31 revisions with structural allograft and cementless cup followed-up on average 46 months after surgery . Twelve (44%) cups were radiographically loose and five of these hips were revised. In contrast, Paprosky et al. reported a failure rate of 19 per cent (6/31) at an average follow-up of 5.7 years after revision with use of a structural allograft . The only failures in that series were in hips in which the allograft supported more than 50 per cent of the cup. In another study, Morsi et al. found a success rate of 86% (25/29) at mean follow-up of 7.1 years . They used a minor bulk allograft that supported less than 50% of the cup.
Although results after revision with structural allograft are controversial, most authors agree that the rate of success increases if more than 50% of the cup is in contact with viable host bone [50,68,116]. According to Morsi et al.  and Pollock and Whiteside  a repeat revision does not mean failure of the reconstruction. This complex reconstruction can be considered successful if bone stock is restored for future revisions.
Revision with structural allograft is a suitable option for restoration of hip center. The role of the allograft is to support the cementless cup with partial stability until adequate ingrowth occurs. The success after the procedure is technically-related. In order to optimize result after revision with structural allograft a number of principles should be followed. Structural allografts combined with antiprotrusio cages, and a cemented cup should be considered only in cases with insufficient host bone to provide a stable fixation for a press-fit cup .
For an optimal result, an appropriate allograft must be selected to match the mechanical requirements of the desired reconstruction. The method of processing of the bone allografts is important for the clinical result. Greater success rate with fresh frozen bone allografts was obtained compared to freeze-dried allografts . The trabecullae of the allograft should be in the direction of load for optimal stress transfer. After trimming of the allograft in order to obtain maximal contact with the host bone the allograft is fixed with 6.5 mm parallel screws in the direction of load. In case of pelvic discontinuity, the column should be fixed with a plate before fixating the allograft. Use of reinforcement cages improves results after reconstruction with structural allograft [50,138].
6.2.6. Custom triflanged implants
Custom triflanged prostheses have been proposed for treatment of massive acetabular defects and pelvic discontinuity, but the experience is limited and the rate of complications is high . The implant is manufactured from 3-D CT data reconstruction of the degree and localization of bone loss as well as its spatial orientation.
In 2007, DeBoer et al. evaluated the outcome of revision with a custom-designed porous-coated triflanged acetabular implant in 20 hips at an average 10-year follow-up . A definite healing of the pelvic discontinuity was found in 18 hips (90%). The remaining two implants were radiographically stable and did not migrate even when discontinuity persisted. However, the overall dislocation rate in the series was 25%. Christie et al. followed-up retrospectively 67 complex revisions with custom-made triflanged implant . Two discontinuities persisted, but both were asymptomatic and no implant was revised. Six (7.8%) hips were revised for recurrent dislocation. Using custom triflanged acetabular components Dennis reported three failures in 24 revisions with mid-term follow-up of 4 years . He questions the value of the technique in pelvic discontinuity unless supplemented with additional column plating.
The technique has high cost, it is time consuming, requires extensile exposure, and lacks modularity. It could not be used in urgent clinical situation where it is not possible to wait for manufacturing the product. With complex implant and technically challenging surgery custom-designed triflanged prosthesis should be reserved for cases where less costly and less technically demanding options could not be used. Many surgeons consider it a salvage procedure for cases where the bone loss is catastrophic.
6.2.7. Trabecular metal cups
Trabecular metal cups can be used in massive contained or uncontained defects. As tantalum provides favorable environment for biological fixation, TM cups have been suggested for revision of Paprosky type 3 defects [57,148] instead of an allograft-cage construct.
Early results with use of TM implants have been encouraging . TM has decreased the need for at least 50% contact of the implant with viable host bone. In Paprosky 3A an 3B defects, because cages do not provide biologic fixation, Gross suggested use of a cup and cage construct (the so-called cup-cage technique) when less than 30% contact can be made with viable host bone . The rationale behind the technique is that load will be taken off the cage, once bone ingrowth occurs into the trabecular metal cup. So early and mid-term failures of the cages will be prevented.
Sporer et al. reported on 13 hips with pelvic discontinuity revised with tantalum cups with or without augments . At mean 2.6 years after revision 12 of the 13 cups were radiographically stable. Lakstein et al. reported on 53 revisions of contained defects with 50% or less contact with host bone using TM cups . Two cups (4%) were revised, and two additional cups (4%) had radiographical evidence of probable loosening at a mean 45-month follow-up. The fact that some of the TM cups lacked contact with a viable host bone is impressive. Four hips (8%) dislocated and one (2%) sciatic nerve palsy was observed. In a large multicenter study, 263 revisions with tantalum TM cups were followed-up at an average 7 years after surgery. At the most recent follow-up, all cups were radiographically stable and no revision for loosening was reported. Eight dislocations (3%) in the series were successfully treated with closed reduction, and one sciatic palsy partially resolved at last follow-up. Kosashvili et al. reported on 26 revisions of pelvic discontinuities using cages combined with trabecular metal components and morsellized bone (cup-cage technique) . At mean follow-up of 45 months 23 hips (88.5%) were radiographically stable.
Promising midterm results have been demonstrated after revision with use of these new techniques. Currently, the preference is to biological fixation whenever possible, and to alternative options when initial stability could not be obtained.
6.2.8. Modular components (metal augments)
Modular metal augments of various sizes and shape are used to decrease defect size and to restore bone defect to contained one capable of supporting a revision cup (Figure 10).
The size and placement of augments is highly dependable on the bone loss pattern. Augments are secured with multiple screws to host bone and remaining defects are filled in with morsellized bone. The hemispherical cup is impacted into the defect with the interface between the shell and the metal augment cemented.
During the last two decades, revision hip arthroplasty is constantly in the focus of orthopedic surgeons as the numbers of these difficult and risky surgeries are increasing. Up to date, he paradigms of revision surgery has been evolving constantly. From polyethylene wear, osteolysis and loosening, to complexities such as pelvic discontinuity, there is a wide range of surgical options for successful reconstruction. Analysis of clinical results from various studies outlines the preference for biological fixation of the revision implant whenever possible. It is vital for the surgeon to be familiar with different treatment approaches and to anticipate various intraoperative scenarios. Systematic approach with considerable preoperative evaluation and planning will achieve a good result. Prerequisite for a successful and durable revision include viable host bone, adequate surgical technique, and stable and endurable implant. Current improvements in surgical techniques, implant designs, as well as biomaterials and bearing surfaces are a significant contribution for obtaining favorable outcome after revision hip arthroplasty. However, we do not have complex solution. The optimal surgical approach for revision THA varies considerably among different settings. On the other hand, the economic burden of total joint replacement is increasing at a steep rate. This necessitates improved methods for evaluation of existing technology and particularly patient-derived outcomes assessment instruments. Further research and well-designed clinical studies are needed in order to provide optimal treatment to the increasing number of patients requiring revision surgery in the future.
Amanatullah DF, Meehan JP, Cullen AB, Kim SH, Jamali AA. Intermediate-term radiographic and patient outcomes in revision hip arthroplasty with a modular calcar design and porous plasma coating. J Arthroplasty 2011; 26(8):1451-4.
Amstutz HC, Ma SM, Jinnah RH, Mai L. Revision of aseptic loose total hip arthroplasties. Clin Orthop Relat Res 1982; 170:21-33.
Babis GC, Sakellariou VI, Chatziantoniou AN, Soucacos PN, Megas P. High complication rate in reconstruction of Paprosky type IIIa acetabular defects using an oblong implant with modular side plates and a hook. J Bone Joint Surg Br 2011; 93(12):1592-6.
Barrack RL. Orthopaedic crossfire - Stem modularity is unnecessary in revision total hip arthroplasty: in the affirmative. J Arthroplasty 2003; 18(3 Suppl 1):98-100.
Berry DJ, Sutherland CJ, Trousdale RT, Colwell CW Jr, Chandler HP, Ayres D, Yashar AA. Bilobed oblong porous coated acetabular components in revision total hip arthroplasty. Clin Orthop Relat Res 2000; 371:154-60.
Bin Nasser A, Beaulé PE, O'Neill M, Kim PR, Fazekas A. Incidence of groin pain after metal-on-metal hip resurfacing. Clin Orthop Relat Res 2010; 468(2):392-9.
Blackley HR, Davis AM, Hutchison CR, Gross AE. Proximal femoral allografts for reconstruction of bone stock in revision arthroplasty of the hip. A nine to fifteen-year follow-up. J Bone Joint Surg Am 2001; 83(3):346-54.
Bozic KJ, Kurtz SM, Lau E, Ong K, Vail TP, Berry DJ. The epidemiology of revision total hip arthroplasty in the United States. J Bone Joint Surg Am 2009; 91:128-33.
Bradford MS, Jablonsky WS, Paprosky WG, Younger TI. Revision of the difficult femur using the extended proximal femoral osteotomy. AAOS 1999; 48:19.
Bruns J, Delling G, Gruber H, Lohmann CH, Habermann CR. Cementless fixation of megaprostheses using a conical fluted stem in the treatment of bone tumours. J Bone Joint Surg Br 2007; 89(8):1084-7.
Busch VJ, Gardeniers JW, Verdonschot N, Slooff TJ, Schreurs BW. Acetabular reconstruction with impaction bone-grafting and a cemented cup in patients younger than fifty years old: a concise follow-up, at twenty to twenty-eight years, of a previous report. J Bone Joint Surg Am 2011; 93(4):367-71.
Callaghan JJ. Revision of the femoral component using cement fixation. In: Bono JV, McCarthy JC, Thornhil TS, Bierbaum BE, Turner RH. (eds.) Revision total hip arthroplasty. New York: Springer-Verlag; 1999. p. 204.
Callaghan JJ, Salvati EA, Pellicci PM, Wilson PD Jr, Ranawat CS. Results of revision for mechanical failure after cemented total hip replacement, 1979 to 1982. A two to five-year follow-up. J Bone Joint Surg Am 1985; 67(7):1074-85.
Cameron HU. Orthopaedic crossfire - Stem modularity is unnecessary in revision total hip arthroplasty: in opposition. J Arthroplasty 2003; 18(3 Suppl 1):101-3.
Chandler H, Clark J, Murphy S, McCarthy J, Penenberg B, Danylchuk K, Roehr B. Reconstruction of major segmental loss of the proximal femur in revision total hip arthroplasty. Clin Orthop Relat Res 1994; 298:67-74.
Chen WM, Engh CA Jr, Hopper RH Jr, McAuley JP, Engh CA. Acetabular revision with use of a bilobed component inserted without cement in patients who have acetabular bone stock deficiency. J Bone Joint Surg Am 2000; 82(2):197-206.
Chiang PP, Burke DW, Freiberg AA, Rubash HE. Osteolysis of the pelvis. Evaluation and treatment. Clin Orthop Relat Res 2003; 417:164-74.
Christie MJ, Barrington SA, Brinson MF, Ruhling ME, DeBoer DK. Bridging massive acetabular defects with the triflange cup: 2- to 9-year results. Clin Orthop Relat Res 2001; 393:216-27.
Civinini R, Capone A, Carulli C, Villano M, Gusso MI. Acetabular revisions using a cementless oblong cup: five to ten year results. Int Orthop 2008; 32(2):189-93.
Comba F, Buttaro M, Pusso R, Piccaluga F. Acetabular reconstruction with impacted bone allografts and cemented acetabular components: a 2- to 13-year follow-up study of 142 aseptic revisions. J Bone Joint Surg Br 2006; 88(7):865-69.
Cook SD, Barrack RL, Thomas KA, Haddad RJ Jr. Tissue growth into porous primary and revision femoral stems. J Arthroplasty 1991; 6(Suppl):S37-46.
Cummins F, Reilly PO, Flannery O, Kelly D, Kenny P. Defining the impaction frequency and threshold force required for femoral impaction grafting in revision hip arthroplasty. A human cadaveric mechanical study. Acta Orthop 2011; 82(4):433-7.
D'Antonio J, McCarthy JC, Bargar WL, Borden LS, Cappelo WN, Collis DK, Steinberg ME, Wedge JH. Classification of femoral abnormalities in total hip arthroplasty. Clin Orthop Relat Res 1993; 296:133-39.
D’Antonio JA. Periprosthetic bone loss of the acetabulum. Classification and management. Orthop Clin North Am 1992; 23:279-90.
D'Antonio JA, Capello WN, Borden LS, Burgar WL, Bierbaum BE, Boettcher WG, Steinberg ME, Stulberg SD, Wedge JH. Classification and management of acetabular abnormalities in total hip arthroplasty. Clin Orthop Relat Res 1989; 243:126-37.
Dearborn JT, Harris WH. Acetabular revision arthroplasty using so-called jumbo cementless components: an average 7-year follow-up study. J Arthroplasty 2000; 15(1):8-15.
Dearborn JT, Harris WH. High placement of an acetabular component inserted without cement in a revision total hip arthroplasty. Results after a mean of ten years. J Bone Joint Surg Am 1999; 81(4):469-80.
DeBoer DK, Christie MJ, Brinson MF, Morrison JC. Revision total hip arthroplasty for pelvic discontinuity. J Bone Joint Surg Am 2007; 89(4):835-40.
DeBoer DK, Christie MJ. Reconstruction of the deficient acetabulum with an oblong prosthesis: three- to seven-year results. J Arthroplasty 1998; 13(6):674-80.
Della Valle CJ, Shuaipaj T, Berger RA, Rosenberg AG, Shott S, Jacobs JJ, Galante JO. Revision of the acetabular component without cement after total hip arthroplasty. A concise followup, at fifteen to nineteen years, of a previous report. J Bone Joint Surg Am 2005; 87(8):1795-800.
Della Valle CJ, Berger RA, Rosenberg AG, Galante JO. Cementless acetabular reconstruction in revision total hip arthroplasty. Clin Orthop Relat Res 2004; 420:96-100.
Della Valle CJ, Paprosky WG. Classification and an algorithmic approach to the reconstruction of femoral deficiency in revision total hip arthroplasty. J Bone Joint Surg Am 2003; 85:(Suppl. 4):1-6.
Dennis DA. Management of massive acetabular defects in revision total hip arthroplasty. J Arthroplasty 2003; 18(3 Suppl 1):121-5.
Deirmengian C, Hallab N, Tarabishy A, et al. Synovial fluid biomarkers for periprosthetic infection. Clin Orthop Relat Res 2010; 468(8):2017-23.
Duncan WW, Hubble MJW, Howell JR, Whitehouse SL, Timperley AJ, Gie GA. Revision of the femoral stem using a cement-in-cement technique: a five- to 15-year review. J Bone Joint Surg Br 2009; 91(5):577-82.
Eftekhar NS. Principles of total hip arthroplastv. St Louis : CV Mosby Co; 1978.
Eisler T, Svensson O, Iyer V, Wejkner B, Schmalholz A, Larsson H, Elmstedt E. Revision total hip arthroplasty using third-generation cementing technique. J Arthroplasty. 2000 Dec;15(8):974-81.
Eldridge JD, Smith EJ, Hubble MJ, Whitehouse SL, Learmonth ID. Massive early subsidence following femoral impaction grafting. J Arthroplasty 1997; 12(5):535-40.
Engelbrecht DJ, Weber FA, Sweet MBE, Jakim I. Long term results of revision total hip arthroplasty. J Bone Joint Surg Br 1990; 72:41-5.
Engelbrecht E, Siegel A. Komplikationen nach Hüftgelenks-endoprothesen. Radiologe 1989; 29(10): 508-18.
Engh CA Jr, Hopper RH Jr, Engh CA Sr. Distal ingrowth components. Clin Orthop Relat Res 2004; 420:135-41.
Engh CA, Glassman AH, Griffin WL, Mayer JG. Results of cementless revision for failed cemented total hip arthroplasty. Clin Orthop Relat Res 1988; 235:91-110.
Estok DM 2nd, Harris WH. Long-term results of cemented femoral revision surgery using second-generation techniques. An average 11.7-year follow-up evaluation. Clin Orthop Relat Res 1994; 299:190-202.
Etienne G, Bezwada HP, Hungerford DS, Mont MA. The incorporation of morselized bone grafts in cementless acetabular revisions. Clin Orthop Relat Res 2004; 428:241-46.
Farfalli GL, Buttaro MA, Piccaluga F. Femoral fractures in revision hip surgeries with impacted bone allograft. Clin Orthop Relat Res 2007; 462:130-6.
Fink B, Grossmann A, Schulz MS. Bone regeneration in the proximal femur following implantation of modular revision stems with distal fixation. Arch Orthop Trauma Surg 2011; 131(4):465-70.
Fink B, Grossmann A. Modified transfemoral approach to revision arthroplasty with uncemented modular revision stems. Oper Orthop Traumatol 2007; 19(1):32-55.
Flannery OM, Britton JR, O'Reilly P, Mahony N, Prendergast PJ, Kenny PJ. The threshold force required for femoral impaction grafting in revision hip surgery. Acta Orthop 2010; 81(3):303-7.
Franzén H, Toksvig-Larsen S, Lidgren L, Onnerfält R. Early migration of femoral components revised with impacted cancellous allografts and cement. A preliminary report of five patients. J Bone Joint Surg Br 1995; 77(6):862-4.
Garbuz D, Morsi E, Gross AE. Revision of the acetabular component of a total hip arthroplasty with a massive structural allograft. Study with a minimum five-year follow-up. J Bone Joint Surg Am 1996 May;78(5):693-7.
Garcia-Cimbrelo E, Cruz-Pardos A, Garcia-Rey E, Ortega-Chamarro J. The survival and fate of acetabular reconstruction with impaction grafting for large defects. Clin Orthop Relat Res 2010; 468(12):3304-13.
Gerber SD, Harris WH: Femoral head autografting to augment acetabular deficiency in patients requiring total hip replacement. J Bone Joint Surg 1986; 68:1241-8.
Gie GA, Linder L, Ling RS, Simon JP, Slooff TJ, Timperley AJ. Impacted cancellous allografts and cement for revision total hip arthroplasty. J Bone Joint Surg Br 1993; 75:14-21.
Gokhale S, Soliman A, Dantas JP, Richardson JB, Cook F, Kuiper JH, Jones P. Variables affecting initial stability of impaction grafting for hip revision. Clin Orthop Relat Res 2005; 432:174-80.
Goodman S, Saastamoinen H, Shasha N, Gross A. Complications of ilioischial reconstruction rings in revision hip surgery. J Arthroplasty 2004;19:436-46.
Greenwald AS, Narten NC, Wilde AH. Points in the technique of recementing in the revision of an implant arthroplasty. J Bone Joint Surg Br 1978; 60(1):107-10.
Gross AE. Restoration of acetabular bone loss 2005. J Arthroplasty 2006; 21(4 Suppl 1):117-20.
Gross AE, Duncan CP, Garbuz D, Mohamed EM. Revision arthroplasty of the acetabulum in association with loss of bone stock. Instr Course Lect 1998;48:57-66.
Gross AE, Hutchison CR, Alexeeff M, Mahomed N, Leitch K, Morsi E. Proximal femoral allografts for reconstruction of bone stock in revision arthroplasty of the hip. Clin Orthop Relat Res 1995; 319:151-8.
Gustilo RB, Pasternak HS. Revision total hip arthroplasty with titanium ingrowth prosthesis and bone grafting for failed cemented femoral component loosening. Clin Orthop Relat Res 1988; 235:111-9.
Haddad FS, Spangehl MJ, Masri BA, Garbuz DS, Duncan CP. Circumferential allograft replacement of the proximal femur. A critical analysis. Clin Orthop Relat Res 2000; 371:98-107.
Haentjens P, De Boeck H, Opdecam P. Proximal femoral replacement prosthesis for salvage of failed hip arthroplasty: complications in a 2-11 year follow-up study in 19 elderly patients. Acta Orthop Scand 1996; 67(1):37-42.
Haentjens P, de Boeck H, Handelberg F, Casteleyn PP, Opdecam P. Cemented acetabular reconstruction with the Müller support ring. A minimum five-year clinical and roentgenographic follow-up study. Clin Orthop Relat Res 1993; 290:225-35.
Halliday BR, English HW, Timperley AJ, Gie GA, Ling RS Femoral impaction grafting with cement in revision total hip replacement. Evolution of the technique and results. J Bone Joint Surg Br 2003; 85(6):809-17.
Harris WH, Krushell RJ, Galante JO. Results of cementless revisions of total hip arthroplasties using the Harris-Galante prosthesis. Clin Orthop Relat Res 1988; 235:120-6.
Herberts P, Malchau H. Long-term registration has improved the quality of hip replacement: a review of the Swedish THR Register comparing 160,000 cases. Acta Orthop Scand. 2000 Apr;71(2):111-21.
Holt G, Hook S, Hubble M. Revision total hip arthroplasty: the femoral side using cemented implants. Int Orthop 2011; 35(2):267-73.
Hooten JP Jr, Engh CA Jr, Engh CA. Failure of structural acetabular allografts in cementless revision hip arthroplasty. J Bone Joint Surg Br 1994; 76(3):419-22.
Hunter GA, Welsh RP, Cameron HU, Bailey WH. The results of revision of total hip arthroplasty. J Bone Joint Surg Br 1979; 61:419-21.
Huo MH, Salvati EA. Revision total hip arthroplasty using cement. Tech Orthop1993; 7:58-64.
Iorio R, Healy WL, Presutti AH. A prospective outcomes analysis of femoral component fixation in revision total hip arthroplasty. J Arthroplasty 2008; 23:662-9.
Issack PS, Nousiainen M, Beksac B, Helfet DL, Sculco TP, Buly RL. Acetabular component revision in total hip arthroplasty. Part II: management of major bone loss and pelvic discontinuity. Am J Orthop 2009; 38(11):550-6.
Issack PS, Nousiainen M, Beksac B, Helfet DL, Sculco TP, Buly RL. Acetabular component revision in total hip arthroplasty. Part I: cementless shells. Am J Orthop 2009; 38(10):509-14.
Jamali AA, Dungy DS, Mark A, Schule S, Harris WH. Isolated acetabular revision with use of the Harris-Galante Cementless Component. Study with intermediate-term follow-up. J Bone Joint Surg Am 2004; 86(8):1690-7.
Joshi AB, Lee J, Christensen C. Results for a custom acetabular component for acetabular deficiency. J Arthroplasty 2002;17:643-8.
Kang MN, Huddleston JI, Hwang K, Imrie S, Goodman SB. Early outcome of a modular femoral component in revision total hip arthroplasty. J Arthroplasty 2008; 23: 220-5.
Karrholm J, Hultmark P, Carlsson L, Malchau H. Subsidence of a non-polished stem in revisions of the hip using impaction allograft: evaluation with radiostereometry and dual-energy x-ray absorptiometry. J Bone Joint Surg Br 1999; 81:135-42.
Katz RP, Callaghan JJ, Sullivan PM, Johnston RC. Long-term results of revision total hip arthroplasty with improved cementing technique. J Bone Joint Surg Br 1997; 79:322-6.
Kavanagh BF, Wallrichs S, Ilstrup D, Berry D, Fitzgerald RH. Ten year follow up of cemented revision total hip replacement. Orthop Trans 1993; 17:943.
Kavanagh BF, Fitzgerald RH Jr. Multiple revisions for failed total hip arthroplasty not associated with infection. J Bone Joint Surg Am 1987; 69(8):1144-9.
Kavanagh BF, Ilstrup DM, Fitzgerald RH Jr. Revision total hip arthroplasty. J Bone Joint Surg Am 1985; 67(4):517-26.
Kelly MP, Bozic KJ. Cost drivers in total hip arthroplasty: effects of procedure volume and implant selling price. Am J Orthop 2009; 38(1):E1-4.
Kershaw Ci, Atkins RM, Dodd CAF, Bulstrode CJK. Revision total hip arthroplasty for aseptic failure: a review of 276 cases. J Bone Joint Surg Br 1991; 73:564-8.
Kinov P. Aseptic loosening of hip arthroplasty. PhD thesis. Medical University of Sofia; 2007.
Kinov P, Bukarev D, Dimov V, Kazakov K, Tivchev P. Revision total hip arthroplasty with cementing technique-five-year results. Chir Narzadow Ruchu Ortop Pol 2007; 72(4):293-6.
Knight JL, Helming C. Collarless polished tapered impaction grafting of the femur during revision total hip arthroplasty: pitfalls of the surgical technique and follow-up in 31 cases. J Arthroplasty 2000; 15(2):159-65.
Kosashvili Y, Backstein D, Safir O, Lakstein D, Gross AE. Krishnamurthy AB, MacDonald SJ, Paprosky WG. 5- to Acetabular revision using an anti-protrusion (ilio-ischial) cage and trabecular metal acetabular component for severe acetabular bone loss associated with pelvic discontinuity. J Bone Joint Surg Br 2009; 91(7):870-6.
Krishnamurthy AB, MacDonald SJ, Paprosky WG. 5-13-year follow-up study on cementless femoral components in revision surgery. J Arthroplasty 1997; 12(8):839-47.
Kurtz S, Mowat F, Ong K, Chan N, Lau E, Halpern M. Prevalence of primary and revision total hip and knee arthroplasty in the United States from 1990 through 2002. J Bone Joint Surg Am 2005; 87:1487-97.
Kwong LM, Jasty M, Harris WH. High failure rate of bulk femoral head allografts in total hip acetabular reconstructions at 10 years. J Arthroplasty 1993; 8:341-6.
Lachiewicz PE, Hussamy OD. Revision of the acetabulum without cement with use of the Harris-Galante porous-coated implant: two to eight-year results. J Bone Joint Surg Am 1994; 76:1834-9.
Lakstein D, Kosashvili Y, Backstein D, Safir O, Lee P, Gross AE. Revision total hip arthroplasty with a modular tapered stem. Hip Int 2010; 20(2):136-42.
Lakstein D, Backstein D, Safir O, Kosashvili Y, Gross AE. Trabecular Metal cups for acetabular defects with 50% or less host bone contact. Clin Orthop Relat Res 2009; 467(9):2318-24.
Langlais F, J. C. Lambotte, P. Collin, and H. Thomazeau, Long-term results of allograft composite total hip prostheses for tumors. Clin Orthop Relat Res 2003; 414:197-211.
Lawrence JM, Engh CA, Macalino GE. Revision total hip arthroplasty. Long-term results without cement. Orthop Clin North Am 1993; 24(4):635-44.
Li PL, Ingle PJ, Dowell JK. Cement-within-cement revision hip arthroplasty: should it be done? J Bone Joint Surg Br 1996; 78(5):809-11.
Lieberman JR, Moeckel BH, Evans BG, Salvati EA, Ranawat CS. Cement-within-cement revision hip arthroplasty. J Bone Joint Surg Br 1993; 75(6):869-71.
Malkani AL, F. H. Sim, and E. Y. S. Chao. Custom-made segmental femoral replacement prosthesis in revision total hip arthroplasty. Orthop Clin North Am 1993; 24(4):727-33.
Mallory TH. Preparation of the proximal femur in cementless total hip revision. Clin Orthop Relat Res 1988; 235:47-60.
Masterson S, Lidder S, Scott G. Impaction femoral allografting at revision hip arthroplasty: uncemented versus cemented technique using a Freeman femoral component. J Bone Joint Surg Br 2012; 94(1):51-5.
Masterson EL, Masri BA, Duncan CP. The cement mantle in the Exeter impaction allografting technique. A cause for concern. J Arthroplasty 1997; 12(7):759-64.
McAuley JP, Engh CA Jr. Femoral fixation in the face of considerable bone loss: cylindrical and extensively coated femoral components. Clin Orthop Relat Res 2004; 429:215-21.
Meding JB, Ritter MA, Keating EM, Faris PM. Impaction bonegrafting before insertion of a femoral stem with cement in revision total hip arthroplasty: a minimum two-year follow-up study. J Bone Joint Surg Am 1997; 79:1834-41.
Miner TM, Momberger NG, Chong D, Paprosky WL. The extended trochanteric osteotomy in revision hip arthroplasty: a critical review of 166 cases at mean 3-year, 9-month follow-up. J Arthroplasty 2001; 16(Suppl 1):188-94.
Moreland JR, Moreno MA. Cementless femoral revision arthroplasty of the hip: minimum 5 years followup. Clin Orthop Relat Res 2001; 393:194-201.
Moreland JR, Bernstein ML. Femoral revision hip arthroplasty with uncemented, porous-coated stems. Clin Orthop Relat Res 1995; 319:141-50.
Morsi E, Garbuz D, Gross AE. Revision total hip arthroplasty with shelf bulk allografts. A long-term follow-up study. J Arthroplasty 1996;11(1):86-90.
Moskal JT, Higgins ME, Shen J. Type III acetabular defect revision with bilobed components: five-year results. Clin Orthop Relat Res 2008; 466(3):691-5.
Mulroy WF, Harris WH. Revision total hip arthroplasty with use of so-called second-generation cementing techniques for aseptic loosening of the femoral component. A fifteen-year-average follow-up study. J Bone Joint Surg Am 1996; 78:325-30.
Nadaud MC, Griffin WL, Fehring TK, Mason JB, Tabor OB Jr, Odum S, Nussman DS. Cementless revision total hip arthroplasty without allograft in severe proximal femoral defects. J Arthroplasty 2005; 20(6):738-44.
Nehme A, Lewallen DB, Hanssen AD. Modular porous metal augments for treatment of severe acetabular bone loss during revision hip arthroplasty. Clin Orthop Relat Res 2004; 429:201-8.
Nelson CL. Cemented cup revisions. Am J Orthop 2002; 31(8):479-80.
Obenaus C, Winkler H, Girtler R, Huber M, Schwägerl W. Extra-large press-fit cups without screws for acetabular revision. J Arthroplasty 2003; 18(3):271-7.
Ornstein E, Linder L, Ranstam J, Lewold S, Eisler T, Torper M. Femoral impaction bone grafting with the Exeter stem - the Swedish experience: survivorship analysis of 1,305 revisions performed between 1989 and 2002. J Bone Joint Surg Br 2009; 91(4):441-6.
Paprosky W G, Greidanus N V, Antoniou J. Minimum 10-year-results of extensively porous-coated stems in revision hip arthroplasty. Clin Orthop Relat Res 1999; 369:230-42.
Paprosky WG, Perona PG, Lawrence JM. Acetabular defect classification and surgical reconstruction in revision arthroplasty. A 6-year follow-up evaluation. J Arthroplasty 1994; 9(1):33-44.
Park DK, Della Valle CJ, Quigley L, Moric M, Rosenberg AG, Galante JO. Revision of the acetabular component without cement. A concise follow-up, at twenty to twenty-four years, of a previous report. J Bone Joint Surg Am 2009; 91:350-5.
Parvizi J, Jacovides C, Antoci V, Ghanem E. Diagnosis of periprosthetic joint infection: the utility of a simple yet unappreciated enzyme. J Bone Joint Surg Am 2011; 93(24):2242-48.
Parvizi J, Della Valle CJ. AAOS Clinical Practice Guideline: diagnosis and treatment of periprosthetic joint infections of the hip and knee. J Am Acad Orthop Surg 2010; 18(12):771-2.
Parvizi J, Tarity TD, Slenker N, Wade F, Trappler R, Hozack WJ, Sim FH. Proximal femoral replacement in patients with non-neoplastic conditions. J Bone Joint Surg Am 2007; 89(5):1036-43.
Pellicci PM, Wilson Jr PD, Sledge CB, Salvati EA, Ranawat CS, Poss R, Callaghan JJ.. Long-term results of revision total hip replacement. A follow-up report. J Bone Joint Surg Am 1985; 67:513-6.
Pellicci PM, Wilson PD Jr, Sledge CB, Salvati EA, Ranawat CS, Poss R. Revision total hip arthroplasty. Clin Orthop Relat Res 1982; 170:34-41.
Pekkarinen J, Alho A, Lepistö J, Ylikoski M, Ylinen P, Paavilainen T. Impaction bone grafting in revision hip surgery. A high incidence of complications. J Bone Joint Surg Br 2000; 82(1):103-7.
Piccaluga F, González Della Valle A, Encinas Fernández JC, Pusso R. Revision of the femoral prosthesis with impaction allografting and a Charnley stem. A 2- to 12-year follow-up. J Bone Joint Surg Br 2002; 84(4):544-9.
Pollock FH, Whiteside LA. The fate of massive allografts in total hip acetabular revision surgery. J Arthroplasty 1992; 7:271-6.
Pulido L, Rachala SR, Cabanela ME. Cementless acetabular revision: past, present, and future. Revision total hip arthroplasty: the acetabular side using cementless implants. Int Orthop 2011; 35(2):289-98.
Raut VV, Siney PD, Wroblewski BM. Cemented Charnley revision arthroplasty for severe femoral osteolysis. J Bone Joint Surg Br 1995; 77(3):362-5.
Raut VV, Siney PD, Wroblewski BM: Cemented revision for aseptic acetabular loosening: a review of 387 hips. J Bone Joint Surg Br 1995; 77:357-61.
Regis D, Sandri A, Bonetti I, Bortolami O, Bartolozzi P. A minimum of 10-year follow-up of the Burch-Schneider cage and bulk allografts for the revision of pelvic discontinuity. J Arthroplasty 2012; 27(6):1057-63.
Ritter MA, Thong AE. The role of cemented sockets in 2004: is there one? J Arthroplasty 2004; 19(4 suppl 1):92-4.
Rogers BA, Sternheim A, Backstein D, Safir O, Gross AE. Proximal femoral allograft for major segmental femoral bone loss: a systematic literature review. Adv Orthop 2011; 2011:257572.
Roque PJ, Mankin HJ, Malchau H. Proximal femoral allograft: prognostic indicators. J Arthroplasty 2010; 25(7):1028-33.
Rosenstein A, MacDonald W, Iliadis A, McLardy-Smith P. Revision of cemented fixation and cement-bone interface strength. Proc Inst Mech Eng (H) 1992; 206(1):47-9.
Rubash HE, Harris WH. Revision of nonseptic, loose, cemented femoral components using modem cementing techniques. J Arthroplasty 1988; 3:241-8.
Russotti GM, Harris WH. Proximal placement of the acetabular component in total hip arthroplasty. A long-term follow-up study. J Bone Joint Surg Am 1991; 73(4):587-92.
Safir O, Kellett CF, Flint M, Backstein D, Gross AE. Revision of the deficient proximal femur with a proximal femoral allograft. Clin Orthop Relat Res 2009; 467(1):206-12.
Saleh KJ, Holtzman J, Gafni A, Saleh L, Davis A, Resig S, Gross AE. Reliability and intraoperative validity of preoperative assessment of standardized plain radiographs in predicting bone loss at revision hip surgery. J Bone Joint Surg Am 2001; 83:1040-6.
Saleh KJ, Jaroszynski G, Woodgate I, Saleh L, Gross AE. Revision total hip arthroplasty with the use of structural acetabular allograft and reconstruction ring: a case series with a 10-year average follow-up. J Arthroplasty 2000; 15(8):951-8.
Schreurs BW, Luttjeboer J, Thien TM, de Waal Malefijt MC, Buma P, Veth RP, Slooff TJ. Acetabular revision with impacted morselized cancellous bone graft and a cemented cup in patients with rheumatoid arthritis. A concise follow-up, at eight to nineteen years, of a previous report. J Bone Joint Surg Am 2009; 91(3):646-51.
Schreurs BW, Arts JJ, Verdonschot N, Buma P, Slooff TJ, Gardeniers JW. Femoral component revision with use of impaction bone-grafting and a cemented polished stem. Surgical technique. J Bone Joint Surg Am 2006; 88(Suppl 1 Pt 2):259-74.
Schreurs BW, Arts JJ, Verdonschot N, Buma P, Slooff TJ, Gardeniers JW. Femoral component revision with use of impaction bone-grafting and a cemented polished stem. J Bone Joint Surg Am 2005; 87(11):2499-507.
Schreurs BW, Bolder SB, Gardeniers JW, Verdonschot N, Slooff TJ, Veth RP. Acetabular revision with impacted morsellised cancellous bone grafting and a cemented cup. A 15- to 20-year follow-up. J Bone Joint Surg Br 2004; 86(4):492-7.
Schreurs BW, Slooff TJ, Buma P, Gardeniers JW, Huiskes R. Acetabular reconstruction with impacted morsellised cancellous bone graft and cement. A 10- to 15-year follow-up of 60 revision arthroplasties. J Bone Joint Surg Br 1998; 80(3):391-395.
Shinar AA, Harris WH. Bulk structural autogenous grafts and allografts for reconstruction of the acetabulum in total hip arthroplasty. Sixteen-year-average follow-up. J Bone Joint Surg Am 1997; 79(2):159-68.
Silverton CD, Rosenberg AG, Sheinkop MB, Kull LR, Galante JO. Revision of the acetabular component without cement after total hip arthroplasty. A follow-up note regarding results at seven to eleven years. J Bone Joint Surg Am 1996; 78(9):1366-70.
Slooff TJ, Huiskes R, van Horn J, Lemmens AJ. Bone grafting in total hip replacement for acetabular protrusion. Acta Orthop Scand 1984; 55: 593-6.
Sporer SM, Paprosky WG. Acetabular revision using a trabecular metal acetabular component for severe acetabular bone loss associated with a pelvic discontinuity. J Arthroplasty 2006; 21(6 Suppl 2):87-90.
Sporer SM, O’Rourke M, Paprosky WG. The treatment of pelvic discontinuity during acetabular revision. J Arthroplasty 2005; 20(4 Suppl 2):79-84.
Sternheim A, Rogers BA, Kuzyk PR, Safir OA, Backstein D, Gross AE. Segmental proximal femoral bone loss and revision total hip replacement in patients with developmental dysplasia of the hip: the role of allograft prosthesis composite. J Bone Joint Surg Br 2012; 94(6):762-7.
Sutherland AG, D’Arcy S, Smart D, Ashcroft GP. Removal of the subchondral plate in acetabular preparation. Int Orthop 2000; 24(1):19-22.
Sutherland CJ. Early experience with eccentric acetabular components in revision total hip arthroplasty. Am J Orthop 1996; 25:284-9.
Tanzer M, Drucker D, Jasty M, McDonald M, Harris WH. Revision of the acetabular component with an uncemented Harris-Galante porous-coated prosthesis. J Bone Joint Surg Am 1992; 74(7):987-94.
Templeton JE, Callaghan JJ, Goetz DD, Sullivan PM, Johnston RC. Revision of a cemented acetabular component to a cementless acetabular component. A ten to fourteen-year follow-up study. J Bone Joint Surg Am 2001; 83(11):1706-11.
van Egmond N, De Kam DC, Gardeniers JW, Schreurs BW. Revisions of extensive acetabular defects with impaction grafting and a cement cup. Clin Orthop Relat Res 2011; 469(2):562-73.
Vastel L, Lemoine CT, Kerboull M, Courpied JP. Structural allograft and cemented long-stem prosthesis for complex revision hip arthroplasty: use of a trochanteric claw plate improves final hip function. Int Orthop 2007; 31(6):851-7.
Wagner H, Wagner M. Cone prosthesis for the hip joint. Arch Orthop Trauma Surg 2000; 120(1-2):88-95.
Wedemeyer C, Neuerburg C, Heep H, von Knoch F, von Knoch M, Löer F, Saxler G. Jumbo cups for revision of acetabular defects after total hip arthroplasty: a retrospective review of a case series. Arch Orthop Trauma Surg 2008; 128:545-50.
Weeden SH, Paprosky WG. Minimum 11-year follow up of extensively porous coated stems in femoral revision total hip arthroplasty. J Arthroplasty 2002; 17(Suppl 1):134-7.
Whaley AL, Berry DJ, Harmsen WS. Extra-large uncemented hemispherical acetabular components for revision total hip arthroplasty. J Bone Joint Surg Am 2001; 83(9):1352-7.
Winter E, Piert M, Volkmann R, Maurer F, Eingartner C, Weise K, Weller S. Allogeneic cancellous bone graft and a Burch-Schneider ring for acetabular reconstruction in revision hip arthroplasty. J Bone Joint Surg Am. 2001 Jun;83-A(6):862-7.
Wraighte PJ, Howard PW. Femoral impaction bone allografting with an Exeter cemented collarless, polished, tapered stem in revision hip replacement: a mean follow-up of 10.5 years. J Bone Joint Surg Br 2008; 90(8):1000-4.
Wysocki RW, Della Valle CJ, Shott S, Rosenberg AG, Berger RA, Galante JO. Revision of the acetabulum with a contemporary cementless component. J Arthroplasty 2009; 24(6 Suppl):58-63.
Zehntner MK, Ganz R. Midterm results (5.5-10 years) of acetabular allograft reconstruction with the acetabular reinforcement ring during total hip revision. J Arthroplasty 1994; 9:469-79.
Zehr RJ, Enneking WF, Scarborough MT. Allograft-prosthesis composite versus megaprosthesis in proximal femoral reconstruction. Clin Orthop Relat Res 1996; 322:207-23.