Electrochemical biosensors are transducers that convert biological information into electrical information. Electrochemical biosensors provide qualitative and quantitative information (Wang 1999) on the existence and concentration of the target compounds in the analyte in the form of current (amperometric biosensor) or voltage (potentiometric biosensor).
A typical amperometric biosensor consists of three components: the analyte, the transduction element (electrode and conductive nanomaterials) and the biorecognition element (enzyme) (McLamore et al., 2010a; McLamore et al., 2010b; McLamore et al., ; Shi et al., 2010). During biosensor operation, target compound in the sample is specifically recognized by the enzymes immobilized on the electrode. Electrooxidative intermediate is produced by this enzyme-substrate interaction. The produced electrooxidative intermediate is oxidized or reduced by the voltage applied on the biosensor, and current proportional to substrate concentration is generated and recorded. By calibrating the biosensor using solutions with known concentration, the relationship between measured current and substrate concentration is obtained. The sensitivity and specificity of the sensor is ensured by the high selectivity of enzymes.
Considering the functional mechanism of biosensors, surface modification of the electrode is vital to biosensor performance. The most straightforward and also widely used approach is to immobilize enzymes on the electrode with a polymer layer. However, this method has two major limitations. One is that the activity of the enzymes can be affected by structural change due to the polymer layer, and affected by the pH of the layer (Zou et al., 2008). The other is that the thickness of the polymer layer cannot be precisely controlled, so the response time and sensitivity of the biosensor could be affected (Li et al., 1996). To overcome these limitations, some groups used polymers with neutral pH such as silicate sol-gel for enzyme immobilization to preserve enzyme activity (Salimi et al., 2004) while some groups used electric methods such as cyclic voltammetry to control layer deposition (Llaudet et al., 2005; Smutok et al., 2006). Furthermore, to obtain better performance, nanomaterials including carbon nanotubes (CNTs) and metal nanomaterials are often involved in surface modification (McLamore et al., 2010a; McLamore et al., 2010b; McLamore et al., ; Shi et al., 2010). Since different modification approaches result in quite distinct biosensor performance, problems with evaluating and comparing different approaches, and sorting out the optimal ones have arisen. To solve this problem, a standardization method which evaluates the performance of biosensors constructed by different approaches is needed.
In this chapter, followed by a comprehensive literature review of surface modification approaches, a tentative protocol for comparing different approaches will be discussed.
2. Immobilization approaches for enzymes
As was mentioned previously, enzymes are the biorecognition element of biosensors. Biosensors function based on the highly selective enzyme-substrate interactions. Thus, the enzymes immobilized on electrode determine the target compound, the activity of the enzymes determines the sensitivity, and the selectivity of the enzymes determines the specificity of the biosensors. As a result, it is important to develop proper enzyme immobilization approaches with high enzyme loading and well-preserved enzyme activity.
2.1. Enzyme based biosensing
Enzymes are usually immobilized on the electrode by polymer encapsulation or covalent linking (McLamore et al., 2010b; McLamore et al., 2011; Rickus et al., 2002; Shi et al., 2010). During biosensor operation, when analyte solution diffuses into the enzyme layer, a series of biochemical and electrochemical reactions will take place. Take the
In the first step (biorecognition), GOx converts glucose into H2O2 and gluconic acid. The main purpose of this step is to produce the electrooxidative intermediate H2O2, because glucose cannot be directly electrooxidized. Because the enzyme-substrate interaction in this step is specific to glucose, biorecognition step ensures the selectivity of the biosensors.
Step 1. Glucose + O2 GOx> Gluconic acid + H2O2
In the second step (transduction), an electric potential is applied to the electrode. The value of the potential is determined by the type of electrode used, and the type of the electroactive intermediate produced in
Step 2. H2O2 → O2 + 2H+ + 2e-
Since the concentration of H2O2 is proportional to glucose according to
2.2. Enzyme immobilization approaches
One of the most widely used approach for immobilizing enzymes is to entrap enzymes within polymer layers. The layer containing enzymes can be deposited on electrodes by cast-and-dry, or electropolymerization. Many polymers have been reported for such applications, including nafion (Fortier et al., 1992; Vaillancourt et al., 1999), polypyrrole (Branzoi & Pilan 2008; Ekanayake et al., 2007), polytyramine (Situmorang et al., 1999) and silicate sol-gels (Llaudet et al., 2005; Rickus et al., 2002; Salimi et al., 2004).
Nafion is a negatively charged sulfonated tetrafluorethylene copolymer, which possesses a strong surface adhesion to electrode surface and a low swelling capability in aqueous media (Gong et al., 2005; Liaw et al., 2006; Wang et al., 2003b). Thus, nafion is quite appropriate for enzyme immobilization. Biosensors based on nafion/enzyme composite for the detection of glucose and other compounds have been reported (Fortier et al., 1992; Vaillancourt et al., 1999). One noticeable advantage of nafion over other polymers is that the negative charges repel the diffusion of many negatively charged compounds such as ascorbate and acetaminophen into the layer (Ni et al., 1999), significantly enhancing biosensing selectivity.
Polypyrrole (PPy) is a conductive polymer mainly made up of pyrroles. Polypyrroles can be formed through electropolymerization using cyclic voltammetry, resulting in a uniformly doped PPy film with positive charges on electrode surface (Schuhmann 1991; Schuhmann & Kittsteiner-Eberle 1991; Schuhmann et al., 1990). One advantage with PPy is that enzymes with negative charges can be absorbed into PPy layers via electrostatic forces (Gao et al., 2003). Another advantage is that the thickness of the PPy layer can be quantitatively controlled by controlling the number of cycles during cyclic voltammetry. The selectivity of polypyrrole film can be enhanced by the addition of various counter ions (Sadik 1999; Teasdale & Wallace 1993; Zotti 1992). Biosensors based on PPy for versatile sensing applications have been reported (Dumont & Fortier 1996; Ekanayake et al., 2007; Umana & Waller 2002). Excellent reproducibility in amperometric response and resistance towards high temperature have been reported for PPy over a number of polymers including polyaniline, poly(aniline/p-phenylediamine), polyindole, and poly(o-phenylediamine) (Dumont & Fortier 1996). The major disadvantage with PPy is that the layer is most stable under pH range of 5.5-6.0 (Dumont 1996), which may greatly lower the activities of certain enzymes that favor basic pH, such as glycerol kinase (optimal pH=9.8) and glycerol-3-phosphate oxidase (optimal pH=8.1), both of which are used in adenosine-3-phosphate (ATP) sensing (Llaudet et al., 2005). In addition, Schuhmann et al. reported that the enzyme loading capability of PPy was low (Schuhmann 1991), which may result in a low biosensor sensitivity.
Silicate sol-gels are polymers formed by ethyl esters of orthosilicic acid, among which tetraethyl orthosilicate (TEOS) and tetramethyl orthosilicate (TMOS) are most commonly used in the immobilization of enzymes (Llaudet et al., 2005; Salimi et al., 2004; Yang et al., 1998). The hydrolysis and condensation of sol-gels at low temperature (usually 4°C) generate a 3-dimensitional polymer matrix of silica, which can entrap enzymes (Rickus et al., 2002). Biosensors based on sol-gel approach for the detection of glucose (Salimi et al., 2004), ATP (Llaudet et al., 2005) and other compounds with linear response range covering physiological concentrations have been reported. One advantage of sol-gel immobilization is that enzymes are entrapped within the matrix with no covalent linking involved, thus enzyme activity may be better preserved. Another advantage is that the porous structure of sol-gel matrix facilitates the diffusion of substrates into the matrix and provides space for the interaction between substrates and enzymes. However, since immobilization approaches based on sol-gels require dip coating and the distribution of dissolved enzymes in the sol-gel is not uniform, the thickness of the layer and the amount of loaded enzymes may vary a lot, affecting the reproducibility of biosensors.
Other polymers such as chitosan (Kang et al., 2007; Miscoria et al., 2006) have been used for enzyme immobilization as well. Some approaches directly entrap enzymes in the polymer. The common drawback with these approaches is the relatively low efficacy of enzyme loading that often results in inconsistency in amperometric response and reduced sensitivity during long-term biosensor operation (Schuhmann 1991; Schuhmann & Kittsteiner-Eberle 1991). Thus, cross-linking agents have been combined with polymer layers for better enzyme loading. These agents include glutaraldehyde (GA) (via NH2- bond) (Guerrieri et al., 1998), 1-ethyl-3-(3-diamino)propyl-carbodiimide (EDC) and
3. Immobilization of nanomaterials
One problem with biosensors based only on polymers and enzymes is the undesired low signal-to-noise ratio, because catalytic ability of enzymes is limited. Consequently, biosensor’s amperometric response may be submerged by noise. One of the most commonly used approaches to resolve this problem is to modify biosensors with nanomaterials. Two most commonly used nanomaterials are carbon nanotubes and metal nanomaterials (McLamore et al., 2010a; McLamore et al., 2010b; McLamore et al., 2011; Shi et al., 2010).
Ever since Iijima reported the synthesis method for CNT in 1991(Iijima 1991), this allotrope of carbon has demonstrated versatile applications in biomedical imaging (Choi et al., 2007b), chemical batteries (Wang et al., 2003a), and biosensing (McLamore et al., 2010a; McLamore et al., 2010b; McLamore et al., 2011; Shi et al., 2010 ). CNTs have two types: single-walled CNT (SWNT) and multi-walled CNT (MWNT). SWNT is a seamless cylinder formed by rolling-over a one-atom-thick layer of graphite namely graphene (Iijima & Ichihashi 1993) (Fig. 1a), while MWNT has the structure of sheets of graphite arranged in concentric cylinders (Ajayan 1999; Dai 2002) (Fig. 1b). SWNT has a diameter on the order of 1.2 nm (Fig. 1d) while MWNT has a diameter on the order of 10 nm to 20 nm with concentric nanotubes 0.34 nm apart (Ajayan 1999; Dai 2002) (Fig. 1c).3.1. Electrochemical basis for CNT
STM/STS studies have shown that CNTs consist of both metallic and semi-conductive tubes (Odom et al., 1998; Wilder et al., 1998). Both SWNTs (Wang et al., 2003b) and MWNTs (McLamore et al., 2010a; McLamore et al., 2010b; McLamore et al., 2011; Shi et al., 2010 ) have been widely used in biosensing. CNTs have been demonstrated to possess the ability to facilitate the electron transfer process during electroreduction and electrooxidation of electroactive species, such as NADH and hydrogen peroxide (Hrapovic et al., 2004; Wang et al., 2003b), and the electron transfer process during enzyme-substrate interaction, even when the enzyme redox center is deeply embedded (Gooding et al., 2003).
Researches have been carried out to explore the underlying mechanism for CNT to enhance biosensor performance. The reasons for CNT to greatly improve biosensor’s response are summarized as follows:
First, CNTs enlarge the effective surface area when immobilized on the surface of the electrodes. The electrode impedance is decreased, and the current is increased due to the increase in surface area (Azamian et al., 2002). Another advantage due to enlarged surface area is that more enzymes can be immobilized. MWNTs have been used as a matrix for enzyme immobilization (Shi et al., 2010 ).
Second, CNTs act as a catalyst that increases electron transfer rate. The carbon atoms at the ends of CNT behave like the edge plane of highly oriented pyrolytic graphite (HOPG) from a mechanistic point of view (Li et al., 2002). When CNTs are pretreated by purifying and refluxing using strong acid such as nitric acid (McLamore et al., 2010a), the tube ends will be connected with oxygenated species, such as carboxylic acids, alcohols and quinines (Gooding 2005; Koehne et al., 2003). The oxygenated tube ends allow efficient electron transfer (Gooding 2005; Koehne et al., 2003), which is the origin for the catalytic ability of CNTs. This underlying mechanism is further supported by comparing peak separation in cyclic voltammogram of potassium ferricyanide between one electrode with aligned SWNTs perpendicular to its surface and another electrode with SWNTs with random orientations. The former has much a smaller separation than the latter, indicating improved electrochemical property (Liu et al., 2005).
Third, the electrodes are endowed with better wetting properties due to the porous structure of CNTs (Nugent et al., 2001). As a result, analyte solution will diffuse into the CNT bundles with lower friction (Verweij et al., 2007), which contributes to a higher current sensitivity when biosensing is diffusion limited (Cambiaso et al., 1996).
3.2. Surface modification approaches using CNTs
3.2.1. Abrasive immobilization
CNT, as an allotrope of carbon, can be attached to carbon electrode surface by non-covalent forces. Salimi et al. prepared glucose biosensor based on abrasive immobilization approach, by gently rubbing the polished basal plane pyrolytic graphite (bppg) electrode surface on a filter paper containing MWNTs (Salimi et al., 2004). Decreased oxidation and reduction potentials for H2O2 were discovered compared with bare bppg electrodes, indicating the improvement in electrocatlytic activities of the electrodes due to CNT immobilization (Salimi et al., 2004). In amperometric tests, well-defined response to glucose addition was reported for the bppg/CNT/sol-gel/GOx biosensor while hardly any response could be observed with the bppg/sol-gel/GOx electrodes (Salimi et al., 2004), demonstrating that the low signal-to-noise issue with biosensors based on conventional materials could be resolved by adding nanomaterials. In addition, compared with glucose biosensors with no CNT involved (Wang et al., 1997; Yang et al., 1998), the analytical parameters (sensitivity, detection limit, response time and linear range) for bppg/CNT/sol-gel/GOx biosensor were comparable or better (Salimi et al., 2004).
3.2.2. Immobilization with MPS
(3-Mercaptopropyl) trimethoxysilane (MPS), a silanization reagent with methoxy and thiol functional groups, has been applied to attach MWNTs to electrodes (McLamore et al., 2010a; Zeng & Huang 2004). The thiol groups form covalent bonds to link CNTs to electrodes. Biosensors based on this approach exhibited increased peak current in cyclic voltammetry with potassium ferricyanide, and high sensitivity towards the direct oxidation of IAA, due to the CNTs on electrode surface which facilitated electron transfer. MPS immobilization of CNTs provides an alternative to abrasive immobilization which can be applied to metal electrodes. Desirable reproducibility has been reported for biosensors based on this approach (Zeng & Huang 2004).
3.2.3. Immobilization with polymer entrapment
The major obstacle to immobilizing CNTs for biosensing is that CNTs tend to aggregate due to van der Walls forces among tubes. As a result, CNTs are insoluble in almost all solvents (Chen et al., 1998; Star et al., 2001). Since almost all conventional approaches for building enzyme based biosensors rely on polymer layers to entrap enzymes, similar approaches can be developed to immobilize CNT. Researches have shown that many polymer layers can suspend CNT, including nafion (McLamore et al., 2010b; McLamore et al., 2011; Shi et al., 2010 ; Tsai et al., 2005; Wang et al., 2003b), chitosan (Kang et al., 2007, 2008) and silicate sol-gels (Chen & Dong 2007; Gavalas et al., 2004).
Nafion is a conductive sulfonated tetrafluorethylene copolymer and its negatively charged layer is capable of suspending CNTs and enzymes. SEM image showed that MWNTs were well dispersed within nafion layer, and formed a conductive network which will facilitate electron transfer during electrochemical reactions (Shi et al., 2010 ) (Fig. 2).
Chitosan is a linear polysaccharide with fine biocompatibility and adhesive capability to chemically modified surfaces. Pretreated CNTs with –COOH groups on tube ends could disperse among chitosan containing –NH2 groups due to the peptide bonds formed between –COOH and –NH2 (Kang et al., 2007). Biosensors based on chitosan polymers with CNT and enzymes involved have been reported (Kang et al., 2007, 2008). Similar to other CNT modified electrodes, the oxidation potential for electrooxidative species is significantly lowered (Zhang 2004). A low oxidation potential ensures that interferences such as acetaminophen and ascorbic acid, that can only be oxidized at high voltages, are excluded, which greatly enhances the selectivity of the biosensors. However, one disadvantage with chitosan is that the peptide bonds formed between CNTs and chitosan eliminate the –COOH groups on CNT, which may lower the catalytic ability of CNTs, as the ability mainly comes from the oxidative species at tube ends.
Polypyrrole (PPy) is a highly conductive polymer formed from a number of connected pyrrole rings. Wang et al. reported that “oxidized CNT” together with enzymes could act as combined dopants to form a covalently linked PPy-CNT-Enzyme layer (Wang & Musameh 2005). When electro-oxidized at +650 mV using platinum (Pt) or glass carbon (GC) electrodes as working electrodes, each pyrrole ring will carry one positive charge. With the presence of charge balancing anionic dopants, such as negatively charged enzymes (Kang et al., 2007; Umana & Waller 2002) or –COOH modified CNTs (Wang & Musameh 2005), polymer layers with enzymes or CNTs will form on the working electrode surface after electropolymerization (Wang & Musameh 2005). Glucose biosensors based on this approach showed significantly increased response to glucose compared with no MWNT involved. In addition, thanks to irreversibly oxidized PPy’s special property to reject electroactive interferences (Malitesta et al., 1990), glucose biosensors based on PPy/MWNT exhibited no response towards uric and ascorbic acids even at +900 mV (Wang & Musameh 2005), showing excellent selectivity. Besides PPy, immobilization approaches based on similar electropolymerization process using polyaniline (PAN) was also reported (Ma et al., 2006). In addition, the auto-assembly linking of negatively charged oxygenated groups on modified CNTs to positively charged polyelectrolyte poly(diallyldimethylammonium chloride) (PDDA) layer with no need of electropolymerization was reported (Mamedov et al., 2002; Rouse & Lillehei 2002).
Silicate sol-gels, including tetramethyl orthosilicate (TMOS) and tetraethyl orthosilicate (TEOS), have been widely used in enzyme immobilization due to the formed porous 3-D matrix structure which physically entraps enzymes (Llaudet et al., 2005; Salimi et al., 2004; Yang et al., 1998). The use of sol-gels to immobilize CNTs on biosensors have been reported by directly dispersing CNTs within pretreated methyltriethoxysilane (MTEOS) (Gavalas et al., 2004), Propyltrimethoxysilane (PTMOS) (Gong et al., 2004) and methyltrimethoxysilane (MTMOS) (Chen & Dong 2007) solutions. Homogeneous suspensions were obtained after ultrasonication and sol-gel/CNT layers were formed on electrodes. TEM image of CNT and the CNT/sol-gel composite (Gong et al., 2004) showed that small MWNT bundles were separated into several independent nanoelectrodes, which greatly increased the contacting area between CNTs and analytes.
3.2.4. CNT paste electrodes
Almost all previously reviewed approaches immobilized CNTs on a substrate electrode, such as glassy carbon (GC), platinum (Pt) and gold (Au). CNTs can be directly packed into a carbon electrode with or without binder materials (Britto et al., 1996; Rubianes & Rivas 2003; Valentini et al., 2003; Wang & Musameh 2003a; Zare et al., 2010) (Wang & Musameh 2003b; Zhao et al., 2003). Britto et al. first reported biosensors based on CNT paste electrode by packing a paste of MWNTs with bromoform into a glass tube for dopamine detection, and the resulted paste electrode showed desirable electrochemical reversibility in cyclic voltammetry compared with conventional carbon electrodes (Britto et al., 1996). Enhanced amperometric response was also reported for CNT paste electrodes compared with carbon paste electrodes (Wang & Musameh 2003b).
3.2.5. Immobilization of aligned CNTs
As has been discussed previously, the catalytic activities of CNTs are mainly due to the carbon atoms at the end of the tubes, especially when tube ends are attached with oxygenated species (Chou et al., 2005; Gooding 2005; Koehne et al., 2003; Nugent et al., 2001). If the CNTs are perpendicular to the electrode surface, carbon atoms at tube ends will be sufficiently exposed and the catalytic activities of CNTs can be further increased. Almost none of the approaches reviewed previously had control over the orientation of CNTs. Take the polymer layer approach as an example, when CNTs are dispersed in the polymer layer, the orientations of CNTs are completely random. Huang et al. developed a method of preparing aligned CNT thin film on a quartz plate which can be easily transferred to other surfaces, such as electrode surface (Huang et al., 1999). Gao et al. developed a glucose biosensor based on this approach using gold electrodes (Gao et al., 2003). Increased glucose sensitivity was reported and a decreased irreversible single oxidation peak was observed compared with glassy carbon electrodes with no CNTs. Yun et al. developed a needle biosensor for H2O2 based on the same approach with modifications, and enhanced amperometric and voltammetric properties were observed (Yun et al., 2006). Wang et al. reported another CNT alignment approach by microwave plasma enhanced chemical vapor deposition using nickel as a catalyst (Wang et al., 2003c). Resulted CNTs grew densely and vertically along the grain of the catalytic particles (Yudasaka et al., 2009) and the CNTs were aligned straight by virtue of the nickel used, except for few entangled and cross-linked tubes. Wang et al. further reported that by controlling the thickness of the nickel layer, the diameter of aligned CNTs could be controlled (Wang et al., 2003c). Glucose biosensor based on this approach with direct absorption of glucose oxidase by MWNTs has been reported and more than 91% of the initial sensitivity towards glucose remained after three months, indicating good stability (Wang et al., 2003c). Liu et al. developed a self-assembled SWNT alignment method under room temperature, which was much easier to implement compared with Dai and Wang’s approaches requiring high temperature (Liu 2000). Derivative CNT alignment approaches have been reported (Chattopadhyay et al., 2001; Kim & Sigmund 2003).
3.2.6. Other surface modification approaches using CNTs
Other surface modification approaches using CNTs have been reported, including CNT-nanoelectrode ensembles (NEE) (Lin et al., 2004) and screen-printed CNT (Wang & Musameh 2004), which all provided biosensors with enhanced performance by virtue of the unique structure and properties of CNTs.
3.3. Combination of CNTs and metal nanomaterials
Transition metal nanomaterials possess high catalytic activities and facilitate electron transfer for many electrochemical reactions. Metal nanomaterials also enhance the performance of the biosensors by enlarging the effective surface area (Hrapovic et al., 2004). Biosensors incorporating metal nanomaterials, including platinum black (McLamore et al., 2010a; McLamore et al., 2010b; McLamore et al., 2011; Shi et al., 2010 ), copper (Xu et al., 2006), silver (Ren et al., 2005), palladium (Claussen et al., 2009; Lim et al., 2005) and gold (Daniel 2004) have exhibited well biocompatibility and enhanced performance. Especially the combination of metal nanomaterials and CNTs for surface modification of biosensors has proved to be feasible and more effective than using either nanomaterial alone (Evans et al., 2002; Hrapovic et al., 2004; McLamore et al., 2010a; McLamore et al., 2010b; McLamore et al., 2011; Shi et al., 2010 ; Zou et al., 2008). Increased H2O2 sensitivity and effective surface area have been demonstrated for Pt black/MWNT/Nafion electrodes over bare electrodes (Shi et al., 2010 ) (Fig. 3).
Due to the high hydrophobicity of CNTs, most metal nanomaterials would not attach to CNTs via physical absorption. Hrapovic et al. linked Pt nanoparticles to SWNTs using the charge interaction between Pt and nafion-suspended SWNT, where Pt was positively charged and nafion was negative. A uniform layer containing Pt-CNT was formed (Hrapovic et al., 2004) (Fig. 4a). Kang et al. reported glucose biosensors based on a uniform Pt-CNT-chitosan film because the amino group of chitosan facilitates the dissolving of both Pt nanoparticles and –COOH modified CNT. Pt nanoparticles and CNT were dispersed in chitosan sol–gel as shown in the TEM image (Kang et al., 2008) (Fig. 4b). Electrodeposition of Pt and Au nanoparticles on CNT modified electrodes using H2PtCl6 and HAuCl4 as Pt and Au source for glucose biosensing has also been reported, and SEM image showed that the porous MWNT film provided an ideal matrix for the distribution of Pt nanoparticles (Kang et al., 2007; Zou et al., 2008) (Fig. 4c).
The combination of metal nanomaterials and CNTs in biosensor surface modification integrates the catalytic capabilities of both nanomaterials and has been proved to be more effective in enhancing the biosensor’s performance than using either material alone, including amperometric response, detection limit, linear range and stability (Claussen et al., 2009; Claussen et al., 2010; Hrapovic et al., 2004; McLamore et al., 2010a; McLamore et al., 2010b; McLamore et al., 2011; Shi et al., 2010 ).
3.4. Attaching enzymes to nanomaterials
Various surface modification approaches based on CNTs and metal nanomaterials have been reviewed in the previous sections. A good question is how to immobilize enzymes on
nanomaterial modified electrodes, so that the biosensors have biorecognition capability while the catalytic activities of nanomaterials are preserved. Existing methods include direct absorption of enzymes by MWNTs due to the porous structure (McLamore et al., 2011; Shi et al., 2010 ), encapsulating enzymes and nanomaterials in the same polymer layer (Chen & Dong 2007; Choi et al., 2007a; Lim et al., 2005; Tsai et al., 2005), depositing multiple layers containing nanomaterials and enzymes (Zou et al., 2008) and attaching enzymes to modified electrodes via cross-linking agents (Claussen et al., 2009; Claussen et al., 2010). Gooding et al. reported a self-assembled attachment approach by incubating CNTs in microperoxidase MP-11 solution in HEPES buffer and showed that the enzymes were attached to the ends of the tubes via covalent bonds instead of being entrapped in the gaps among tubes (Gooding et al., 2003). They further demonstrated that no enzymes were linked to the side walls of CNTs with AFM showing that the number of CNTs was almost the same as the number of MP-11 enzymes (Gooding et al., 2003). Peak current of cyclic voltammetry in PBS for biosensors with MP-11 linked to SWNT was more than three times that of biosensors with no SWNT, demonstrating the electrochemical catalytic activities of CNTs (Gooding et al., 2003). Willner et al. reported similar approaches and linked the enzyme redox active center of flavin adenine dinucleotide (FAD) to the end of CNTs (Fernando et al., 2004). The attachment of enzyme redox center not only provided increased biosensing sensitivity due to enhanced electron transfer, but also allowed the direct electron transfer between enzymes and CNTs, which was the basis for third generation biosensors measuring direct electron transfer.
Attaching enzymes to nanomaterials facilitates the “electrical communication” between enzymes and nanomaterials, resulting in improved biosensor response. However, one potential drawback associated this technique is that the structure of enzymes may be changed due to the covalent bonds which link enzymes to CNTs. The catalytic activities of enzymes may be affected due to the structural change.
Almost all the literatures reviewed in this chapter reported biosensor performance in terms of amperometric and/or voltammetric response to the target compounds. Due to the large amount of existing surface modification approaches, problems with evaluating and comparing different approaches, and sorting out the optimal ones have arisen.
For biosensors, two important biophysical factors will affect amperometric response: 1.Enzyme activity. When covalent bonds exist between the enzymes and cross-liking agents, polymer layers, nanomaterials or the electrode surface, the structural change of enzymes will decrease the catalytic activities. 2. Diffusion properties of layer immobilized on the surface of the electrode, including polymers, enzymes and nanomaterials. The ideal case is that the layer is most permeable to target compounds, while most resistant to the interfering compounds that will otherwise generate interference. For surface modification approaches aimed at enhancing the performance of biosensors, such as the immobilization of CNTs, two factors should be considered as well: 1. Enlarged surface area. Both CNTs and metal nanomaterials can enlarge the effective surface area of the electrodes. 2. Enhanced electron transfer rate due to the catalytic ability of nanomaterials.
The complexity arising from the many factors affecting biosensing has posed great difficulty for comparing different surface modification approaches considering analyzing various configurations of membranes, the underlying connections and interactions among components within membranes. Consequently, certain standards should be set up to evaluate the performance of the biosensors. Amperometric response is the commonly used standard. However, different approaches are based on electrodes of different geometric shape, such as disk electrode, wire electrode, and needle electrode, with quite distinct effective surface area. Therefore, it is more reasonable to use the current density (current per surface area) to evaluate the performance of biosensors instead of amperometric response. Current density is defined as:
where n is the number of transferred electrons during the oxidation and reduction of potassium ferricyanide,
Ajayan P. M. 1999Nanotubes from Carbon. Chem. Rev, 1787 1800.
Azamian B. R. Davis J. J. Coleman K. S. Bagshaw C. B. Green M. L. H. 2002 Bioelectrochemical Single-Walled Carbon Nanotubes. 124 43(2002), 12664 12665.
Bard A. J. Faulkner L. R. 2000 Electrochemical methods : fundamentals and applications, (2nd), Wiley, 978-0-47105-542-6New York.
Branzoi V. Pilan L. 2008 Electropolymerization Compounds Used for the Obtainment of Modified Electrodes and Electrochemical Biosensors. 484(2008), 303 321.
Britto P. J. Santhanam K. S. V. Ajayan P. M. 1996 Carbon nanotube electrode for oxidation of dopamine. 41 1(1996), 121 125.
Cambiaso A. Delfino L. Grattarola M. Verreschi G. Ashworth D. Maines A. Vadgama P. 1996 Modelling and simulation of a diffusion limited glucose biosensor. B: Chemical, 33 1-3, (1996), 203 207.
Chattopadhyay D. Galeska I. Papadimitrakopoulos F. 2001 Metal-Assisted Organization of Shortened Carbon Nanotubes in Monolayer and Multilayer Forest Assemblies. 123 38(2001), 9451 9452.
Chen H. Dong S. 2007 Direct electrochemistry and electrocatalysis of horseradish peroxidase immobilized in sol-gel-derived ceramic-carbon nanotube nanocomposite film. 22 8(2007), 1811 1815.
Chen J. Hamon M. A. Hu H. Chen Y. Rao A. M. Eklund P. C. Haddon R. C. 1998 Solution Properties of Single-Walled Carbon Nanotubes. ., 282 5386(1998), 95
Choi H. N. Han J. H. Park J. A. Lee J. M. Lee W. Y. 2007a Amperometric Glucose Biosensor Based on Glucose Oxidase Encapsulated in Carbon Nanotube-Titania-Nafion Composite Film on Platinized Glassy Carbon Electrode. 19 17(2007a), 1757 1763.
Choi J. H. Nguyen F. T. Barone P. W. Heller D. A. Moll A. E. Patel D. Boppart S. A. Strano M. S. 2007b Multimodal Biomedical Imaging with Asymmetric Single-Walled Carbon Nanotube/Iron Oxide Nanoparticle Complexes. 7 4(2007b), 861 867.
Chou A. Bocking T. Singh N. K. Gooding J. J. 2005 Demonstration of the importance of oxygenated species at the ends of carbon nanotubes for their favourable electrochemical properties. 842 844.
Claussen J. C. Franklin A. D. ul Haque A. Porterfield D. M. Fisher T. S. 2009 Electrochemical Biosensor of Nanocube-Augmented Carbon Nanotube Networks. , 3 1(2009), 37 44.
Claussen J. C. Kim S. S. Haque A. U. Artiles M. S. Porterfield D. M. Fisher T. S. 2010Electrochemical Glucose Biosensor of Platinum Nanospheres Connected by Carbon Nanotubes. 4 2(2010), 312
Dai H. 2002Carbon nanotubes: opportunities and challenges. 500 1-3, (2002), 218 241.
Daniel M. C. 2004Gold nanoparticles: assembly, supramolecular chemistry, quantum-size-related properties, and applications toward biology, catalysis, and nanotechnology. 104 1(2004), 293
Dumont J. Fortier G. 1996Behavior of glucose oxidase immobilized in various electropolymerized thin films. 544 552.
Dumont J. Fortier G. 1996Behavior of Glucose Oxidase Immobilized in Various Electropolymerized Thin Films. 49(1996), 544 552.
Ekanayake E. M. I. M. Preethichandra D. M. G. Kaneto K. 2007Polypyrrole nanotube array sensor for enhanced adsorption of glucose oxidase in glucose biosensors. 23 1(2007), 107 113.
Evans S. A. G. Elliott J. M. Andrews L. M. Bartlett P. N. Doyle P. J. Denuault G. 2002Detection of Hydrogen Peroxide at Mesoporous Platinum Microelectrodes. 74 6(2002), 1322 1326.
Fernando P. Yossi W. Itamar W. 2004Long-Range Electrical Contacting of Redox Enzymes by SWCNT Connectors. 2113 2117.
Fortier G. Vaillancourt M. Belanger D. 1992Evaluation of nafion as media for glucose oxidase immobilization for the development of an amperometric glucose biosensor. 275 283.
Gao M. Dai L. Wallace G. 2003Biosensors Based on Aligned Carbon Nanotubes Coated with Inherently Conducting Polymers. 15 13(2003), 1089
Gavalas V. G. Law S. A. Christopher Ball. J. Andrews R. Bachas L. G. 2004Carbon nanotube aqueous sol-gel composites: enzyme-friendly platforms for the development of stable biosensors. 329 2(2004), 247
Gong K. Yan Y. Zhang M. Su L. Xiong S. Mao L. 2005Electrochemistry and Electroanalytical Applications of Carbon Nanotubes: A Review. 21(2005), 1383 1394.
Gong K. Zhang M. Yan Y. Su L. Mao L. Xiong S. Chen Y. 2004Sol-Gel-Derived Ceramic-Carbon Nanotube Nanocomposite Electrodes: Tunable Electrode Dimension and Potential Electrochemical Applications. 76 21(2004), 6500 6505.
Gooding J. J. 2005Nanostructuring electrodes with carbon nanotubes: A review on electrochemistry and applications for sensing. 50 15(2005), 3049 3060.
Gooding J. J. Wibowo R. Liu J. Yang W. Losic D. Orbons S. Mearns F. J. Shapter J. G. Hibbert D. B. 2003Protein electrochemistry using aligned carbon nanotube arrays. 125 30(2003), 9006 9007.
Guerrieri A. De Benedetto G. E. Palmisano F. Zambonin P. G. 1998Electrosynthesized non-conducting polymers as permselective membranes in amperometric enzyme electrodes: a glucose biosensor based on a co-crosslinked glucose oxidase/overoxidized polypyrrole bilayer. 13 1(1998), 103 112.
Heinze J. 1993Ultramicroelectrodes in Electrochemistry. 32 9(1993), 1268 1288.
Hrapovic S. Liu Y. Male K. B. Luong J. H. 2004Electrochemical biosensing platforms using platinum nanoparticles and carbon nanotubes. 76 4(2004), 1083 1088.
Huang S. Dai L. Mau A. W. H. 1999Patterned Growth and Contact Transfer of Well-Aligned Carbon Nanotube Films. 103 21(1999), 4223 4227.
Iijima S. 1991Helical microtubules of graphitic carbon. 354 6348(1991), 56 58.
Iijima S. Ichihashi T. 1993Single-shell carbon nanotubes of 1-nm diameter. 363 6430(1993), 603 605.
Kang X. Mai Z. Zou X. Cai P. Mo J. 2007A novel glucose biosensor based on immobilization of glucose oxidase in chitosan on a glassy carbon electrode modified with gold-platinum alloy nanoparticles/multiwall carbon nanotubes. 369 1(2007), 71 79.
Kang X. Mai Z. Zou X. Cai P. Mo J. 2008Glucose biosensors based on platinum nanoparticles-deposited carbon nanotubes in sol-gel chitosan/silica hybrid. 74 4(2008), 879 886.
Kim B. Sigmund W. M. 2003Self-Alignment of Shortened Multiwall Carbon Nanotubes on Polyelectrolyte Layers. 19 11(2003), 4848 4851.
Koehne J. Chen H. Li J. Cassell A. M. Ye Q. Ng H. T. Han J. Meyyappan M. 2003Ultrasensitive label-free DNA analysis using an electronic chip based on carbon nanotube nanoelectrode arrays. 14(2003), 1239
Li J. Cassell A. Delzeit L. Han J. Meyyappan M. 2002Novel Three-Dimensional Electrodes: Electrochemical Properties of Carbon Nanotube Ensembles. 106 36(2002), 9299 9305.
Li J. Tan S. N. Ge H. 1996Silica sol-gel immobilized amperometric biosensor for hydrogen peroxide. 335 1-2, (1996), 137 145.
Liaw H. W. Chen J. M. Tsai Y. C. 2006Development of an amperometric ethanol biosensor based on a multiwalled carbon nanotube-Nafion-alcohol dehydrogenase nanobiocomposite. 6 8(2006), 2396 2402.
Lim S. H. Wei J. Lin J. Li Q. Kua You. J. 2005A glucose biosensor based on electrodeposition of palladium nanoparticles and glucose oxidase onto Nafion-solubilized carbon nanotube electrode. 20 11(2005), 2341 2346.
Limbut W. Kanatharana P. Mattiasson B. Asawatreratanakul P. Thavarungkul P. 2006A comparative study of capacitive immunosensors based on self-assembled monolayers formed from thiourea, thioctic acid, and 3-mercaptopropionic acid. 22 2(2006), 233 240.
Lin Y. Lu F. Tu Y. Ren Z. 2004Glucose Biosensors Based on Carbon Nanotube Nanoelectrode Ensembles. 4 2(2004), 191 195.
Liu J. Chou A. Rahmat W. Paddon-Row M. N. Gooding J. J. 2005Achieving Direct Electrical Connection to Glucose Oxidase Using Aligned Single Walled Carbon Nanotube Arrays. 17 1(2005), 38 46.
Liu Z. Ziyong S. Tao Z. Shifeng H. Lizhen Y. 2000Organizing Single-Walled Carbon Nanotubes on Gold Using a Wet Chemical Self-Assembling Technique. Langmuir, 16(2000), 3569 3573.
Llaudet E. Hatz S. Droniou M. Dale N. 2005Microelectrode biosensor for real-time measurement of ATP in biological tissue. 77 10(2005), 3267 3273.
Ma Y. Ali S. R. Dodoo A. S. He H. 2006Enhanced Sensitivity for Biosensors: Multiple Functions of DNA-Wrapped Single-Walled Carbon Nanotubes in Self-Doped Polyaniline Nanocomposites. 110 33(2006), 16359 16365.
Malitesta C. Palmisano F. Torsi L. Zambonin P. G. 1990Glucose fast-response amperometric sensor based on glucose oxidase immobilized in an electropolymerized poly (o-phenylenediamine) film. 62 24(1990), 2735 2740.
Mamedov A. A. Kotov N. A. Prato M. Guldi D. M. Wicksted J. P. Hirsch A. 2002Molecular design of strong single-wall carbon nanotube/polyelectrolyte multilayer composites. 1 3(2002), 190 194.
Mc Lamore E. S. Diggs A. Calvo Marzal. P. Shi J. Blakeslee J. J. Peer W. A. Murphy A. S. Porterfield D. M. 2010aNon-invasive quantification of endogenous root auxin transport using an integrated flux microsensor technique. 63 6(2010a), 1004 1016.
Mc Lamore E. S. Mohanty S. Shi J. Claussen J. Jedlicka S. S. Rickus J. L. Porterfield D. M. 2010bA self-referencing glutamate biosensor for measuring real time neuronal glutamate flux. 189 1(2010b), 14 22.
(A self referencing platinum nanoparticle decorated enzyme-based microbiosensor for real time measurement of physiological glucose transport. Biosens. Bioelectron., Vol.In Press, Corrected Proof. Mc Lamore E. S. Shi J. Jaroch D. Claussen J. C. Uchida A. Jiang Y. Zhang W. Donkin S. S. Banks M. K. Buhman K. K. Teegarden D. Rickus J. L. Porterfield D. M.
Mc Lamore E. S. Shi J. Jaroch D. Claussen J. C. Uchida A. Jiang Y. Zhang W. Donkin S. S. Banks M. K. Buhman K. K. Teegarden D. Rickus J. L. Porterfield D. M. 2011A self referencing platinum nanoparticle decorated enzyme-based microbiosensor for real time measurement of physiological glucose transport. 26 5(2011), 2237 2245.
Miscoria S. A. Desbrieres J. Barrera G. D. Labb P. Rivas G. A. 2006Glucose biosensor based on the layer-by-layer self-assembling of glucose oxidase and chitosan derivatives on a thiolated gold surface. 578 2(2006), 137 144.
Ni J.-A. Ju H.-X. Chen H.-Y. Leech D. 1999Amperometric determination of epinephrine with an osmium complex and Nafion double-layer membrane modified electrode. Anal. Chim. Acta, 378 1-3, (1999), 151 157.
Nugent J. M. Santhanam K. S. V. Rubio A. Ajayan P. M. 2001Fast Electron Transfer Kinetics on Multiwalled Carbon Nanotube Microbundle Electrodes. 1 2(2001), 87 91.
Odom T. W. Huang J. L. Kim P. Lieber C. M. 1998Atomic structure and electronic properties of single-walled carbon nanotubes. 391 6662(1998), 62 64.
Ren X. Meng X. Chen D. Tang F. Jiao J. 2005Using silver nanoparticle to enhance current response of biosensor. 21 3(2005), 433 437.
Rickus J. L. Dunn B. Zink J. I. Frances S. L. Chris A. R. T. 2002Optically Based Sol-Gel Biosensor Materials. Optical Biosensors, 427 456. Elsevier Science, Amsterdam.
Rouse J. H. Lillehei P. T. 2002Electrostatic Assembly of Polymer/Single Walled Carbon Nanotube Multilayer Films. 3 1(2002), 59 62.
Rubianes M. D. Rivas G. A. 2003Carbon nanotubes paste electrode. 5 8(2003), 689 694.
Sadik O. A. 1999Bioaffinity Sensors Based on Conducting Polymers: A Short Review. 11 12(1999), 839 844.
Salimi A. Compton R. G. Hallaj R. 2004Glucose biosensor prepared by glucose oxidase encapsulated sol-gel and carbon-nanotube-modified basal plane pyrolytic graphite electrode. 333 1(2004), 49 56.
Schuhmann W. 1991Amperometric substrate determination in flow-injection systems with polypyrrole- enzyme electrodes. 4(1991), 41 49.
Schuhmann W. Kittsteiner-Eberle R. 1991Evaluation of polypyrrole/glucose oxidase electrodes in flow-injection systems for sucrose determination. 6 3(1991), 263 273.
Schuhmann W. Lammert R. Uhe B. Schmidt H. L. 1990Polypyrrole, a new possibility for covalent binding of oxidoreductases to electrode surfaces as a base for stable biosensors. 1 1-6, (1990), 537 541.
Shi J. Mc Lamore E. Jaroch D. Claussen J. Rickus J. Porterfield D. M. 2010(Oscillatory glucose flux in INS1 pancreatic [beta] cells: A self-referencing microbiosensor study. Anal. Biochem., 411, 185 193.
Situmorang M. Gooding J. J. Hibbert D. B. 1999Immobilisation of enzyme throughout a polytyramine matrix: a versatile procedure for fabricating biosensors. 394 2-3, (1999), 211 223.
Smutok O. Ngounou B. Pavlishko H. Gayda G. Gonchar M. Schuhmann W. 2006A reagentless bienzyme amperometric biosensor based on alcohol oxidase/peroxidase and an Os-complex modified electrodeposition paint. 113 2(2006), 590 598.
Star A. Stoddart J. F. Steuerman D. Diehl M. Boukai A. Wong E. W. Yang X. Chung S. W. Choi H. Heath J. R. 2001Preparation and Properties of Polymer-Wrapped Single-Walled Carbon Nanotubes. 40(2001), 1721 1725.
Teasdale P. R. Wallace G. G. 1993Molecular recognition using conducting polymers: basis of an electrochemical sensing technology桺lenary lecture. 118 4(1993), 329 334.
Tsai Y. C. Li S. C. Chen J. M. 2005Cast thin film biosensor design based on a Nafion backbone, a multiwalled carbon nanotube conduit, and a glucose oxidase function. 21 8(2005), 3653 3658.
Umana M. Waller J. 2002Protein-modified electrodes. The glucose oxidase/polypyrrole system. 58 14(2002), 2979 2983.
Vaillancourt M. Chen J. W. Fortier G. Beanger D. 1999Electrochemical and Enzymatic Studies of Electron Transfer Mediation by Ferrocene Derivatives with Nafion-Glucose Oxidase Electrodes. 23 31.
Valentini F. Amine A. Orlanducci S. Terranova M. L. Palleschi G. 2003Carbon Nanotube Purification: Preparation and Characterization of Carbon Nanotube Paste Electrodes. 75 20(2003), 5413 5421.
Verweij H. Melissa S. Ju L. 2007Fast Mass Transport Through Carbon Nanotube Membranes. 1996 2004.
Wang J. Pamidi P. V. A. 1997Sol-gel-derived gold composite electrodes, American Chemical Society Washington, DC, ETATS-UNIS.
Wang G. X. Ahn J.h. Yao J. Lindsay M. Liu H. K. Dou S. X. 2003aPreparation and characterization of carbon nanotubes for energy storage. 119121, (2003a), 16 23.
Wang J. 1999Amperometric biosensors for clinical and therapeutic drug monitoring: a review. 19 1-2, (1999), 1 2.
Wang J. Musameh M. 2003aCarbon Nanotube/Teflon Composite Electrochemical Sensors and Biosensors. 75 9(2003a), 2075 2079.
Wang J. Musameh M. 2003bEnzyme-dispersed carbon-nanotube electrodes: a needle microsensor for monitoring glucose. 128 11(2003b), 1382 1385.
Wang J. Musameh M. 2004Carbon nanotube screen-printed electrochemical sensors. 129 1(2004), 1 2.
Wang J. Musameh M. 2005Carbon-nanotubes doped polypyrrole glucose biosensor. 539 1-2, (2005), 209 213.
Wang J. Musameh M. Lin Y. 2003bSolubilization of carbon nanotubes by Nafion toward the preparation of amperometric biosensors. 125 9(2003b), 2408 2409.
Wang S. G. Zhang Q. Yoon S. F. Ahn J. 2003cSynthesis and characterization of MWNTs with narrow diameter over nickel catalyst by MPCVD. 48 4(2003c), 409
Wilder J. W. G. Venema L. C. Rinzler A. G. Smalley R. E. Dekker C. 1998Electronic structure of atomically resolved carbon nanotubes. 391 6662(1998), 59 62.
Xu Q. Zhao Y. Xu J. Z. Zhu J. J. 2006Preparation of functionalized copper nanoparticles and fabrication of a glucose sensor. 114 1(2006), 379 386.
Yang S. Lu Y. Atanossov P. Wilkins E. Long X. 1998Microfabricated glucose biosensor with glucose oxidase entrapped in sol-gel matrix. 47 3(1998), 735 743.
Yudasaka M. Kikuchi R. Ohki Y. Ota E. Yoshimura S. 2009Behavior of Ni in carbon nanotube nucleation. 70 14(2009), 1817 1818.
Yun Y. Bange A. Shanov V. N. Heineman W. R. Halsall H. B. Dong Z. Jazieh A. Yi T. Wong D. Pixley S. Behbehani M. Schulz M. J. 2006Fabrication and Characterization of a Multiwall Carbon Nanotube Needle Biosensor. Nanotechnology, 2006. IEEE-NANO 2006. Sixth IEEE Conference on, 280 283.
Zare H. R. Shishehbore M. R. Nematollahi D. Tehrani M. S. 2010Electrochemical behavior of nano-composite containing 4-hydroxy-2-(triphenylphosphonio)phenolate and multi-wall carbon nanotubes spiked in carbon paste and its application for electrocatalytic oxidation of hydrazine. 151 1(2010), 153 161.
Zeng B. Huang F. 2004Electrochemical behavior and determination of fluphenazine at multi-walled carbon nanotubes/(3-mercaptopropyl)trimethoxysilane bilayer modified gold electrodes. 64 2(2004), 380 386.
Zhang M. Smith A. Gorski W. 2004Carbon nanotube-chitosan system for electrochemical sensing based on dehydrogenase enzymes. 76(2004), 5045 5050.
Zhao Y.-D. Zhang W.-D. Chen H. Luo Q.-M. 2003Electrocatalytic oxidation of cysteine at carbon nanotube powder microelectrode and its detection. Sensors and Actuators B: Chemical, 92 3(2003), 279 285.
Zotti G. 1992Electrochemical sensors based on polyconjugated conducting polymers. 51 1-3, (1992), 373 382.
Zou Y. Xiang C. Sun L. X. Xu F. 2008Glucose biosensor based on electrodeposition of platinum nanoparticles onto carbon nanotubes and immobilizing enzyme with chitosan-SiO2 sol-gel. 23 7(2008), 1010 1016.