Development of a Modular Biosensor System for Rapid Pathogen Detection Development of a Modular Biosensor System for Rapid Pathogen Detection

Progress in the field of pathogen detection relies on at least one of the following three qualities: selectivity, speed, and cost-effectiveness. Here, we demonstrate a proof of concept for an optical biosensing system for the detection of the opportunistic human pathogen Pseudomonas aeruginosa while addressing the abovementioned traits through a modular design. The biosensor detects pathogen-specific quorum sensing molecules and generates a fluorescence signal via an intracellular amplifier. Using a tailored measurement device built from low-cost components, the image analysis software detected the presence of P. aeruginosa in 42 min of incubation. Due to its modular design, individual components can be optimized or modified to specifically detect a variety of different pathogens. This biosensor system represents a successful integration of synthetic biology with software and hardware engineering.


Introduction
A prerequisite for countermeasures against opportunistic pathogens is their rapid detection [1,2]. In contrast, conventional diagnostic methods often utilize time-consuming techniques such as microscopy and cultivation in different media [3] and bear the risk of false-positive

The five key elements of the proposed biosensor
The present work provides proof of concept for a novel approach toward a cost-efficient, optical biosensor, which enables safe and simple detection of pathogens and does not require highly trained staff for operation. The detection system was designed for investigation of solid surfaces, for example, to assess cleaning success in a hospital environment, which is receiving increasing interest [10]. This project was performed and has successfully competed in the International Genetically Engineered Machine (iGEM) competition 2014 [11].
The potential of the proposed system lies within the combination of biology and engineering as the development of biosensors is highly interdisciplinary [7]. Five key components, namely, biomolecular detection (I) with intracellular signal amplification (II) embedded into a twodimensional sensor chip (III), a custom incubation device (IV), and automated image analysis (V), constitute the functional biosensor as displayed in Figure 1. In terms of the biological component, the present project comprised the genetic engineering of sensor cells (introduction of the amplifying reporter circuit in Escherichia coli) as well as the optimization of the interface and immobilization of the resulting sensor cells. The transduction element (hardware), a customized detection unit, and image analysis software for automated evaluation were developed.
As a model organism for demonstrating the biosensor's functionality, the well-studied opportunistic pathogen Pseudomonas aeruginosa [12] was chosen as it has become a major cause of nosocomial infections; about 10% of nosocomial infections in most European Union hospitals are currently caused by P. aeruginosa alone [13]. Additionally, this bacterium often acquires multiple drug resistances and is a threat to patients suffering from cystic fibrosis, severe burns, or immunodeficiency [14].

Quorum sensing in Pseudomonas aeruginosa
Bacteria have evolved complex systems to sense their environment. Quorum sensing (QS) networks present a way to synchronize behavior, such as bioluminescence, biofilm formation, sporulation, and the secretion of virulence factors, on a population-wide scale [15].
In QS systems of bacteria, an autoinducer (AI) is produced by one or more synthases and is secreted from the cell. The cell can in turn detect the autoinducers through receptors in the Figure 1. The five key elements of the proposed sensor system. A biomolecular signal originating from pathogens in the sample is recognized (I), converted, and amplified (II) by sensor cells embedded in a two-dimensional sensor chip (III). The chip is incorporated in a detection device capable of real-time monitoring (IV) and equipped with software for an automated image analysis (V). In combination, the setup gives feedback to the user if pathogens were detected. cytosol (single-step response regulation in Gram-negative bacteria) or in the membrane (twostep response regulation in Gram-positive bacteria). Once a minimal threshold concentration is reached at higher cell densities, the activated AI receptors can induce or repress specific gene expression programs. The induction of the QS regulon leads to the expression of more AI synthase, amplifying the QS signaling [16]. However, most often the QS systems of one bacterial species extend beyond the basic circuit described above. Such configurations can include a multitude of circuits in parallel or series as well as competitive setups and on-off switches [17].
P. aeruginosa is commonly found in soil and is of particular interest due to its role in nosocomial infections. QS is essential for the persistence and disease progression, because it governs cell adhesion, biofilm formation, and virulence factor secretion [14]. The bacterium has three interconnected QS circuits: LasIR and RhlIR, two LuxIR-type circuits, and the Pseudomonas quinolone signal (PQS) system. In LasIR, the AI synthase LasI synthesizes the AI 3-oxo-C 12 -homoserine lactone (3OC 12 -HSL). LasR is a cytosolic receptor for 3OC 12 -HSL that acts as an inducer on the lasI promoter once bound to the AI. LasR is only stable in the complex with its matching AI, in this case 3OC 12 -HSL. However, LasR not only activates the expression of the Las regulon; it also acts as an inducer for the transcription of rhlR and rhlI, the receptor and AI synthetase, respectively, in the second LuxIR-type QS system of P. aeruginosa. The interaction between the LasIR and RhlIR systems is illustrated in Figure 2. The details of P. aeruginosa QS have been described in literature [17,18].
The implementation of the P. aeruginosa QS system in E. coli is already a well-established example for the use of such components in synthetic biology. Here, the LasIR circuit is used as a reporter system in E. coli to detect P. aeruginosa. The engineered E. coli cells constitutively express the protein LasR. Once 3OC 12 -HSL is secreted by P. aeruginosa cells, it diffuses into the E. coli cells and Figure 2. The LuxIR-type QS systems in P. aeruginosa and its translation into a biosensor. The AI synthase LasI (horizontal stripes) produces 3OC 12 -HSL (filled triangles) which bind to the transcription factor LasR (vertical stripes). The LasR-3OC 12 -HSL complex induces the expression of the Las regulon as well as rhlI and rhlR. RhlI (light shade) synthesizes the AI C 4 -HSL (open triangles) which in turn binds to RhlR (dark shade) and activates the expression of the Rhl regulon as well as the PQS system (not shown). RBS, ribosome binding site; CDS, coding sequence. binds to LasR. The LasR-3OC 12 -HSL complex then activates the reporter system, resulting in a fluorescent signal that can be read out by the detection device. However, the working principle of the biosensor is not limited to the detection of P. aeruginosa. Ultimately, the sensing E. coli cells can be engineered to include reporter circuits based on QS systems of other bacteria.

Molecular signal amplification
The biological component of the proposed biosensor was embodied by genetically modified E. coli, which were engineered to generate a fluorescence signal upon the presence of QS molecules specific for P. aeruginosa (specifically 3OC 12 -HSL). The core component of the sensor cells is the activation of a pool of quenched fluorophores, which will be discussed in detail later. Desired properties of the sensor cells were a rapid response, specificity, and high sensitivity [10].
The traditional way to report the binding of 3OC 12 -HSL to the constitutively expressed LasR would be the expression of a fluorescent protein, such as GFP, under the control of the lasI promoter. The presence of the autoinducer would then lead to a detectable fluorescent signal. A rapid generation of the signal, however, would be limited by transcription, translation, folding, posttranslational modification, and maturation of GFP. Therefore, a novel reporter strategy to accelerate the signal generation was chosen. In the proposed system, a quencher-linked GFP fusion protein is constitutively expressed in the cells, but does not exhibit fluorescence as long as the quencher subunit is in close proximity to the GFP subunit. Binding of 3OC 12 -HSL to LasR induces the expression of a tobacco etch virus (TEV) protease, which cleaves the fusion protein. Thereby, GFP is released from the quencher and emits a fluorescence signal. Compared to the conventional approach, the signal is generated faster by maintaining a stock of fusion proteins in the cells, which can be readily cleaved. Additional signal amplification is achieved by the ability of a single TEV protease to cut multiple fusion proteins, while expression of a fluorescent protein upon the presence of 3OC 12 -HSL would only result in a single fluorescent molecule at a time.

Quenching of GFP fluorescence
The quenching of GFP fluorescence in the fusion protein is based on Förster resonance energy transfer (FRET), a process by which the energy of an excited donor fluorophore is transferred to an acceptor molecule whose absorption spectrum overlaps with the emission spectrum of the donor [19]. The energy can then be released, for example, by fluorescence of a longer wavelength or by heat. Yellow fluorescent protein (YFP) represents a suitable FRET acceptor for GFP. Emission resulting from YFP was avoided by using a nonfluorescent mutant of YFP called resonance energy-accepting chromoprotein (REACh [20]). Two REACh variants were generated by introducing the mutation Y145W (REACh1) and the double mutation Y145W/H148 (REACh2) into an enhanced YFP (eYFP) by QuikChange mutagenesis. Ganesan et al. [20] reported a reduction in fluorescence of 82 and 98% for REACh1 and REACh2, respectively.
Both REACh variants were genetically fused to GFP (mut3b [21]) via a linker, which brings both proteins in close proximity, facilitating FRET [22] from GFP to REACh, thus quenching the fluorescence. The linker harbors a cleavage site for the TEV protease (ENLYFQ\S) allowing the separation from the quencher. In the present study, the TEV protease is expressed under control of the lasI promoter, making it inducible by the QS autoinducer 3OC 12 -HSL. For this purpose, a TEV protease gene with codon optimization for E. coli and the anti-self-cleavage mutation S219 V was designed [23]. The GFP-REACh fusion protein is expressed constitutively to ensure continuous supply of protease substrate. Figure 3 illustrates the interplay between the GFP-REACh fusion protein and the TEV protease. The expression cassette for the GFP-REACh fusion protein was cloned into a pSB3K3 [24] vector backbone, and the TEV protease expression cassette was inserted into a pSB1C3 [25] vector.

Validation of the reporter system
For initial validation the developed reporter system was tested via β-D-1-thiogalactopyranoside (IPTG) induction using a well-characterized T7 promoter instead of the lasI promoter. Two plasmids, one carrying the GFP-REACh fusion protein and one carrying the TEV protease, were introduced into E. coli BL21 (DE3). The resulting strain allowed the IPTG-inducible expression of the fusion protein. A growth experiment was conducted in which the fluorescence of the double plasmid strains, containing either variant of the fusion protein and the TEV protease, was compared to cells constitutively expressing GFP as positive control and a nonfluorescent strain as negative control (Figure 4, left). For both REACh variants, IPTGinduced as well as IPTG-non-induced cultures were grown in parallel, and all measurements were done in a biological triplicate. The fluorescence was normalized to the observed optical density (OD). The induction with IPTG leads to a rapid increase of the fluorescence signal. At the end point, a signal strength comparable to the positive control was reached, indicating a complete cleavage of the fusion proteins by the TEV protease. The higher base level of fluorescence in the non-induced cells can be attributed to imperfect quenching. This experiment demonstrated the quenching ability of the REACh proteins in our fusion constructs as well as the functionality of the E. coli-produced TEV protease. To test the hypothesis that the GFP-REACh fusion proteins in combination with the cleavage amplification results in a faster response than the conventional approach, the kinetics of our reporter strategy were compared to a strain expressing GFP under the control of an IPTGinducible lacI promoter. Using the new reporter strategy, a much stronger and faster increase in fluorescence was observed compared to IPTG-induced expression of GFP (Figure 4, right). The high variation for the development of fluorescence by the GFP-REACh systems may have originated from inhomogeneous expression levels of the TEV protease and different sizes of the fluorophore pools. As the signal is amplified by the cleavage of the GFP-REACh fusion protein by the TEV protease, even slight temporal differences in the expression of the TEV protease are expected to cause great shifts in the temporal signal responses, thus resulting in high error bars when different cultures are averaged. The errors were increased even further with the Gaussian error propagation.

Immobilization of sensor cells
The sensor cells were immobilized in rectangular layers (chips), thus creating an interface between the biological component and the technical component (transducer). Main objectives during the design of the interface were to enable viability and storability of the immobilized sensor cells, reproducibility of the fluorescence response, as well as cost-efficiency. For proof of concept, a simple and robust design was chosen.
A variety of different methods have been used for immobilization of whole cells, which can be divided into six general types: covalent coupling, affinity immobilization, adsorption, confinement in liquid-liquid emulsion, capture behind semipermeable membranes, and entrapment [26]. An established technique for immobilization of living cells is entrapment, which refers to the physical containment of organisms inside a matrix or fibers, thus creating a protective barrier around the cells [27]. Matrices used for entrapment can be synthetic polymers, such as polyester, or natural polymers, such as agar, agarose, or alginate [27]. Entrapment allows to preserve and prolong cell viability, for example, during storage [26,27], which matched the intentions of this work.
Important prerequisites for the entrapment matrix of the sensor cells were physical rigidity, safety, resistance against biological degradation, transparency, as well as the possibility to conduct matrix synthesis at mild conditions, suitable for living cells. Inorganic polymers such as polyacrylamide were ruled out due to the carcinogenicity of the monomers and rather harsh polymerization conditions [28]. Natural polymers allow for higher diffusion rates than inorganic polymers (tested for small molecules [28]) and are less expensive and less hazardous in production than synthetic polymers. The organic polymer agarose offers several advantages including easy handling, resistance to microbial degradation, and favorable conditions for entrapped cells [27]. Thus, agarose was the polymer of choice for immobilization of cells and formation of chips.

Optimization of chip casting mold and medium
First, a casting mold for rapid and reproducible manufacturing of the 2D sensor chip was developed. A plain surface was a prerequisite for automated image evaluation. Low agarose concentrations (<3.0%) were chosen to reduce consumable costs and to ensure rapid diffusion of the analyte (HSL) to the immobilized sensor cells.
Manufacturing of the agarose gel was conducted based on existing protocols for entrapping living cells in melted polymers. In brief, the temperature of the polymer solution was adjusted to 45°C and was quickly poured into the respective mold after mixing with the sensor cells. Sensor cells were spun down from a liquid culture (50 mL LB, 5 g•L −1 NaCl, 10 g•L −1 tryptone, 5 g•L −1 yeast extract) and resuspended in 1 mL LB medium (21°C) before mixing with the temperature-adjusted agarose solution, resulting in a cell concentration of approximately 5.6×10 9 cells/mL. Before usage, solidified and cutout sensor chips were incubated for 1 h at 37°C.
An open casting mold, which exploited the surface tension of the polymer solution to achieve a plain chip surface, was most successful for the production of sensor chips. After discarding a small gel area in direct contact with the edges of the mold (Figure 5, left), bubble-free sensor chips with a plain surface were readily obtained from this approach. The open mold allowed for simple, reproducible, and rapid manufacturing of sensor chips and was hence the method of choice for this work. An agarose concentration of 1.5% was found to be sufficient to cast robust sensor chips. For an accelerated manufacturing process, multiple sensor chips were casted simultaneously using an extended mold (Figure 5, left).
Further, to meet the nutritional needs of the sensor cells while minimizing background fluorescence, different complex media (Luria-Bertani or LB medium, Terrific-Broth or TB medium, nutrient agar or NA medium) as well as minimal media (Hartmans minimal or HM medium, M9 minimal medium) were tested with respect to sensor cell growth and the presence of background fluorescence. Background fluorescence was investigated in a commercial gel imaging system (GelDoc™ XR, Biorad, Germany) as well as in the custommade optical detection device constructed in this work as described in the following section. The results are summarized in Table 1, and a comparison of the background fluorescence of sensor chips comprising the respective media is displayed in Figure 5 (right). Only LB medium allowed for sufficient growth of the sensor cells. Its background fluorescence in the custom-made optical detection device was acceptable, most likely due to the narrow excitation profile compared to the commercial device.
Background fluorescence appeared to be more intense in complex media than in minimal media. To identify a possible cause for this observation, minimal M9 medium was supplemented with 2% Casamino acids (Figure 5, right, bottom row). Background fluorescence was stronger in supplemented minimal medium matching reports in literature [29], possibly due to an increased concentration of aromatic amino acids possessing inherent fluorescence.
Activity of the sensor cells after immobilization was investigated in a subsequent experiment by inducing a fluorescent signal with 0.2 μL of a 500 μg•mL −1 HSL (3-oxo-C 12 ) solution ( Figure 6A).  One and a half hours post induction, a fluorescence signal was visible even to the naked eye, indicating that the sensor cells were in fact still viable after immobilization. No apparent change in fluorescence was observable for the negative control ( Figure 6B). Additional experiments were carried out to investigate the biosafety of the proposed sensor chips, because a release of the genetically modified sensor cells from the sensor chips represented a possible risk in handling. A simple approach for investigating the biosafety of the sensor chips was replica plating on agar plates containing the respective antibiotic. An average of five colony-forming units (CfU) was found (n = 3), indicating that some cells were in fact able to escape the agarose entrapment. Therefore, measures to achieve a complete entrapment, for example by increasing the agarose concentration, should be evaluated to render the system as safe for the use in non-GMO-certified areas.

Integrated cultivation and detection device
The two-dimensional approach of sensing pathogens on agarose chips requires a specialized device for detecting and interpreting the fluorescent signals generated by the immobilized sensor strain. Since the results from commercially available plate readers and gel imaging systems did not yield a sufficient spatial resolution, a custom-made device was designed and constructed as pictured in Figure 7 (left). The device consists of two enclosed compartments, separated by laser-cut plates of acrylic glass. The inner compartment serves for cultivation and illumination of the sensor chip. The outer compartment contains a Raspberry Pi microcomputer, an Arduino microcontroller, and a camera for imaging. Figure 7 (right) schematically shows the individual components of the device and their interaction.
Once the chip is prepared and a sample taken, a petri dish containing the chip is inserted into the inner compartment, which serves as an in situ incubation chamber for both pathogens and genetically modified sensor organisms. A UV lamp could be integrated to facilitate built-in inactivation of microorganisms.
During the experiment, the parameters are controlled by an Arduino Uno and a Raspberry Pi. The Arduino has two main functions: first, it is responsible for controlling the incubation temperature in the inner compartment. Based on measurements from the temperature sensor, it sets the power input for the Peltier elements, thus heating or cooling the interior of the device. Second, the Arduino controls the LEDs illuminating the chip. When a control command from the Raspberry Pi is received, the two channels of the connected relay are turned on or off, switching the state of the LEDs, respectively. Thus, the chip is exposed to the specific wavelength emitted by the LEDs, in this case 480 nm for the excitation of the unquenched GFP.
Upon user input, the Raspberry Pi triggers the camera module to take an image of the chip. A filter slide is placed in front of the lens to block the excitation wavelength from the LEDs and to specifically transmit the emission wavelength of the fluorophore. In this configuration, a highly resolved fluorescent signal is obtained. The image is further processed by the Raspberry Pi and displays the analysis results via the graphical user interface (GUI) on a built-in 7-inch display located in the outer casing. The GUI (Figure 8, left) runs on either the Raspberry Pi or an externally connected computer; it enables the user to adjust the camera settings, take a single image or start time-lapse imaging, and to monitor the imaging process. Moreover, it allows execution of the analysis software for saved images as described in detail below. The communication between the GUI and the hardware is ensured by the backend software. It receives the respective commands (e.g., for capturing an image) from the GUI and subsequently forwards them to the according hardware. Therefore, the backend is responsible for image acquisition. An exemplary chain of commands for taking an image is depicted in Figure 8 (right). The backend runs on the Raspberry Pi.
For the detection of P. aeruginosa using the sensor system presented in the previous sections, time-lapse imaging was performed, taking pictures in intervals of 5 min. The agarose chip was incubated at 37°C and excited with four LEDs (Superflux LED blue 3 lm NSPBR70BSS-PU/ PV-W, Nichia Corporation) emitting a peak wavelength of 480 nm. The filter "010 Medium Yellow" (LEE Filters) was installed in front of the camera to allow the emission wavelengths of the fluorophore to pass while blocking the peak wavelength of the LEDs.

Analysis of spatial fluorescence
Automated, fast, and reliable analysis of raw sensor data is critical for a diagnostic device. Since, in the case of the 2D biosensor, the raw sensor measurement is a series of pictures taken by the onboard camera, an image analysis pipeline is required. Here, a novel pipeline is presented involving segmentation through statistical region merging (SRM [30]), thresholding in hue-saturation-value (HSV)-color space, and a final classification step. This leads to segmentation of the fluorescent regions in the biosensor chip, thus identifying chips or chip regions containing pathogens.

Image segmentation
Onboard image analysis on embedded computing hardware is subject to rigorous performance constraints due to the poor availability of existing analysis packages and the limited computing power. This complicates the use of sophisticated analysis pipelines. At the same time, the need for quantification of fluorescent regions on the biosensor mandates the image to be segmented into foreground (fluorescent) and background regions, also called super-pixels. This is necessary because only after a segmentation mask is computed for an input image, the number of independent fluorescent regions in the image, their intensity, and their area can be quantified. Statistical region merging is an image segmentation algorithm which is both light-weight and does not require expensive tuning of algorithm-specific hyperparameters [30]. In contrast to other clustering algorithms, it also produces deterministic results, which increases the reproducibility of the analysis pipeline. The SRM algorithm has one important hyperparameter Q which influences the merging process. A Q-level of 256 resulting in many fine regions was chosen (Figure 9, top-left).
The input image (Figure 10A) is segmented into super-pixels, and the list of regions is filtered to obtain only candidate regions of fluorescence ( Figure 10B). Since the color of the fluorescence signal is known, the regions can be thresholded based on their HSV color representation. For selection of GFP-fluorescent regions, super-pixels that have hue (color shade) in the interval [0.462, 0.520], saturation of 0.99, and value (brightness) in the interval [0.25, 0.32] were considered. This thresholding step removes background regions and is performed at low computational cost ( Figure 10C).
Since false positives can remain after filtering, they are removed from the list of candidate regions by classifying each region into noise or signal. First, the classification applies a smoothing procedure to the region mask. This is achieved by convolving the region mask with a disk filter ( Figure 10D). Then, for each pixel p′ in that smoothed image, the smoothness index [31] is calculated (Figure 10E) as the sum of the difference with respect to each of its neighbors N k (Eq. (1)). In the implementation, the neighbors in a radius of i=4px were used. Finally, the matrix of smoothness indices is normalized the interval [0, 1]: A subsequent thresholding step selects pixels that fulfill S p ≥ T S ∧ I p ≥ 255 where S p denotes the smoothness index at pixel p, T s =0.85 is an empirically determined smoothness threshold, and I p is the intensity of the pixel in the smoothed mask. The final classification step removes regions with high-edge curvature and selects smooth, blob-like regions ( Figure 10F). Thereby, artifacts are removed from the analysis, and only fluorescent pixels are quantified.

Quantification of the fluorescence signal
The image analysis pipeline outlined above was implemented in both MATLAB and C++. It allowed the detection of fluorescence with little tuning of hyperparameters (Q-level and thresholds). When a time-lapse of images is automatically acquired with the software, the image analysis pipeline can be applied to each frame (Figure 11, top). The number of pixels in the resulting foreground regions can be quantified over time. After about 40 min of incubation, a region of pathogen-induced fluorescence was detected, which then grew linearly with respect to its area over time (Figure 11, bottom). As a proof of principle, 6×10 5 cells were applied on the chip. However, serial dilution testing is needed to determine the lower detection limit. The expected lower limit of detection is 1 CFU due to cell proliferation during the incubation step.

Discussion and outlook
In this work, a modular biosensor for the detection of the opportunistic human pathogen P. aeruginosa was developed. Five key components, (2.1) a selective molecular detection mechanism, (2.2) an integrated amplification step, (2.3) a gentle immobilization technique, (2.4) a low-cost cultivation and optical detection device, and (2.5) a graphical analysis software, were integrated. The resulting modular biosensor demonstrates the power of combining synthetic biology with software and hardware engineering by detecting P. aeruginosa in less than 1 h of analysis time. Table 2 provides a comparison of the sensor system developed in this study to existing detection methods for P. aeruginosa.
In addition to the detection methods compared in Table 2, there are several whole-cell approaches. Most of the previously developed whole-cell biosensors deliver an optical output [39]. In a previous work, Struss et al. developed a whole-cell biosensor detecting AHLs of gram-negative bacteria, particularly P. aeruginosa [40]. Similar to the approach presented herein, they used components of the AHL-mediated QS regulatory system to generate an optical signal. A portable format was developed by liquid-drying the sensor cells on filter paper strips. While Struss et al. met many criteria for a successful portable on-site field kit, such as easy handling, inexpensiveness, and simple transportation, it lacks a rapid, integrated analysis and is dependent on the user's subjective evaluation.
Enhancement and optimization of the proposed biosensor system beyond the proof of principle demonstrated in this work can be realized by modifying each of the five key elements as well as their interactions. The individual key elements can be optimized as follows.
The utilization of the pathogen's inherent QS system guarantees a high specificity as the receptor for the AI is unique. However, this poses a challenge if multiple pathogens are desired to be detected simultaneously. First, only QS molecules can be recognized by a molecular sensing system of the presented type. In theory, other secreted compounds can be used for detection, though potentially reducing the specificity. Second, the sensing system should be introduced into a separate sensing organism to completely avoid interaction, especially if a closely related QS system and a signal amplification as presented here are utilized. This may lead to insufficient spatial resolution as many different sensing cells are required to be incorporated in the same sensor chips. An equal distribution of each type of sensing cell needs to be ensured and reciprocal interference avoided. The feasibility hereof has already been proven in previous work [41].
By introducing the REACh quenching system, the fluorescence response was amplified and accelerated compared to conventional GFP expression. Quenched fluorophores are constitutively expressed, and a constant pool of reporter molecules is built up. Upon the presence of inducers and a subsequent expression of the protease, they are unquenched resulting in a fast and strong fluorescent signal. Since the two expression cassettes for the GFP-REACh fusion protein and the TEV protease are currently on two separate plasmids, using a single plasmid would increase the robustness of the detection system, as two plasmid expression systems are considered less stable. As a proof of principle, the system was tested using IPTG-induced expression of the TEV protease. As a next step, the system would be adjusted by exchanging the T7 promoter with the HSL-bound LasR-inducible lasI promoter to render the expression of the TEV protease inducible by 3OC 12 -HSL. Subsequently, extensive testing with different concentrations of 3OC 12 -HSL and varying cell numbers of P. aeruginosa should be performed to determine detection limits. Based on the results, the expression can be fine-tuned, for example, by improving the promoters. On the protein level, the linker length between GFP and REACh can be optimized with respect to the protein folding, protease accessibility, and quenching efficiency [22].
Engineering of the agarose chips for entrapment of the sensor cells represents a simple yet efficient way for a two-dimensional detection method. The immobilized sensor cells survived and still performed as expected, even after short-term storage at 4°C. A fluorescence signal was generated upon induction, thus proving a sufficient diffusion of the inducer through the chip. As discussed above, adjustment of the agarose concentration used for production of the sensor chips represents a simple way to further optimize the sensor chips. Increasing the agarose concentration could focus the fluorescent response on a smaller area by restricting diffusion of the analyte, however, under the prerequisite that the diffusion is fast enough to reach the sensor cells within a short time. Additionally, adjustment of the agarose concentration affects the biosafety as the ability of the chip to contain the sensor cells is altered. To ensure a sufficient quantity and spread of the cells, an array-based technique for pattering the sensor cells onto a chip surface could be used to enable high-throughput analysis [41]. Several techniques for printing bacteria on surfaces have already been used successfully [42,43].
The optical detection device represents a simple and cost-effective solution for the rapid visualization and analysis of the 2D fluorescent signal. In situ cultivation with automatic, real-time monitoring of the fluorescence resulted in the detection of P. aeruginosa within 42 min, even without using the optimized sensor cells. Compartmentation and the possibility to install a UV sterilization light ensures a high standard of biosafety. The settings described in Section 2.4 are highly specific for the presented two-dimensional biosensor; however, as the device is modular, single components such as the LEDs and the filter sets can be exchanged to adjust the optical settings to different reporter systems. An extension of the device, for example, by using a filter wheel or a monochromator and LEDs emitting different wavelengths bears the potential of simultaneously detecting various pathogens if respective molecular reporter systems can be constructed, thus allowing a high degree of multiplexing. The extensive modularity and the inexpensive parts in comparison to common commercial devices grant an easy access for potential users and researchers customizing the system for other biosensors.
The analysis software pipeline recognized and distinguished fluorescent signals of certain shapes and marked them for an easy interpretation by the user. However, the lack of sufficient amounts of real input data may imply a subjectivity of the analysis. Further testing needs to be done to prove universal applicability. In this regard, the precision vs. recall trade-off of the software is required to be further investigated to determine ratios between false positives and false negatives. Additionally, time-lapse data should be featured not only in the GUI but in the analysis as well. Since the project was conducted, the computational capabilities of embedded hardware have dramatically improved. Future adoptions of this work should therefore utilize state-of-the-art embedded hardware and software packages.
In general, the presented biosensor represents a proof of concept of a modular whole-cell, point-of-contact biosensing system. It enables rapid and inexpensive detection of P. aeruginosa, providing intuitive feedback through integrated, real-time analysis. The applicability of this sensor platform in other fields, such as food, water, and environmental safety, offers further innovation potential.