Thermal conductivities and spin destruction rates for Xe and N2/He buffer gases used in the SEOP hyperpolarization process .
Spin-exchange optical pumping (SEOP) can be used to “hyperpolarize” 129Xe for human lung MRI. SEOP involves transfer of angular momentum from light to an alkali metal (Rb) vapor, and then onto 129Xe nuclear spins during collisions; collisions between excited Rb and N2 ensure that incident optical energy is nonradiatively converted into heat. However, because variables that govern SEOP are temperature-dependent, the excess heat can complicate efforts to maximize spin polarization—particularly at high laser fluxes and xenon densities. Ultra-low frequency Raman spectroscopy may be used to perform in situ gas temperature measurements to investigate the interplay of energy thermalization and SEOP dynamics. Experimental configurations include an “orthogonal” pump-and-probe design and a newer “inline” design (with source and detector on the same axis) that has provided a >20-fold improvement in SNR. The relationship between 129Xe polarization and the spatiotemporal distribution of N2 rotational temperatures has been investigated as a function of incident laser flux, exterior cell temperature, and gas composition. Significantly elevated gas temperatures have been observed—hundreds of degrees hotter than exterior cell surfaces—and variances with position and time can indicate underlying energy transport, convection, and Rb mass-transport processes that, if not controlled, can negatively impact 129Xe hyperpolarization.
- low-frequency Raman
- spin-exchange optical pumping
- xenon NMR/MRI
- low-field NMR
- remote temperature measurement
Magnetic resonance imaging (MRI) is a powerful tool for diagnostic imaging of soft tissue, combining superior contrast, dynamic capability, and no risk of damage caused by ionizing radiation compared to computerized tomography (CT) . Most MRI techniques conveniently detect the nuclear spin of protons in water, which makes up a significant portion of the human body. This detection modality poses a problem within the lungs, however, as the low proton density translates to greatly reduced signal-to-noise ratios for conventional MRI techniques. Even at high magnetic field strengths, the equilibrium nuclear spin polarization is very low (~10−4 to 10−6). To facilitate MRI inside the lungs, far greater signal-to-noise ratios are needed, requiring nuclear spin polarizations to be increased far above equilibrium levels—a process known as hyperpolarization. Indeed, inhalation of hyperpolarized gases provides greatly improved sensitivity and bright images, outweighing the low spin densities of the gas phase. Gas hyperpolarization can now be readily achieved using established techniques; however in these approaches, maximizing the available spin polarization is hampered by complex system dynamics and codependence of many variables that are heavily temperature-dependent.
In most studies relevant to lung imaging, this hyperpolarized state is achieved by utilizing circularly polarized photons to shift particles with a nonzero magnetic moment between discrete spin states. During gas-phase collisions, spin angular momentum is transferred from alkali metal electrons to noble gas nuclei, a technique known as spin-exchange optical pumping (SEOP) . Noble gases are used in SEOP because of their lack of reactivity with alkali metal vapors and their ultra-long hyperpolarization lifetimes. Nitrogen is present as a buffer gas to prevent radiation trapping, helping to maximize the alkali metal electron spin polarization—and hence, the noble gas nuclear spin polarization. Thus, inhalation of a hyperpolarized noble gas sample such as 129Xe while inside the MRI scanner enables both structural and functional high-resolution imaging of the lungs in real time. Despite significant advances in polarizer technology in recent years, many underlying aspects of the SEOP process, such as energy- and mass transport mechanisms within the OP cell and the temperature dependences of many parameters, remain poorly understood. Ironically, the desire to maximize the available nuclear spin magnetization leads to a demand for higher pump laser powers and richer noble gas mixtures, which only serve to compound this issue.
In an attempt to overcome these challenges, Raman spectroscopy may be used to complement more commonly used techniques such as optical absorption spectroscopy and low-field NMR in order to optimize the multidimensional SEOP parameter space, thereby improving the 129Xe hyperpolarization process for MR applications. Conventional temperature measurement (e.g., using a thermocouple) would be challenging to implement, given that the SEOP process takes place inside a sealed vessel, within a magnetic field, and under constant uniform illumination by high-power laser light; moreover, temperature measurement of the exterior cell walls may often provide only a poor reflection of the true gas behavior within the cell. However, if optical access is available, Raman spectroscopy can be used to acquire
2.1. Spin-exchange optical pumping
The SEOP process consists of two major steps. Firstly, circularly polarized photons (necessary to satisfy the selection rules for optical absorption) are used to transfer angular momentum to a gaseous target with nonzero spin, such as spin-½ electrons. A vaporized alkali metal such as rubidium is typically used, since its single outer shell electron is easier to manipulate and existing laser technology at the required transition wavelengths is well developed. The net polarization,
The second stage in the SEOP process is that of spin-exchange. Bringing an unpolarized gaseous system into contact with the polarized alkali metal vapor results in spin-coupling via a hyperfine interaction. Gas phase collisions then lead to transfer of polarization from alkali metal electrons to noble gas nuclei on short timescales. The net spin polarization of the noble gas during the spin-exchange process is calculated using 
As previously mentioned, temperature is one of the most important variables governing the SEOP process. Elevated temperatures raise the alkali metal vapor density within the cell, increasing the probability of collisions between pump laser photons and alkali metal electrons and more importantly resulting in increased spin-exchange rates. As evidenced by Eq. (2), the latter effect should lead to a greater net spin polarization of the noble gas, provided there is sufficient pump light intensity to maintain good illumination of the optical cell (and hence, good alkali metal polarization). This prediction holds to a certain extent—as temperature continues to rise however, the alkali metal number density can increase rapidly in a self-propagating fashion; the alkali metal vapor absorbs more light, whose energy is rapidly converted to heat, which can in turn lead to even more alkali metal vaporization. This results in the form of optical opacity, whereby pump laser light is unable to penetrate the length of the optical cell. This actually reduces the efficiency of optical pumping in the areas of the cell furthest from the pump laser, because although the spin-exchange rate may be quite high, the alkali metal polarization in much of the cell is low, greatly reducing noble gas polarization. This is known as a “runaway” process [5, 6], and is clearly detrimental to a stable, efficient SEOP process. In the interest of obtaining the highest possible NMR signal, it is desirable to conduct experiments at an optimal temperature where net spin polarization and build-up rates are maximized, while avoiding the unstable runaway regime.
2.2. Temperature measurement using Raman spectroscopy
As mentioned above, the presence of nitrogen gas inside the optical cell is primarily meant to quench the rate of radiative spin-destruction, achieved by collisions with electronically excited alkali metal atoms. The energy transferred as a result of these collisions is pooled in the rotational and vibrational modes of the N2 molecules; these modes quickly relax to the translational degrees of freedom, thereby increasing the local gas temperature inside of the cell. Since SEOP must take place inside a closed system due to the high reactivity of alkali metals in air, physical insertion of a thermocouple is impractical for the reasons listed above, and merely measuring the cell surface temperature does not provide a true account of the internal temperature, nor the corresponding energy transport processes occurring within the cell. Additionally, progressively stronger light sources have been utilized for SEOP over the years, and lasers emitting tens or hundreds of watts of energy are now standard . Virtually all of the laser energy absorbed by Rb is transferred to the rotational and vibrational degrees of freedom in the N2 buffer gas, which then rapidly equilibrates with the translational temperature (corresponding to the local temperature of the gas mixture). These changes in temperature are capable of significantly affecting SEOP (hence, xenon polarization) through changes to the alkali metal density and absorption profile , degradation of organic coatings on the OP cell surface , and convective gas transport that may bring xenon in closer proximity to paramagnetic relaxation centers in the cell wall surface [4, 10, 11]; changes to temperature-dependent cross-sections that govern polarization and depolarization rates in SEOP may also occur. Remote sensing of the N2 rotational (and vibrational) temperatures inside the pump cell during SEOP can be achieved using
Hickman et al.  demonstrated that the intensity of each peak, numbered
A plot of against
2.3. Hyperpolarized lung imaging
Any noble gas isotope with nonzero spin can theoretically be used as a hyperpolarized contrast agent with a view for clinical lung MRI. Traditionally, helium-3 was most commonly used due to its high gyromagnetic ratio—resulting in stronger MRI signals. However, due to its insolubility in blood and water, perfusion across the alveolar wall cannot occur, primarily limiting the use of 3He to gas-phase ventilation and diffusion imaging. The difficulty in wide-spread clinical 3He adoption is further compounded by the difficulty and expense of acquisition, since 3He is a nonrenewable by-product of tritium decay in nuclear reactors. As such, most current clinical studies on hyperpolarized noble gases use 129Xe for its solubility in tissue and blood, significant chemical shift range, low cost of acquisition, and high natural abundance.
As discussed previously, higher net spin polarizations directly result in stronger MR signal generation and hence, improved image contrast between areas of interest and background noise. Recent hyperpolarization methods can create near-unity net spin polarizations , resulting in spectroscopic signals four to five orders of magnitude greater than nonhyperpolarized samples. Such improvements in image quality facilitate the use of HP noble gas MRI within the lungs as a diagnostic imaging tool for characterizing lung structure and function , and consequently may potentially enable earlier and more reliable diagnosis of various respiratory disorders, such as idiopathic pulmonary fibrosis (IPF) and chronic obstructive pulmonary disease (COPD) 7]. An example of a HP 129Xe lung ventilation image can be seen in Figure 2 .
Early studies carried out by Happer and coworkers  used the N2 rotational temperature to monitor energy transport during optical pumping. These experiments primarily used pump cells loaded with helium and nitrogen gases, and operated under the same experimental conditions as SEOP (without the NMR-active noble gas) using a broadband (2 nm linewidth) 795 nm Rb D1 pump laser. The Raman setup shown in Figure 3(a) consisted of a 514.5 nm probe beam from an argon-ion laser (6 W) that could be focused to a 1 mm waist anywhere within the pump cell; the scattered light was collected using a planoconvex lens and directed into a Spex 1404 double Raman spectrometer (0.85 m focal length) and the photons were counted using a bi-alkali photomultiplier tube (PMT) with computer readout. The results of this work include the observation of convective heat transfer during optical pumping, as the N2 rotational temperature deviated from a linear relationship with absorbed pump laser power, shown in Figure 3(b) . This convective heat transfer was further studied by measuring the N2 temperature at differing spatial locations throughout the pump cell as a function of pump laser absorption. For low values of laser absorption (2.9 W), the calculated Rayleigh number (which provides a measure of convective vs. conductive transfer by comparing buoyant and viscous forces) indicates conductive heat transfer to the pump cell walls. As the absorption of the pump laser is increased to 15 W, the Rayleigh number indicates convective heat transfer; this is important in SEOP, as convection can affect the residing time of polarized 129Xe nuclei near the pump cell walls and lead to depolarization. Switching between a cell loading of predominantly helium gas with some nitrogen (1.0 and 0.33 amagat) to a cell with only high density nitrogen (3.2 amagat) increased the Rayleigh number by a factor of 70; simply changing the pump cell volume and geometry further increased the Rayleigh number by an additional factor of 29. This demonstrates how gas composition, density, and pump cell geometry can affect energy transport during SEOP.
Happer et al. also noticed that the N2 rotational temperature was elevated compared to the surface temperature of the glass walls of the pump cell (measured via thermocouple). The extent of the temperature elevation changed with absorbed pump laser power and gas loading (density/composition), and often varied from 100 to 200°C above the cell wall temperature. Because of the slower vibrational-to-rotational cross-relaxation, N2 vibrational temperatures were found to be significantly elevated in pump cells containing only nitrogen (
3.1. Orthogonal configuration
Recent advances in laser diode array (LDA) technology have resulted in not only an increase in overall laser power (hundreds of watts), but also a narrowing of the spectral output from several nanometers to a few tenths of a nanometer. This increase in resonant flux leads to more efficient absorption by the relatively narrow alkali metal D1 absorption line (c.a. ~0.1 nm), with the consequence of additional thermal management concerns . Furthermore, recent advances in using high xenon partial pressures (
Further work from Whiting et al.  conducted remote temperature sensing using Raman scattering to observe energy thermalization during Rb/129Xe SEOP under conditions of high xenon partial pressures and resonant laser powers. These studies used a laser diode array (~70 W) that was frequency-narrowed (~0.25 nm linewidth) using an internal volume holographic grating; this SEOP pump laser was spectrally tunable independent of output power to allow comparative studies at different excitation frequencies. The cylindrical optical pumping cell (2.5 cm inner diameter; 15.5 cm length) was surrounded by an outer glass cylinder that served as a forced-air oven and allowed 360° optical access . Miniature NMR coils were placed directly under the OP cell to allow correlative low-field NMR studies of HP 129Xe gas along with Raman scattering measurements of nitrogen buffer gas. The Raman apparatus shown in Figure 4 ( a , c ) consisted of a Verdi V5 532 nm excitation laser (5 W) that was fiber-coupled to a modular lens assembly box to focus the light into the OP cell. Scattered light was collected by a similar lens box (positioned orthogonal to the probe laser) that was fiber-coupled to a Horiba Jobin-Yvon U1000 double 1-m Raman spectrometer with 2400 l/mm grating and CCD receiver. Spatial filtering of the laser and Rayleigh scatter was achieved through adjustment of the internal baffles of the spectrometer; this provided spectra that displayed Stokes and anti-Stokes rotational Raman lines within 6 cm−1 of the laser line without the use of notch or edge filters—see Figure 4(b ). The N2 rotational temperature was calculated using the methods explained in Section 2.2, and calibration studies showed a good correlation with different N2 gas temperatures in the absence of optical pumping as shown in Figure 4(d ).
In this work, different xenon partial pressures were studied to determine the effects from increasing incident laser power on both 129Xe hyperpolarization level and N2 rotational temperature—see Figure 5(a, b) . Both HP 129Xe signal and N2 rotational temperatures increased with laser power for the different gas loadings (100/1900, 500/1500, and 1500/500 Torr Xe/N2), with the highest 129Xe signal and N2 temperature observed at the highest xenon partial pressure. This increased temperature at high xenon partial pressure was not primarily attributed to increased laser absorption, but rather to the combination of the lower concentration of nitrogen (increasing the energy absorption per N2) and low thermal conductivity of xenon (which was the significant fraction of total gas volume). Gas loadings with the lowest xenon partial pressure exhibited a relatively weak dependence of N2 rotational temperature on incident laser light. Despite the high nitrogen temperatures, the high xenon density loading provided the highest overall 129Xe NMR signal—typically the overall figure of merit for applications of hyperpolarized contrast agents.
Additional studies examined the effects of heating as a function of longitudinal position within the cylindrical pump cell. Because the external heat source was provided in the rear of the cell, and the forced air oven was poorly insulated, there was a significant temperature gradient (~30–40°C surface temperature) along the length of the cell. Both the 129Xe NMR signal and the N2 rotational temperature reflected the presence of this gradient and showed the maximal values were at the rear of the cell (close to the heating source). This type of back-to-front temperature gradient has both benefits and detriments. One benefit is that the total cell illumination may improve, as the optical density of absorbers gradually increases with penetration depth (unlike other OP systems, which may have conditions where all the laser light is absorbed in the front portion of the cell—leaving the rear of the cell optically opaque); this improved illumination should increase global HP 129Xe signal. On the other hand, an unfavorable result of temperature gradients is longitudinal convective heat transport, which potentially brings the polarized gas in close contact with paramagnetic relaxation centers in the cell wall surface for longer periods of time. These studies were carried out at different illumination wavelengths to examine how changes to the Rb D1 lineshape  (through altering the gas density and composition) affect the N2 rotational temperature. The highest HP 129Xe NMR signals and N2 rotational temperatures were found nearest to the Rb D1 line, and elevated values were observed at red-shifted excitation wavelengths (as expected, given the effects of different gas loadings on the Rb absorption lineshape ).
The effects of slight changes in temperature of the forced air oven on both HP 129Xe NMR signal and N2 rotational temperature over the course of an SEOP experiment were also monitored—see Figure 5(c, d ). Increasing the external cell temperature from ~110°C to ~140°C resulted in significant changes to these parameters. At the lower cell temperature, the N2 rotational temperature reached steady state (~190°C) quickly and held the value throughout the course of the experiment (40 min); correspondingly, other conditions within the OP cell remained stable (e.g., laser absorption, Rb density,
In summary, this work expanded on the previous studies of Happer et al.  by examining a regime of high resonant laser power and xenon-dominant binary gas mixtures in the absence of helium. It also used
3.2. In-line superhead
In an attempt to overcome the limitations of the orthogonal Raman probe arrangement, a confocal approach was taken whereby scattered photons were detected in the same plane as probe laser photon emission. To realize this, the same 532 nm probe laser and Horiba Jobin-Yvon U1000 double 1-m Raman spectrometer (from Section 3.1) were fiber-coupled instead to an on-axis optical module. This design greatly improved the sensitivity and reliability of the measurements due to the significantly reduced chance of misalignment relative to the orthogonal arrangement. Two ultra-narrow-band VHG notch filters were used to discard the vast majority of dominant Rayleigh scattered light, allowing resolution of both Stokes and anti-Stokes ultra-low frequency Raman lines as close as 10 cm−1 to the probe laser wavelength. Rayleigh scattered light was further reduced by an order of magnitude through the inclusion of an ultra-narrow-band beamsplitter filter, which also facilitated the minimization of spontaneous laser diode emissions and fluorescence, thus greatly improving the spectra of the incoming probe photons . Additionally, the entire configuration is mounted on a translational x-axis stage, allowing different positions within the optical cell to be probed independently. Schematics of the inline configuration are shown in Figure 6 , along with a typical Raman spectrum and temperature calibration plot to illustrate the improvements in sensitivity.
A comparison of Raman spectra acquired using the orthogonal and in-line methods was made under identical conditions at room temperature, with no pump laser illumination and over a 15 s integration time in an optical cell containing 100/1900 Torr Xe/N2. The results of this experiment are shown in Figure 7 , indicating a ~23-fold improvement in SNR , facilitating much more accurate determination of temperature. It can be seen from the spectra that additional Raman signals arise from atmospheric N2 and O2 along the optical path of the probe laser external to the optical cell. In later studies, these contributions to the Raman spectrum were mitigated by collection of spectra from an evacuated optical cell, which were subsequently subtracted from the data to remove “background” signals. Care must be taken during this correction process, however, as spectral drift can result in unrealistic difference spectra and less precise temperature measurements.
As an initial demonstration of temperature determination
4. Current work and future directions
Since demonstrating its utility and improved performance, the in-line Raman module has become central to our current work: a series of experiments that provide greater insight into xenon-rich stopped-flow SEOP. So far, these experiments have examined SEOP as a function of (1) buffer gas, in which the ratio of N2 to He was varied; (2) position within the optical cell, in which the portable nature of the in-line Raman module was harnessed to probe in-cell temperatures at multiple points along the length of the optical cell; and (3) alkali metal ratio (the so-called hybrid SEOP), in which pure Rb SEOP was compared to SEOP with Cs also present in the optical cell in varying amounts. The rationale, results, and implications of these experiments are described in the following sections.
4.1. Function of buffer gas/mixture composition
There are two primary buffer gases that are typically utilized in SEOP experiments—N2 and He. N2 is used to perform radiation trapping, quenching the emission of resonant unpolarized photons by excited alkali metal atoms that would otherwise depolarize other Rb atoms. He meanwhile was originally added in high fractions to collisionally broaden the absorption line of the alkali metal to more optimally couple the incident laser line , a job to which it was well suited due to its low Rb spin-destruction rate—see Table 1 . N2 also provides pressure-broadening effects, albeit at the cost of increased spin-destruction. This practice was necessary because the pump laser linewidths were initially at least an order of magnitude wider than the Rb D1 transition line, resulting in inefficient optical pumping. Although the role of He for pressure-broadening has become increasingly redundant as pump laser linewidths narrow to approach the Rb linewidth, its high thermal conductivity relative to N2 and Xe is important to note. In the emerging regime of 129Xe-rich stopped-flow SEOP, where both the poor thermal conductivity of the Xe and longer gas residence time in the cell compound thermal runaway issues, He may be uniquely situated to facilitate the dissipation of thermal energy from the optical cell, ultimately improving yields of polarized 129Xe.
|Gas||Thermal conductivity (10−2 W m−1 K−1)||Spin destruction rate (cm3 s−1)|
|Xenon||0.70||6.02 × 10−15 (T/298 K)|
|Nitrogen||3.09||3.44 × 10−18 (T/298 K)|
|Helium||17.77||3.45 × 10−19 (T/298 K)|
The impact of He on the thermal behavior within the optical cell can be seen in Figure 9 . Here, three SEOP experiments are compared in which the partial pressure of Xe and the total cell pressure were held constant at 100 Torr and 2000 Torr, respectively, but increasing amounts of He gas were substituted in place of N2. As the fraction of He within the cell was increased it can be seen that TN2 is reduced from roughly 200°C with no He present, to 100°C with 1400 Torr He included in the mixture. The onset of Rb runaway is evident from the sharp spike in 129Xe polarization in the first mix, where the polarization initially climbs sharply, peaking abruptly at ~6 min before dropping. At this point the high Rb vapor density creates optical opacity that limits the polarization process. It can be seen that this effect is suppressed as more He is added to the mixture at little cost to the ultimate 129Xe polarization. Given that existing 129Xe-rich stopped-flow polarizers operate with substantially higher powered lasers, which only further exacerbate these thermal management issues, it may be that He becomes a valuable tool for optimization.
4.2. Position-dependent temperature mapping
The experiments that vary buffer gas density/composition indicate that polarization and temperature are likely to vary along the length of the cell, particularly under conditions of Rb runaway. Figure 10 illustrates how Raman measurements were taken at coincident locations to NMR measurements at the front, middle, and back of the optical cell (a) using the translational stage (b). Figure 11 shows the results of two experiments where this configuration was used. Both experiments were conducted on a gas mix of 1000/1000 Torr Xe/N2 but at oven temperatures of 120 and 130°C, respectively. At 120°C, the temperature at the front, middle, and back correspond well to the xenon polarizations, which are both highest at the back for the cell. However, it can be seen that a mere 10°C increase in oven temperature is enough to induce significant variation in TN2 along the length of the cell. Closer examination shows that after 4 min of pump laser illumination during SEOP, when the 129Xe polarization prematurely spikes, a switch in the Raman temperatures occurs–indicative of a change in the thermal flow patterns within the cell. This effect is likely caused by Rb runaway, a conclusion corroborated by the elevated pump laser light absorption, which indicates a spike in absorbers.
These measurements as a function of position will be improved by introducing an automated translational stage that will permit more reliable positioning of the Raman probe relative to the cell with increased speed and precision. This will facilitate N2 temperature maps in two dimensions—both along the width and the length of the optical cell.
4.3. Function of alkali metal ratio (Rb-Cs hybrid SEOP)
Rubidium has long been the alkali metal of choice for SEOP, owing to its large absorption cross-section and the availability of affordable, high-powered lasers that are well matched to its D1 transition line. Additionally, Rb is solid at room temperature, with a low melting point that facilitates easy storage and transfer into optical cells. However, it has been shown that cesium can produce greater polarization under similar experimental conditions , but comparably high-powered lasers at the Cs D1 transition wavelength are not commonplace. Alternatively, for 3He hyperpolarization by SEOP, it was shown that using two alkali metals simultaneously—rubidium and potassium—resulted in four times faster build-up of net spin polarization once the optimal ratio of the alkali metals had been achieved . In this example it was Rb that underwent optical pumping, with angular momentum transferred to He by two distinct spin-exchange routes; one directly from Rb to He, and from Rb to He via K. The reason for the increase in speed is that the exchange cross sections for Rb-K and K-He are both larger than those for Rb-He alone. Inspired by these findings, we conducted analogous experiments in an Rb-Cs SEOP system to hyperpolarize 129Xe. The SE cross-sections for Rb-Cs and Cs-Xe have been shown to be greater than for Rb-Xe in separate experiments [27, 28]. If successful, the performance of existing polarizers could be improved with the simple addition of Cs to the existing Rb within the optical cell. As with previous studies, a 60 W pump laser tuned to the Rb D1 transition wavelength was used for polarization.
Figure 12(a) shows a summary of major hybrid results to date. From this figure it can be seen that, contrary to that suggested by the SE cross-sections, the resulting 129Xe polarization in our initial efforts decreases as more Cs is added to the alkali metal hybrid mixture. The data in Figure 12(b–g) can be used to explain this observation, indicating that 129Xe polarization and TN2 actually decrease with increasing Cs fraction. The latter is indicative of decreasing rates of optical pumping, likely caused by a suppression of the Rb vapor by a dominant Cs vapor. Indeed, the intrinsic vapor pressure of Cs is higher than that of rubidium due to its lower boiling point, a relation that is reversed in the analogous K-Rb system for 3He hyperpolarization. Thus, our Rb-Cs hybrid experiments to date indicate that the addition of Cs to the standard Rb-129Xe SEOP method serves only to suppress the 129Xe polarization. However, considering only a small number of different Rb-Cs ratios and conditions have been investigated, future studies on leaner Cs mixtures, where the Rb vapor density is not suppressed to such an extent, may yet yield a more favorable outcome.
An overview of gas temperature measurement using Raman spectroscopy and its potential as a diagnostic tool for improving understanding of energy transport processes during spin-exchange optical pumping has been discussed. The main approaches taken to correlate low-field NMR with Raman spectroscopy, using both orthogonal and in-line detection methods, have been compared and contrasted to relevant historical studies as well as each other to provide insight on poorly understood thermal processes present during SEOP. Future research will aim to build upon the in-line configuration, where automation of the translational stage could allow three-dimensional temperature mapping for improved characterization of these processes with greater resolution. Additionally, the demand for higher laser powers and richer Xe mixtures will likely result in thermal management becoming paramount, as the tightrope between maximizing polarization efficiency and avoiding the perilous “runaway” regime becomes narrower still.
This improved understanding of such thermal and energy transport processes should be directly translated to improving the next generation of clinical hyperpolarizers for use in producing HP noble gas for MRI studies in the lungs, which will hopefully facilitate safer, earlier, and more reliable diagnosis of a variety of serious and life-threatening diseases. Such advances will serve to streamline the diagnostic process, improving cost-effectiveness and reducing strain on medical services alongside the obvious benefit to those in society.