Maximum and minimum values of pressure in the endograft surface, for the different phases of the cardiac cycle. Excessively high values of pressure due to alteration in the iliac limbs geometry were excluded (outlier values).
Endovascular Treatment (EVAR) is considered the treatment of choice for the majority of Abdominal Aortic Aneurysms (AAA) nowadays, since it demonstrates improved perioperative morbidity and aneurysm-related mortality, comparingto conventional open repair. However, despite the initial technical success and early discharge of the patient, this technique is amenable to early and late complications, the most important of which are the endoleaks (ie. recurrence of blood flow detection within the aneurysm sac) accompanied sometimes with variable degrees of intrasac pressurization (Georgakarakos et al, 2012a). Furthermore, the hemodynamic changes that the endograft sustains during the follow-up period make it prone to positional changes with subsequent risk for endograft migration and loss of sealing between the endograft and either the aneurysm neck or the iliac fixation sites.
Computer-enhanced geometric modeling and Finite Volume Analysis have been used to study the biomechanical behavior of the aortic aneurysms before and after the insertion of the endograft device (Georgakarakos et al, 2012b). Numerical modeling of endovascular-treated AAA is used to determine the stresses and forces developed on AAA sac and stent-graft materials in-vivo, estimating hemodynamic parameters, such as the pressure and stress distribution over the main body, the bifurcation, the limbs of a stent-graft or the drag and displacement forces predisposing to graft migration. Consequently, the study of flow dynamics within aortic endografts holds a fundamental role in the delineation of the endograft behavior under pulsatile flow, providing useful information for developing and modifying the endograft design and surgical techniques. This chapter discusses the aforementioned changes, by using three-dimensional (3D) reconstructed endograft model.
2. Reconstruction of the Aaa endograft model
Finite Volume Analysis technique has a crucial role in the computational research of hemodynamic systems, utilizing small subsections (elements) of 3-dimentional structures created by segmentation and meshing.By solving Navier-Stokes equations for all finite volumes of the model, Computational flow dynamics (CFD) techniques utilize numerical methods and algorithms to analyze problems that involve fluid flows. Furthermore, Fluid Structure Interaction (FSI) methods combine fluid and structural equations, solved either simultaneously or separately (partitioned approach), in order to determine the flow fields and solid body stresses on a deformable model. Most researchers acquire information on the 3D AAA realistic, complex geometry using patient-specific DICOM data derived from high-resolution spiral CT or MR angiography (Georgakarakos et al, 2012b).
Our study group used a reconstructed 3D model of a AAA endograft using commercially available appropriate, validated software (MIMICS 13.0, Materialise NV, Leuven, Belgium), based on the DICOM images derived from contrast-enhanced high-resolution computed tomography. The computational model (Figure1) includes the aortic neck proximal to the endograft and the iliac arteries distal to the endograft limbs. A validated Finite Volume analysis software ANSYS v 12.1 (Ansys Inc., Canonsburg, PA, USA) was used for Computational Fluid Dynamics (CFD). The velocity and pressure waveforms during a period of 1.2 s as previously described in a one-dimensional fluid-dynamics model for the abdominal aorta(Olufsen et al, 2000 and Li et al, 2005) were used for both models as inlet and outlet boundary conditions. Blood was assumed to be non-Newtonian fluid, according to the Carreau-Yasuda model, with a density of 1050 kg/m3.
Accordingly, the velocity streamlines and the pressure distribution were calculated over the entire surface of the endograft and are demonstrated in 6 distinct time-phases through the cardiac cycle (Figure 2).For study reasons, the cardiac cycle was dived in six distinct phases, namely the late diastole (t1), the accelerating systolic phase (t2), the peak systolic phase (t3), the late deceleration (t4), the end-systolic (t5) and the early diastolic phase(t6).
3. Changes in flow patterns and pressure distribution
Figures 3-8 depict the flow patterns in the endograft throughout the cardiac cycle. A flow disturbance is seen near the inlet zone (panel top-left) during the late diastole, t1 (Figure 3). The flow pattern is normalized during the entire systolic phase, ie. t2 to t4 (Figures 4-6) and exhibits disturbance again, from the the end-systole t5early diastole t6(Figures 7,8). Interestingly, there is disturbed flow in the iliac limb unilaterally (left) during the decelerating systolic phase (Figure 6), whereas the irregular flow is also transmitted in the contralateral (right) iliac limb, during the next time-step (end-systolic phase, t5, Figure 7).
|Pressure values (mmHg)|
Table 1 depicts the minimum and maximum pressure values along the endograft surface, for the different phases of the cardiac cycle (as described above). As depicted in Figures 9-14 and Table 1, there is a similar, homogenous distribution of the pressure values along the different parts of the endograft during the diastolic phase (t6and t1). However, when it comes to the systolic phase (t2and t4), there is a marked linear decrease of the pressure values from the endograft inlet to the iliac limbs (outlet). The greatest pressure value difference is marked in the accelerating systolic phase (t2). Interestingly, the highest and lowest pressure values are demonstrated in the inlet-main body area and the iliac limbs of the endograft, respectively, during the accelerating and peak systolic phase, whereas this pressure relation is reversed in the decelerating systolic phase (t4), where the highest values are located distally (outflow). Moreover, there seems to be a narrower range of pressure distribution in the peak systolic phase (t3). Finally, in the early diastolic phase (t6) there is again a reverse in the pressure distribution compared to the early systolic phase, with the highest pressure being located in the inflow area of the endograft.
Figures 15-17 demonstrate the vertical velocity patterns and the secondary flow fields in the different parts of the endograft, during the peak systolic and the diastolic phase. The bifurcation of the endograft in two distinct outflow tracts (iliac limbs) causes a disturbance of flow especially in the secondary flow fields and generation of local vortices mainly in the proximal iliac parts (Figure 16), before this marked difference is subsided in the most distal iliac outflow parts (Figure 17). This pattern is also met in the diastolic phase, but with a greater discrepancy being present in this phase (Figures 15-17). In both iliac limbs there was a skewing of the flow towards the inner wall and significant flow separation towards the outer wall.
4. The forces exerted on the endograft surface
The forces applied on the surface of the endograft are demonstrated in Figures 18-20. The forces generated by the pressure are directed mainly vertical to the endograft surface (Figure 18) throughout the cardiac cycle. The tangential forces are mainly caused by theflow of blood and the boundary layer that is formed near the aortic wall, while their direction is depended on the cardiac phase. So, their vector heads forward during the early, peak (Figure 19) and late systolic phase, whereas the direction is reversed during the end systole (Figures 2 and 20) and late diastole. Notably, the values of the tangential forces are lower than the pressure ones by many orders of magnitude. The total sum of the pressure and viscous forces acting on the surface of the graft resolving into the x, y and z components, determines the drag forces that the endograft is subjected to, making it prone to migration.
(CFD) techniques provide a valuable and reliable tool in the study of the hemodynamic behavior of the cardiovascular system after therapeutic interventions (Frauenfelder, 2006).
The insertion of an endograft causes alterations in the hemodynamic environment of the AAA, regarding the pressures and stresses exerted on the AAA sac as well as the flow patterns inside the endograft lumen (Molony et al, 2009). There is a reduction in the intrasac pressure and the stress values on the sac of the stented AAA, leading to sac shrinkage. Chong and How (2004) used flow visualization and laser Doppler anemometry to study in vitro the flow patterns within a stent graft in different phases of the cardiac cycle. According to their study, the main trunk of the endograft is characterized by complex flow patterns with evidence of instability in systolic acceleration phase, developing into a number of vortical structures during systolic deceleration. The flow phenomena in the iliac limbs are strongly influenced by the geometry and the configuration of the limbs (Morris 2006 and Molony, 2008) and any degree of existing constriction caused in the iliac limbs. Basically, the flow in both limbs is triphasic, with a large retrograde component in end-systole (Chong and How, 2004) and formation of recirculating zones. The profiles are significantly more disturbed in the deceleration phase than at maximum velocity (Chong and How, 2004).
Finally, local geometric factors play a role in the determination of velocity values and flow patterns (recirculating zones, flow separation, skewed flow, vortices and Dean flows) with the out-of-plane endograft geometry determining greatly the outlet flow rates, flow patterns and drag forces (Morris, 2006). Extrinsic constriction (due to calcified or stenosed iliac vessels) or excessive kinking in the iliac limbs can lead either to thrombosis of the graft limbs or altered flow patterns that induce excessive disturbances in shear stresses (not shown in our model), leading also to recirculating zones and prolonged transit times of platelets with consequent apposition and formation of thrombus in the endografts. The latter constitutes a rather common incidental finding, occurring more frequently that previously assumed (Wu et al, 2009). Finally, the study and understanding of the hemodynamic alterations and the parameters that influence them, could lead to better designs of endovascular grafts, in order to eliminate the factors that predispose to endograft migration as well as to generation of endoleaks (Figueroa 2009 and 2010, Liffman 2001, Mohan 2002).
Aortic endografts are subject to hemodynamic alterations that determine the flow patterns within the different parts of the endografts and influence the values and distribution of pressures and stresses onto their surface during the different phases of the cardiac cycle. Certain geometric factors such as the inlet-to-outlet ratio of the graft as well as the out-of-plane configuration of the main body and iliac limbs have been implicated as major determinants of the aforementioned hemodynamic alterations. Computational simulation techniques can help towards the understanding of these interactions and help us further design better endografts with greater resistance to migration, endoleaks and dislocation of modular stent-grafts, all of which are influenced by the hemodynamic environment that endografts are exposed to.
Georgakarakos E. Georgiadis G. S. Ioannou C. V. Kapoulas K. C. Trellopoulos G. Lazarides M. 2012aAneurysm sac shrinkage after endovascular treatment of the aorta: beyond sac pressure and endoleaks. Vasc Med.; 17 168 73
Georgakarakos E. Georgiadis G. S. Xenakis A. Kapoulas K. C. Lazarides M. K. AS Tsangaris Ioannou. C. V. 2012bApplication of bioengineering modalities in vascular research: evaluating the clinical gain. Vasc Endovascular Surg.; 46 101 8
Chong CK, How TV, Gilling-Smith GL, Harris PL. 2003Modeling endoleaks and collateral reperfusion following endovascular AAA exclusion. J Endovasc Ther.; 10 424 32
Chong CK, How TV. 2004Flow patterns in an endovascular stent-graft for abdominal aortic aneurysm repair. J Biomech.; 37 89 97
CA Figueroa Taylor. CA Yeh V. Chiou A. J. Zarins C. K. 2009Effect of curvature on displacement forces acting on aortic endografts: a 3-dimensional computational analysis. J Endovasc Ther.; 16 284 94
CA Figueroa Taylor. CA Yeh V. Chiou A. J. Gorrepati M. L. Zarins C. K. 2010Preliminary 3D computational analysis of the relationship between aortic displacement force and direction of endograft movement. J Vasc Surg.; 51 1488 97
Frauenfelder T. Lotfey M. Boehm T. Wildermuth S. 2006Computational fluid dynamics: hemodynamic changes in abdominal aortic aneurysm after stent-graft implantation. Cardiovasc Intervent Radiol.; 29 613 23
Li Z. Kleinstreuer C. Farber M. 2005Computational analysis of biomechanical contributors to possible endovascular graft failure. Biomech Model Mechanobiol.; 4 221 34
Liffman K. MM Lawrence-Brown Semmens. J. B. Bui A. Rudman M. Hartley D. E. 2001Analytical modeling and numerical simulation of forces in an endoluminal graft. J Endovasc Ther.; 8 358 71
Mohan IV, Harris PL, Van Marrewijk CJ, Laheij RJ, How TV. 2002Factors and forces influencing stent-graft migration after endovascular aortic aneurysm repair. J Endovasc Ther.; 9 748 55
DS Molony Callanan. A. Morris L. G. Doyle B. J. Walsh M. T. Mc Gloughlin T. M. 2008Geometrical enhancements for abdominal aortic stent-grafts. J Endovasc Ther.; 15 518 29
DS Molony Callanan. A. Kavanagh E. G. Walsh M. T. Mc Gloughlin T. M. 2009Fluid-structure interaction of a patient-specific abdominal aortic aneurysm treated with an endovascular stent-graft. Biomed Eng Online.6;8: 24 EOF
Morris L. Delassus P. Grace P. Wallis F. Walsh M. Mc Gloughlin T. 2006Effects of flat, parabolic and realistic steady flow inlet profiles on idealised and realistic stent graft fits through Abdominal Aortic Aneurysms (AAA). Med Eng Phys.; 28 19 26
MS Olufsen Peskin. C. S. Kim W. Y. Pedersen E. M. Nadim A. Larsen J. 2000Numerical simulation and experimental validation of blood flow in arteries with structured-tree outflow conditions. Ann Biomed Eng.; 28 1281 99
Wu IH, Liang PC, Huang SC, Chi NS, Lin FY, Wang SS. 2009The significance of endograft geometry on the incidence of intraprosthetic thrombus deposits after abdominal endovascular grafting. Eur J Vasc Endovasc Surg.; 38 741 7