Knowing the nonlinear optical properties such as nonlinear refraction, nonlinear absorption and rotational viscosity in dye-doped nematic liquid crystals (NLCs) is crucially important for photonic applications like liquid crystal display (LCD). Versatile optical methods have been developed for measuring the nonlinear refraction and nonlinear absorption including third harmonic generation (deSouza et al., 1999), degenerate four wave mixing (Liao et al., 1998), optical Kerr gate (Gomes et al., 1996), nonlinear interferometry (Yavrian et al., 1999), ellipse rotation (Lefkiry et al., 1998), phase distortion method (Watanabe et al., 1995), among others. Although these techniques are sensitive, they require complicated experimental setup to measure the optical nonlinearities. The Z-scan technique (Sheik-Bahae et al., 1990), is a simple and highly sensitive single beam method that uses the principle of spatial beam distortion to measure both the sign and the magnitude of the optical nonlinearity. Nowadays, the Z scan method has been most popularly used.
To determine the rotational viscosity coefficient, various experimental methods have been extensively investigated which include: the light scattering measurement (Durand et al., 1995), capillary flow measurement (Gähwiller, 1971), rotating magnetic field technique (Prost & Gasparoux, 1971), shear-waves reflectance technique (Kneppe et al., 1982; Martinoty & Candau, 1971), nuclear magnetic resonance technique (Martins et al., 1986), capacitance method (Leenhouts, 1985) and transient current method (Imai et al., 1995). Among them most widely used techniques are capacitance method and transient current method. Capacitance method measures the time constant of an exponentially decaying director relaxation from a non-equilibrium to an equilibrium state in a twisted nematic cell upon switching on or off the applied voltage. Transient current method measures the peak amplitude of the transient current induced by a direct-current voltage pulse application.
In this chapter, we propose simple and accurate optical methods for determining the nonlinear refractive coefficient, the nonlinear absorption and the rotational viscosity coefficient in dye-doped nematic liquid-crystal and also present the theories developed. This chapter is organized as follows. In section 2, the theoretical backgrounds are provided for closed- and open- aperture Z-scan transmittance, knife-edge X-scan and modified Z-scan by switching on or off the applied electric field, taking into account two photon absorption. In section 3, a series of experiments are performed and the experimental results are discussed, and finally, conclusions are drawn in section 4.
2.1. Closed and open aperture Z-scan theory for complex optical nonlinearity
In this section, we present the optical method to determine the nonlinear refraction and nonlinear absorption by Z-scan technique. The nonlinear medium is scanned along the z-axis in the back focal region of an external lens, and the far-field on-axis (i.e., closed aperture) transmittance and the whole (i.e., open aperture) transmittance are monitored as a function of the scan distance z. The open aperture Z-scan transmittance is insensitive to the nonlinear refraction and solely determines the nonlinear absorption, whilst the closed aperture Z-scan transmittance is coupled with both of the nonlinear effects. Actually, both the nonlinear refraction and the nonlinear absorption are often present simultaneously in nonlinear optical materials. Nonlinear absorption is inevitably present for resonant absorption wavelength ranges as well as for transparent regions owing to multi-photon absorption when the laser beam intensity is sufficiently high or because of other nonlinear processes. For simplicity, we only concentrate on two-photon absorption (TPA). Consider the fundamental Gaussian electric field (TEM00 mode) of travelling in the z direction as
where n0 is the linear refractive index, n2 is the nonlinear refractive coefficient,
According to the aberration-free approximation of a Gaussian beam, which requires the Gaussian beam profile be approximated as being parabolic, by expanding the exponential in the intensity and retaining only the quadratic term, the nonlinear phase shift, Eq. (2b) can be approximated as:
2.2. Knife-edge X-scan theory for nonlinear absorption
In this section, we propose an alternative optical method for determining the nonlinear absorption coefficient, so-called knife-edge X-scan method. The knife-edge scanning technique is a simple single beam method for measuring a laser beam profile such as the beam radius and the radius of curvature of the wave front (Suzaki & Tachibana, 1975). Due to its high accuracy, simple apparatus and easy to data analysis, the knife-edge scanning method has been widely used. As the knife-edge along the x-axis moves across the beam propagation direction, the beam power at the far-field gradually decreases and eventually goes to zero. For a Gaussian beam distribution, the (measured) beam power is given by integrating the Gaussian function from negative infinity to present knife-edge position and becomes the error function.
Figure 1 represents schematic diagram for the knife-edge X-scan method proposed in this work to determine the nonlinear absorption coefficient.
The knife edge is positioned in front of a nonlinear optical medium placed at the focus (i.e., z=0) and is transversely scanned to the beam propagation axis from negative infinity to present knife-edge position. In case of two photon absorption process, the variation of beam power for a fundamental Gaussian laser beam passing through the medium can be written as
Figure 2 represents theoretical curves for normalized transmitted power and its derivative relative to knife-edge position x’ for various nonlinear absorbance q0=–0.5 and +0.5. Note that the first term (i.e., m=0) in Eq.(9) reveals one dimensional Gaussian beam power without nonlinear material (i.e., q0=0) for knife-edge X-scan. For negative nonlinear absorption (i.e., q0<0 or amplification), the beam radius or full width at half maximum (FWHM) decreases when compared with q0=0, while for positive nonlinear absorption (i.e., q0>0 or real absorption), the beam radius is much broaden than that of q0=0.
2.3. Orientational nonlinear refraction kinetics in nematic liquid crystals for rotational viscosity: Modified closed-aperture Z-scan
In this section, we will derive the kinetics of orientational refractive index change via director axis torque of nematic liquid crystals (NLCs), which is caused by a Gaussian optical field with/without an applied electric field. We also present a simple and accurate method to measure the rotational viscosity, the response time and the orientational nonlinear refraction in NLCs by modifying the closed Z-scan. Figure 3 shows the experimental setup. The optical method proposed in this work has basically the same experimental geometry used in closed aperture Z-scan. The sole distinction is that the NLC sample is placed at focus (i.e., z=0) of an external lens and is fixed at that place during the experiments, unlike Z-scan technique.
Before supplying an external electric field by a function generator, a focused optical beam is continuously illuminated to the sample, producing the optical field-induced director axis reorientation (Khoo, 1995), which gives rise to the orientational Kerr effect (OKE) and is given by
where γ1 is the rotational viscosity coefficient,
Eqs.(13) can be readily solved by using the boundary conditions of which
Similary, for the case of
The solution to Eqs.(15) is given by
Since the orientational refractive index is proportional to
3. Experiments and discussions
3.1. Sample preparation of nematic liquid crystals cell
We fabricated porphyrin:Zn-doped nematic liquid crystal (NLC) cells filled by capillary phenomenon between two transparent indium-tin-oxide coated glass substrates with 20 μm thick beads as a spacer. Two glass substrates were assembled by UV bond and then filled inside of cells with porphyrin:Zn-doped nematic liquid crystal for various concentrations of dye (0, 0.006, 0.13, 0.50wt%). The liquid crystal used was the eutectic liquid crystal mixture, commercially known as E7 (Merck Ltd.), which has a positive dielectric anisotropy Δε=13.8, the elastic constants
The transmission spectrum for pure E7 NLC cell reveals nearly transparent of about 90 % in visible wavelength range, as shown in Fig. 5. As increasing the concentrations of dye the transmisstion spectrum is gradually decreased. It is also shown from Fig. 5 that Zn-doped porphyrin dye is photosensitive to blue-green wavelength region. The linear absorption coefficients for various dye concentrations at wavelength 632.8nm were estimated by using the Beer-Lambert law
3.2. Determinations of nonlinear absorption coefficient by using knife-edge X-scan and open-aperture Z-scan
In this section, we determine the nonlinear absorption coefficients for various dye concentrations in NLC sample by means of knife-edge X-scan method and open-aperture Z-scan method and compare the experimental results quantitatively. Figure 1 represents the schematic diagram for the knife-edge X-scan method. The cw He-Ne laser of wavelength λ=632.8nm is used for experiments and the laser beam power is 3mW. The focal length of biconvex lens is 20cm. The whole transmitted power is measured by a photo detector during the knife-edge scan. Before conducting the knife-edge X-scan experiment, we have to determine the incident Gaussian laser beam profiles such as beam radius w(z), beam waist w0 and radius of curvature of the wave front R(z) at z. Figure 6(a) shows the typical experimental results of normalized power for knife-edge scan against scan x’ distance at several, which are well fitted with the theoretic formula as
To determine the nonlinear absorption coefficient of the sample we performed two kinds of experiments; one is the knife-edge X-scan in which the sample is placed at rear face of the knife-edge, as shown in Fig. 1, and the other is the conventional open-aperture Z-scan. Since the closed-aperture Z-scan transmittance is entangled with the nonlinear refraction and the nonlinear absorption, as described in Eq.(5), one should determine the nonlinear absorption coefficient before finding the nonlinear refractive coefficient. Once the nonlinear absorption coefficient is extracted from the open aperture Z-scan or the knife-edge X-scan, one can extracts the remaining unknown nonlinear refractive coefficient n2 from the closed aperture Z-scan transmittance. Figure 7 represents the typical experimental results of the knife-edge X-scan and the open aperture Z-scan for various dye concentrations with the theoretical predictions.
Table 1 compares the nonlinear absorption coefficient for various dye concentrations, determined by the knife-edge X-scan method with the open aperture Z-scan method. Nonlinear absorption coefficients determined by two methods are in good agreement with each other.
3.3. Determinations of nonlinear refractive coefficient by using closed aperture Z-scan
Figure 8 depicts the typical closed aperture Z-scan data, revealing a self-defocusing nature. The nonlinear refractive coefficients are determined from the best curve fitting using Eq.(5) with the known nonlinear absorption coefficients obtained from preceding subsection.
3.4. Determinations of rotational viscosity by modified closed-aperture Z-scan
Following the method described in subsection 2.3, we conducted the transient optical transmittance experiments by applying the rectangular electric field with the pulse duration time of
Table 2 compares the rotational viscosity coefficient and nonlinear refractive index coefficient for various dye concentrations, determined by the transient optical transmittance (or modified Z-scan) method with the closed aperture Z-scan method. Nonlinear refractive index coefficient, which is determined by two methods are in good agreement with each other. It reveals that the physical mechanism of the Kerr effect in NLC is caused by the optical field-induced director axis reorientation. It is also noted that the measured value of the rotational viscosity coefficient of 0.23Pas for pure E7 is almost the same value of 0.224Pas at 25 C by means of transient current method (Chen & Lee, 2007).
In this chapter, we propose simple and accurate optical methods to determine the nonlinear refraction, the nonlinear absorption and the rotational viscosity coefficient in dye-doped nematic liquid crystals, and also develop the corresponding theories. The versatile optical methods presented are as follows: (i) closed aperture Z-scan for measuring both the sign and the magnitude of the optical nonlinear refraction, taking into accounting two photon absorption, (ii) open aperture Z-scan and knife-edge X-scan for measuring the nonlinear absorption coefficient unambiguously, and (iii) modified closed aperture Z-scan (or transient optical transmittance) method by applying a rectangular electric field to measure the orientational Kerr effect (OKE) and the rotational viscosity coefficient. The measured values of optical nonlinearities and the rotational viscosity by an optical method are cross-checked by another method, showing excellent agreement with each other.